Home blood pressure monitors are widely used by consumers yet cardiovascular health may be better defined by pulse-wave velocity (PWV). So far, the Withings Body Cardio scale is the only consumer device that has been designed to measure PWV and body composition, including fat mass (FM) and fat-free mass (FFM), in the home setting. While one study has demonstrated that this device meets the acceptable accuracy standards of the ARTERY Society, no study has accounted for the gravitational effect of standing on a scale on aortic-leg PWV. Purpose: The purpose of this study was to assess the accuracy of PWV and body composition as determined by the Body Cardio scale. Methods: Measurements of PWV and body composition in healthy, young males and females (n
= 20) using the Body Cardio device were compared to PWV assessed by applanation tonometry (SphygmoCor) and body composition analysis determined by air displacement plethysmography (Bod Pod). Bland–Altman analysis and mean absolute percent error (MAPE) were used to assess accuracy. Results: Data are reported as the mean bias (95% confidence interval). The Body Cardio overestimated PWV by 0.68 m/s (−0.16, 1.51) and FM by 2.91 kg (−2.91, 8.73). Body Cardio PWV and FM estimations had a MAPE of 9.7% and 25.8%, respectively. The Body Cardio underestimated body mass (BM) and FFM by 0.11 kg (−0.41, 0.18) and 2.87 kg (−9.04, 3.30), respectively. Body Cardio BM and FFM estimations had a MAPE of 0.15% and 5.6%, respectively. Conclusions: The Body Cardio scale provides accurate measures of BM and PWV; however, it should be used cautiously for measures of FM and FFM.
This is an open access article distributed under the Creative Commons Attribution License
which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited