# Real-Time Correction of a Laser Beam Wavefront Distorted by an Artificial Turbulent Heated Airflow

^{*}

## Abstract

**:**

## 1. Introduction

## 2. The Fast Adaptive Optical System

#### 2.1. Deformable Mirror

#### 2.2. Shack–Hartmann Wavefront Sensor

#### 2.3. Adaptive Optical System Control Loop

## 3. Experimental Setup

## 4. Processing the Results of the Experiment

^{(−5/3)}corresponds to the Kolmogorov spectrum. The sampling duration of the focal spot coordinates was 10 s (at a frequency of 2000 Hz, 20,000 samples were recorded), which provided a resolution along the frequency axis of 0.1 Hz.

- A sample of the offset coordinates of the focal spots of the lens array of the WFS with a duration of 10 s was recorded. This made it possible to achieve a resolution along the frequency axis of 0.1 Hz. At a frequency of 2000 Hz, a total of 20,000 values of coordinate offsets of each focal spot were recorded
- The transition was carried out from sampling by coordinates to sampling by the coefficients of the wavefront expansion by Zernike polynomials. In this work, we used a set of 24 Zernike polynomials in Wyant indices [23] (1 and 2—tilts, 3—defocus, 4 and 5—astigmatism, 6 and 7—coma, 8—spherical, etc.).
- Using the Fourier transform, the transition from the time domain to the frequency domain was carried out for sampling each Zernike polynomial;
- The power spectral density was calculated for each mode;
- Further, by integrating the spectral power density, the spectral energy was calculated.

## 5. Discussion

- The combination of FPGA performance and a bimorph wavefront corrector in an adaptive optical system allows correction of artificially created turbulence in real time. The correction frequency in the experiments was chosen to be equal to 2000 Hz. Such a stable correction frequency is almost impossible to obtain using a conventional PC. The PC, unlike the FPGA, performs I/O at the driver level, thereby increasing the time of the closed correction cycle. FPGA exchanges data between external devices (wavefront sensor and corrector control unit) directly. In addition, the FPGA performs parallel processing of information, which has significant limitations in the case of using a PC.
- The speed of the AOS controlled by the FPGA made it possible to analyze the effectiveness of aberration correction up to the 23rd Zernike polynomial, whose bandwidth is about 100 Hz, in detail.
- The bimorph WFC does not have the ability to correct for the slopes of the WF. Accordingly, the correction efficiency of the first two Zernike polynomials is negative, i.e., there is an increase in the amplitude of the initial slopes. To eliminate the slopes, it is necessary to use either a separate beam position stabilization system (see, for example, [25]), or to install a mirror in a tip–tilt mount. In this case, the virtual slopes used in the experiment become real and the voltages calculated during operation are applied to the control drives of the tip–tilt mount.
- WFC used in the experiments has three rings of electrodes and cannot reproduce a spherical aberration of the third order (polynomial # 24), since this aberration has four extremums (max–min). To reduce the aberration # 24, a higher spatial resolution WFC is required.
- The aberrations from 3 to 23 are compensated well enough by this WFC. However, because the initial amplitude of the polynomial # 24 is small compared to other aberrations, the undercompensating of this aberration can be neglected.

## 6. Conclusions

## Author Contributions

## Funding

## Data Availability Statement

## Conflicts of Interest

## References

- Huang, Q.; Liu, D.; Chen, Y.; Wang, Y.; Tan, J.; Chen, W.; Liu, J.; Zhu, N. Secure free-space optical communication system based on data fragmentation multipath transmission technology. Opt. Express
**2018**, 26, 13536–13542. [Google Scholar] [CrossRef] [PubMed] - Vorontsov, M.; Weyrauch, T.; Carhart, G.; Beresnev, L. Adaptive optics for free space laser communications. In Lasers, Sources and Related Photonic Devices; OSA Technical Digest Series (CD), LSMA1; Optical Publishing Group: Washington, DC, USA, 2010. [Google Scholar] [CrossRef]
- Lu, M.; Bagheri, M.; James, A.P.; Phung, T. Wireless charging techniques for UAVs: A review, reconceptualization, and extension. IEEE Access
**2018**, 6, 29865–29884. [Google Scholar] [CrossRef] - Landis, G.A.; Westerlund, H. Laser Beamed Power—Satellite Demonstration Applications; NASA Contractor Report 190793, IAF-92-0600; NASA: Washington, DC, USA, 1992. [Google Scholar]
- Bennet, F.; Conan, R.; D’Orgeville, C.; Dawson, M.; Paulin, N.; Price, I.; Rigaut, F.; Ritchie, I.; Smith, C.; Uhlendorf, K. Adaptive optics for laser space debris removal. Proc. SPIE
**2012**, 8447, 844744. [Google Scholar] [CrossRef] [Green Version] - Tatarskii, V.I. Wave Propagation in a Turbulent Medium; McGraw-Hill: New York, NY, USA, 1961; 304p. [Google Scholar]
- Rui, W.; Yukun, W.; Chengbin, J.; Xianghui, Y.; Shaoxin, W.; Chengliang, Y.; Zhaoliang, C.; Quanquan, M.; Shijie, G.; Li, X. Demonstration of horizontal free-space laser communication with the effect of the bandwidth of adaptive optics system. Opt. Commun.
**2019**, 431, 167–173. [Google Scholar] [CrossRef] - Weyrauch, T.; Vorontsov, M. Free-space laser communications with adaptive optics: Atmospheric compensation experiments. Free-Space Laser Communications. Opt. Fiber Commun. Rep.
**2004**, 2, 247–271. [Google Scholar] [CrossRef] - Shen, Q.; Guan, J.-Y.; Ren, J.-G.; Zeng, T.; Hou, L.; Li, M.; Cao, Y.; Han, J.-J.; Lian, M.-Z.; Chen, Y.-W.; et al. 113 km Free-Space Time-Frequency Dissemination at the 19th Decimal Instability. arXiv
**2022**, arXiv:2203.11272. [Google Scholar] [CrossRef] - Brigham, E.O. The Fast Fourier Transform and Its Applications; Prentice Hall: Upper Saddle River, NJ, USA, 1988; 448p. [Google Scholar]
- Hardy, J. Active optics: A new technology for the control of light. Proc. IEEE
**1978**, 66, 651–697. [Google Scholar] [CrossRef] - Cao, G.; Yu, X. Accuracy analysis of a Hartmann–Shack wave front sensor operated with a faint object. Opt. Eng.
**1994**, 33, 2331–2335. [Google Scholar] [CrossRef] [Green Version] - Rukosuev, A.L.; Nikitin, A.N.; Sheldakova, Y.V.; Kudryashov, A.V.; Belousov, V.N.; Bogachev, V.A.; Volkov, M.V.; Garanin, S.G.; Starikov, F.A. Smart adaptive optical system for correcting the laser wavefront distorted by atmospheric turbulence. Quantum Electron.
**2020**, 50, 707–709. [Google Scholar] [CrossRef] - Born, M.; Wolf, E. Principles of Optics: Electromagnetic Theory of Propagation, Interference and Diffraction of Light, 1st ed.; Pergamon Press: London, UK; New York, NY, USA; Paris, France, 1959; 986p. [Google Scholar]
- Rukosuev, A.; Nikitin, A.; Belousov, V.; Sheldakova, J.; Toporovsky, V.; Kudryashov, A. Expansion of the Laser Beam Wavefront in Terms of Zernike Polynomials in the Problem of Turbulence Testing. Appl. Sci.
**2021**, 11, 12112. [Google Scholar] [CrossRef] - Fried, D.L. Optical Resolution Through a Randomly Inhomogeneous Medium for Very Long and Very Short Exposures. J. Opt. Soc. Am.
**1966**, 56, 1372–1379. [Google Scholar] [CrossRef] - Belousov, V.; Bogachev, V.; Volkov, M.; Garanin, S.; Kudryashov, A.; Nikitin, A.; Rukosuev, A.; Starikov, F.; Sheldakova, Y.; Shnyagin, R. Investigation of spatial and temporal characteristics of turbulent-distorted laser radiation during its dynamic phase correction in an adaptive optical system. Quantum Electron.
**2021**, 51, 992–999. [Google Scholar] [CrossRef] - Sarazin, M.; Roddier, F. The ESO Differential Image Motion Monitor. Astron. Astrophys.
**1990**, 227, 294–300. [Google Scholar] - Kudryashov, A.; Samarkin, V.; Aleksandrov, A. Adaptive optical elements for laser beam control. Proc. SPIE
**2001**, 4457, 170–178. [Google Scholar] [CrossRef] - Toporovskii, V.; Skvortsov, A.; Kudryashov, A.; Samarkin, V.; Sheldakova, Y.; Pshonkin, D. Flexible bimorphic mirror with high density of control electrodes for correcting wavefront aberrations. J. Opt. Technol.
**2019**, 86, 32–38. [Google Scholar] [CrossRef] - Nikitin, A.; Sheldakova, J.; Kudryashov, A.; Borsoni, G.; Denisov, D.; Karasik, V.; Sakharov, A. A device based on the Shack-Hartmann wave front sensor for testing wide aperture optics. Proc. SPIE
**2016**, 9754, 97540K. [Google Scholar] [CrossRef] - Kudryashov, A.; Rukosuev, A.; Nikitin, A.; Galaktionov, I.; Sheldakova, J. Real-time 1.5 kHz adaptive optical system to correct for atmospheric turbulence. Opt. Express
**2020**, 28, 37546–37552. [Google Scholar] [CrossRef] - Goodwin, E.; Wyant, J. Field Guide to Interferometric Optical Testing; SPIE Press Book: Bellingham, WA, USA, 2006; 114p, ISBN 0-8194-6510-0. [Google Scholar]
- Taylor, G.I. The Spectrum of Turbulence. Proc. Roy. Soc.
**1938**, 164, 476–490. [Google Scholar] [CrossRef] [Green Version] - Saathof, R.; Breeje, R.; Klop, W.; Doelman, N.; Moens, T.; Gruber, M.; Russchenberg, T.; Pettazzi, F.; Human, J.; Mata Calvo, R.; et al. Pre-correction adaptive optics performance for a 10 km laser link. In Free-Space Laser Communications XXXI; SPIE: Bellingham, WA, USA, 2019; Volume 10910, pp. 325–331. [Google Scholar] [CrossRef] [Green Version]

**Figure 2.**Bimorph mirror used in experiments: (

**a**) Mirror photo; (

**b**) electrodes structure. The numbers indicate the channels of the mirror. The first electrode, conventionally shown in the lower-left corner, has the dimensions of the mirror aperture and is used to control the defocusing of the wavefront.

**Figure 4.**FPGA timing diagram. The total time of one closed cycle is 500 microseconds, which corresponds to an operating frequency of 2000 Hz.

**Figure 5.**Adaptive optical system experimental setup. WFS—wavefront sensor; mirror CU—mirror control unit.

**Figure 6.**Spectral power density of the oscillation of the X coordinate of the WFS lens array focal spot.

Parameter | Value |
---|---|

Clear aperture | 50 mm |

Electrodes number | 31 |

Control voltage range | −200 V ± 300 V |

Maximal stroke | ±10 µm |

First resonant frequency | 8.3 kHz |

Coating | Protected silver |

Size | Ø 70 mm × 68 mm |

Weight | 320 g |

Parameter | Value |
---|---|

Sensor | Alexima LUX19HS |

Spectral bandwidth | 350–1100 nm |

Dynamic Range (Tilt) | ±50λ |

Accuracy of measurements | λ/90 |

Frame rate | 2500 fps @ 1920 × 1080 |

~4000 fps @ 480 × 480 | |

Interface | Fiber Optic 40 Gb/s |

Lenslet array focal length | 12 mm |

Number of working sub-apertures | 20 × 20 |

Input light beam size | 4.8 × 4.8 mm |

Resolution | 8 bit |

Publisher’s Note: MDPI stays neutral with regard to jurisdictional claims in published maps and institutional affiliations. |

© 2022 by the authors. Licensee MDPI, Basel, Switzerland. This article is an open access article distributed under the terms and conditions of the Creative Commons Attribution (CC BY) license (https://creativecommons.org/licenses/by/4.0/).

## Share and Cite

**MDPI and ACS Style**

Rukosuev, A.; Nikitin, A.; Toporovsky, V.; Sheldakova, J.; Kudryashov, A.
Real-Time Correction of a Laser Beam Wavefront Distorted by an Artificial Turbulent Heated Airflow. *Photonics* **2022**, *9*, 351.
https://doi.org/10.3390/photonics9050351

**AMA Style**

Rukosuev A, Nikitin A, Toporovsky V, Sheldakova J, Kudryashov A.
Real-Time Correction of a Laser Beam Wavefront Distorted by an Artificial Turbulent Heated Airflow. *Photonics*. 2022; 9(5):351.
https://doi.org/10.3390/photonics9050351

**Chicago/Turabian Style**

Rukosuev, Alexey, Alexander Nikitin, Vladimir Toporovsky, Julia Sheldakova, and Alexis Kudryashov.
2022. "Real-Time Correction of a Laser Beam Wavefront Distorted by an Artificial Turbulent Heated Airflow" *Photonics* 9, no. 5: 351.
https://doi.org/10.3390/photonics9050351