Design and Development of a Three-Component Force Sensor for Milling Process Monitoring
AbstractA strain-type three-component table dynamometer is presented in this paper, which reduces output errors produced by cutting forces imposed on the different milling positions of a workpiece. A sensor structure with eight parallel elastic beams is proposed, and sensitive regions and Wheastone measuring circuits are also designed in consideration of eliminating the influences of the eccentric forces. To evaluate the sensor decoupling performance, both of the static calibration and dynamic milling test were implemented in different positions of the workpiece. Static experiment results indicate that the maximal deviation between the measured forces and the standard inputs is 4.58%. Milling tests demonstrate that with same machining parameters, the differences of the measured forces between different milling positions derived by the developed sensor are no larger than 6.29%. In addition, the natural frequencies of the dynamometer are kept higher than 2585.5 Hz. All the measuring results show that as a strain-type dynamometer, the developed force sensor has an improved eccentric decoupling accuracy with natural frequencies not much decreased, which owns application potential in milling process monitoring. View Full-Text
Share & Cite This Article
Li, Y.; Zhao, Y.; Fei, J.; Qin, Y.; Zhao, Y.; Cai, A.; Gao, S. Design and Development of a Three-Component Force Sensor for Milling Process Monitoring. Sensors 2017, 17, 949.
Li Y, Zhao Y, Fei J, Qin Y, Zhao Y, Cai A, Gao S. Design and Development of a Three-Component Force Sensor for Milling Process Monitoring. Sensors. 2017; 17(5):949.Chicago/Turabian Style
Li, Yingxue; Zhao, Yulong; Fei, Jiyou; Qin, Yafei; Zhao, You; Cai, Anjiang; Gao, Song. 2017. "Design and Development of a Three-Component Force Sensor for Milling Process Monitoring." Sensors 17, no. 5: 949.
Note that from the first issue of 2016, MDPI journals use article numbers instead of page numbers. See further details here.