In clinical practice, it is recommended to employ reusable electrodes for the registration of brain waves. Before registering EEG signals, the EEG technician checks the condition of all the electrodes, i.e., the occurrence of mechanical damage and the color of the electrode coating. It should be noticed that there is still no information on the permissible number of EEG examinations performed with one set of electrodes. After placement of the electrodes on the patient’s head, the scalp–electrode impedance is measured with the use of EEG equipment. When the scalp–electrode impedance achieves a value above 5 kΩ, it is necessary to replace the given electrode or to re-execute skin abrasion. The Electrochemical Impedance Spectroscopy (EIS) method was used in order to estimate the permissible number of EEG examinations performed with one set of electrodes. Ten new reusable electrodes were tested. Then, the tests were repeated after subsequent uses of those electrodes. The conducted tests led us to the conclusion that the permissible number of examinations performed with one set of electrodes is up to twenty except for the gold electrodes for which it is up to ten. Furthermore, the use of the EIS method revealed variability of impedance in the case of new electrodes.
This is an open access article distributed under the Creative Commons Attribution License
which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited