# High Sensitivity Photonic Crystal Fiber Refractive Index Sensor with Gold Coated Externally Based on Surface Plasmon Resonance

^{1}

^{2}

^{*}

## Abstract

**:**

^{−1}. Due to its high sensitivity, the proposed sensor can be used for practical biological and chemical sensing.

## 1. Introduction

^{−1}at the refractive index range of 1.390~1.395.

## 2. Structural and Theoretical Modeling

_{0}= 0.2 μm. The air holes in the first ring have a radius of r

_{1}= 0.4 μm, and they are arranged in a clockwise rotation of 60°. The distance between the center of the air holes and the center of the fiber is d = 2 μm. In the second ring, the air holes have a radius of r

_{2}= 0.6 μm and they rotate clockwise at 30°. The distance between the first layer and the second is $\mathsf{\Lambda}$ = 1.2 μm, and two holes are missing in the opposite vertices of the second ring. The thickness of the gold layer is t

_{g}= 35 nm. Outside the gold layer is the analytical layer.

_{1}= 0.6961663, B

_{2}= 0.4079426, B

_{3}= 0.8974764, ${\lambda}_{1}$ = 0.0684043 μm, ${\lambda}_{2}$ = 0.1162414 μm, ${\lambda}_{3}$ = 9.896161 μm.

_{g}. The dielectric constant of gold can be expressed as Drude-Lorentz [19], which is expressed as follows:

## 3. Simulation Results and Analysis

_{g}) on the confinement loss of the core-guided mode. It can be seen from the figure that when t

_{g}increases from 30 to 50 nm, the resonance intensity decreases and the resonance wavelength red-shifts. This is because the thicker the t

_{g}, the higher the damping loss. As a result, the core-guided mode has less energy penetration into the gold film and the coupling effect becomes weaker.

_{0}) on the confinement loss of the core-guided mode. It can be seen from the diagram that, when r

_{0}changes from zero to 0.25 μm, the resonance intensity increases, and the resonance peak becomes sharper and the resonance wavelength shifts red. This is because the change of r

_{0}affects the refractive index of the core-guided mode. Thus, the phase matching between SPP mode and the core-guided mode is affected, which in turn leads to a change in the resonance intensity, resonance peak, and the resonance wavelength. Therefore, we can change the size of r

_{0}to optimize the sensor performance.

_{1}) on the confinement loss of the core-guided mode. It can be seen that when r

_{1}increases from 0.34 to 0.42 μm, the resonance intensity increases slowly and the resonance wavelength shifts red. This is because the change of r

_{1}affects the refractive index of the cladding region, leading to the change of the phase matching condition. Figure 6 displays the influence of the size of the-second-layer air holes (r

_{2}) on the confinement loss of the core-guided mode. It can be seen that with the increase of r

_{2}(from 0.57 to 0.69 μm), the resonance intensity decreases, the resonance peak becomes sharper, and the resonance wavelength shifts blue.

^{−5}RIU.

_{a}, $\partial \alpha (\lambda ,{n}_{a})$ is the difference of confinement loss due to two adjacent refractive indexes of two analytes. When the refractive index of the analyte is 1.390, the maximum amplitude sensitivity of the sensor can reach 641 RIU

^{−1}at the wavelength of 0.893 μm.

## 4. Conclusions

_{0}= 0.2 μm, r

_{1}= 0.4 μm, r

_{2}= 0.6 μm, $\mathsf{\Lambda}$ = 1.2 μm, d = 2 μm, and t

_{g}= 35 nm. Additionally, the maximum wavelength sensitivity of the sensor can reach as high as 11,000 nm/RIU and the amplitude sensitivity 641 RIU

^{−1}at the refractive index range of 1.390~1.395. The proposed sensor is simple, reliable, easy to fabricate, and highly applicable, and it can be used for practical biological and chemical sensing.

## Author Contributions

## Funding

## Conflicts of Interest

## References

- Nguyen, H.; Park, J.; Kang, S.; Kim, M. Surface plasmon resonance: A versatile technique for biosensor applications. Sensors
**2015**, 15, 10481–10510. [Google Scholar] [CrossRef] [PubMed] - Hasan, M.R.; Akter, S.; Rifat, A.A.; Rana, S.; Ali, S. A Highly Sensitive Gold-Coated Photonic Crystal Fiber Biosensor Based on Surface Plasmon Resonance. Photonics
**2017**, 4, 18. [Google Scholar] [CrossRef] - Wang, F.; Sun, Z.; Liu, C.; Sun, T.; Chu, P.K. A Highly Sensitive Dual-Core Photonic Crystal Fiber Based on a Surface Plasmon Resonance Biosensor with Silver-Graphene Layer. Plasmonics
**2016**, 12, 1847–1853. [Google Scholar] [CrossRef] - Zanishevskaja, A.A.; Tuchin, V.V.; Skibina, Y.S. Photonic crystal fibers for food quality analysis. In Proceedings of the Biophotonics: Photonic Solutions for Better Health Care III, Brussels, Belgium, 16–19 April 2012; SPIE: Bellingham, WA, USA, 2012; Volume 8427, p. 87. [Google Scholar]
- Cubillas, A.; Schmidt, M.; Scharrer, M.; Euser, T.G.; Etzold, B.; Taccardi, N.; Wasserscheid, P.; Russell, P.S.J. Monitoring of Catalytic Reactions in Photonic Crystal Fiber. In Proceedings of the 2011 Conference on Lasers and Electro-Optics Europe, Munich, Germany, 22–26 May 2011. [Google Scholar]
- Mohammadi, R.; Pourmahyabadi, M. Design and Analysis of Photonic Crystal Fiber Gas Sensor with High Relative Sensitivity. Available online: http://opsi.ir/browse.php?a_id=869&sid=1&slc_lang=en (accessed on 3 December 2018).
- Dash, J.N.; Jha, R. Graphene based birefringent photonic crystal fiber sensor using surface plasmon resonance. IEEE Photonics Technol. Lett.
**2014**, 26, 1092–1095. [Google Scholar] [CrossRef] - Saitoh, K.; Koshiba, M. Numerical modeling of photonic crystal fibers. J. Lightw. Technol.
**2005**, 23, 3580–3590. [Google Scholar] [CrossRef] [Green Version] - Yoav, S.; Dror, M.; Zeev, Z. Prospects for diode-pumped alkali-atom-based hollow-core photonic-crystal fiber lasers. Opt. Lett.
**2014**, 39, 4655–4658. [Google Scholar] - Dadabayev, R.; Shabairou, N.; Zalevsky, Z.; Malka, D. A visible light RGB wavelength demultiplexer based on silicon-nitride multicore PCF. Opt. Laser Technol.
**2019**, 111, 411–416. [Google Scholar] [CrossRef] - Malka, D.; Katz, G. An Eight-Channel C-Band Demux Based on Multicore Photonic Crystal Fiber. Nanomaterials
**2018**, 8, 845. [Google Scholar] [CrossRef] [PubMed] - Hautakorpi, M.; Mattinen, M.; Ludvigsen, H. Surface-plasmon-resonance sensor based on three-hole microstructured optical fiber. Opt. Express
**2008**, 16, 8427–8432. [Google Scholar] [CrossRef] [PubMed] - Rifat, A.; Mahdiraji, G.; Chow, D.; Shee, Y.; Ahmed, R.; Adikan, F. Photonic crystal fiber-based surface plasmon resonance sensor with selective analyte channels and graphene-silver deposited core. Sensors
**2015**, 15, 11499–11510. [Google Scholar] [CrossRef] - Dash, J.N.; Jha, R. SPR Biosensor Based on Polymer PCF Coated with Conducting Metal Oxide. IEEE Photonics Technol. Lett.
**2014**, 26, 595–598. [Google Scholar] [CrossRef] - Liu, B.; Lu, Y.; Yang, X.; Yao, J. Surface plasmon resonance sensor based on photonic crystal fiber filled with core-shell Ag-Au nanocomposite materials. Opt. Eng.
**2016**, 55, 117104. [Google Scholar] [CrossRef] - Zhou, C. Theoretical analysis of double-microfluidic-channels photonic crystal fiber sensor based on silver nanowires. Opt. Commun.
**2013**, 288, 42–46. [Google Scholar] [CrossRef] - Liu, B.; Lu, Y.; Yang, X.; Yao, J. Tunable Surface Plasmon Resonance Sensor Based on Photonic Crystal Fiber Filled with Gold Nanoshells. Plasmonics
**2018**, 13, 763–770. [Google Scholar] [CrossRef] - Malitson, I. Interspecimen comparison of the refractive index of fused silica. JOSA
**1965**, 55, 1205–1209. [Google Scholar] [CrossRef] - Liu, C.; Yang, L.; Lu, X.; Liu, Q.; Wang, F.; Lv, J.; Sun, T.; Mu, H.; Chu, P. Mid-infrared surface plasmon resonance sensor based on photonic crystal fibers. Opt. Express
**2017**, 25, 14227–14237. [Google Scholar] [CrossRef] [PubMed] - Rifat, A.A.; Mahdiraji, G.A.; Sua, Y.M.; Ahmed, R.; Shee, Y.G.; Adikan, F.M. Highly sensitive multi-core flat fiber surface plasmon resonance refractive index sensor. Opt. Express
**2016**, 24, 2485–2495. [Google Scholar] [CrossRef] - Yang, X.; Lu, Y.; Liu, B.; Yao, J. Analysis of Graphene-Based Photonic Crystal Fiber Sensor Using Birefringence and Surface Plasmon Resonance. Plasmonics
**2016**, 12, 489–496. [Google Scholar] [CrossRef] - Momota, M.R.; Hasan, M.R. Hollow-core silver coated photonic crystal fiber plasmonic sensor. Opt. Mater.
**2018**, 76, 287–294. [Google Scholar] [CrossRef] - Rifat, A.A.; Mahdiraji, G.A.; Sua, Y.M.; Shee, Y.G.; Ahmed, R.; Chow, D.M.; Adikan, F.M. Surface Plasmon Resonance Photonic Crystal Fiber Biosensor: A Practical Sensing Approach. IEEE Photonics Technol. Lett.
**2015**, 27, 1628–1631. [Google Scholar] [CrossRef] - Chakma, S.; Khalek, M.A.; Paul, B.K.; Ahmed, K.; Hasan, M.R.; Bahar, A.N. Gold-coated photonic crystal fiber biosensor based on surface plasmon resonance: Design and analysis. Sens. Bio-Sens. Res.
**2018**, 18, 7–12. [Google Scholar] [CrossRef]

**Figure 2.**(

**a**) Dispersion relation between surface plasmon polaritons (SPP) mode and core-guided mode; optical field distribution of (

**b**) the core-guided mode, (

**c**) the SPP mode, and (

**d**) Optical field distribution in phase matching. $\left({n}_{a}=1.36,{r}_{0}=0.2\mathsf{\mu}\mathrm{m},{r}_{1}=0.4\mathsf{\mu}\mathrm{m},{r}_{2}=0.6\mathsf{\mu}\mathrm{m},\mathsf{\Lambda}=1.2\mathsf{\mu}\mathrm{m},d=2\mathsf{\mu}\mathrm{m},{t}_{g}=35\mathrm{nm}\right)$.

**Figure 3.**The confinement loss of the core-guided mode for different thickness of the gold layer. $\left({n}_{a}=1.36,{r}_{0}=0.2\mathsf{\mu}\mathrm{m},{r}_{1}=0.4\mathsf{\mu}\mathrm{m},{r}_{2}=0.6\mathsf{\mu}\mathrm{m},\mathsf{\Lambda}=1.2\mathsf{\mu}\mathrm{m},d=2\mathsf{\mu}\mathrm{m}\right)$.

**Figure 4.**The confinement loss of the core-guided mode for different size of the central air hole. $\left({n}_{a}=1.36,{r}_{1}=0.4\mathsf{\mu}\mathrm{m},{r}_{2}=0.6\mathsf{\mu}\mathrm{m},\mathsf{\Lambda}=1.2\mathsf{\mu}\mathrm{m},d=2\mathsf{\mu}\mathrm{m},{t}_{g}=35\mathrm{nm}\right)$.

**Figure 5.**The confinement loss of the core-guided mode for different size of the-first-layer air holes. $\left({n}_{a}=1.36,{r}_{0}=0.2\mathsf{\mu}\mathrm{m},{r}_{2}=0.6\mathsf{\mu}\mathrm{m},\mathsf{\Lambda}=1.2\mathsf{\mu}\mathrm{m},d=2\mathsf{\mu}\mathrm{m},{t}_{g}=35\mathrm{nm}\right)$.

**Figure 6.**The confinement loss of the core-guided mode for different size of the-second-layer air holes. $\left({n}_{a}=1.36,{r}_{0}=0.2\mathsf{\mu}\mathrm{m},{r}_{1}=0.4\mathsf{\mu}\mathrm{m},\mathsf{\Lambda}=1.2\mathsf{\mu}\mathrm{m},d=2\mathsf{\mu}\mathrm{m},{t}_{g}=35\mathrm{nm}\right)$.

**Figure 7.**Loss spectral when increasing analyte RI from 1.350 to 1.395. $\left({r}_{0}=0.2\mathsf{\mu}\mathrm{m},{r}_{1}=0.4\mathsf{\mu}\mathrm{m},{r}_{2}=0.6\mathsf{\mu}\mathrm{m},\mathsf{\Lambda}=1.2\mathsf{\mu}\mathrm{m},d=2\mathsf{\mu}\mathrm{m},{t}_{g}=35\mathrm{nm}\right)$.

**Figure 8.**The resonance peak when increasing analyte Re from 1.350 to 1.395. $\left({r}_{0}=0.2\mathsf{\mu}\mathrm{m},{r}_{1}=0.4\mathsf{\mu}\mathrm{m},{r}_{2}=0.6\mathsf{\mu}\mathrm{m},\mathsf{\Lambda}=1.2\mathsf{\mu}\mathrm{m},d=2\mathsf{\mu}\mathrm{m},{t}_{g}=35\mathrm{nm}\right)$.

**Figure 9.**Amplitude sensitivity when increasing analyte RI from 1.350 to 1.395. $\left({r}_{0}=0.2\mathsf{\mu}\mathrm{m},{r}_{1}=0.4\mathsf{\mu}\mathrm{m},{r}_{2}=0.6\mathsf{\mu}\mathrm{m},\mathsf{\Lambda}=1.2\mathsf{\mu}\mathrm{m},d=2\mathsf{\mu}\mathrm{m},{t}_{g}=35\mathrm{nm}\right)$.

The Structure of PCF | Detection RI Range | Maximum Sensitivity |
---|---|---|

Double core structure [2] | 1.35–1.36 | 2200 nm/RIU |

Graphene-Based structure [21] | 1.345–1.350 | 3400 nm/RIU |

Hollow-core silver coated structure [22] | 1.36–1.37 | 4200 nm/RIU |

Double core structure [23] | 1.33–1.34 | 4000 nm/RIU |

Double core structure [24] | 1.36–1.37 | 9000 nm/RIU |

Structure with Gold Coated Externally (our work) | 1.390–1.395 | 11,000 nm/RIU |

© 2018 by the authors. Licensee MDPI, Basel, Switzerland. This article is an open access article distributed under the terms and conditions of the Creative Commons Attribution (CC BY) license (http://creativecommons.org/licenses/by/4.0/).

## Share and Cite

**MDPI and ACS Style**

Li, X.; Li, S.; Yan, X.; Sun, D.; Liu, Z.; Cheng, T.
High Sensitivity Photonic Crystal Fiber Refractive Index Sensor with Gold Coated Externally Based on Surface Plasmon Resonance. *Micromachines* **2018**, *9*, 640.
https://doi.org/10.3390/mi9120640

**AMA Style**

Li X, Li S, Yan X, Sun D, Liu Z, Cheng T.
High Sensitivity Photonic Crystal Fiber Refractive Index Sensor with Gold Coated Externally Based on Surface Plasmon Resonance. *Micromachines*. 2018; 9(12):640.
https://doi.org/10.3390/mi9120640

**Chicago/Turabian Style**

Li, Xudong, Shuguang Li, Xin Yan, Dongming Sun, Zheng Liu, and Tonglei Cheng.
2018. "High Sensitivity Photonic Crystal Fiber Refractive Index Sensor with Gold Coated Externally Based on Surface Plasmon Resonance" *Micromachines* 9, no. 12: 640.
https://doi.org/10.3390/mi9120640