# Towards Measuring Terahertz Photon Statistics by a Superconducting Bolometer

^{*}

## Abstract

**:**

_{3}crystal placed to a cooled cryostat together with the bolometer NbN film. Results of theoretical approximation of experimental histograms reveal the discrete nature of THz detection by superconducting bolometers and open a way for studying their quantum characteristics. It is shown that bolometer readings per pulse consist of discrete counts (“single charges”), with the mean number linearly dependent on the number of input photons. Contributions of single counts to a total analog reading are statistically distributed according to the normal law, with average values slightly depending on the number of counts in each reading. A general formula is proposed to describe the relationship between continuous statistical distribution of the bolometer readings and discrete quantum statistics of the incident photons.

## 1. Introduction

## 2. Materials and Methods

^{3+}:YLF solid-state laser, generated in KTP crystal (2ω), was used to pump Mg:LiNbO

_{3}crystal with 7.1 mol.% Mg content (LN). The pump wavelength was 523.35 nm, the pump pulse duration and repetition frequency were 10 ns and 7 kHz, correspondingly, and the pump beam waist at the crystal was ~100 μm. The crystal was mounted in He cryostat together with superconducting HEB manufactured by company SCONTEL, Russia [32]. Temperatures of the crystal and the bolometer sensitive NbN element were kept at the same value: 4.8 K. According to data provided by manufacturers, the characteristic response time and noise equivalent power of the bolometer were 50 ps and $2.5\text{}\times {\text{}10}^{-13}\mathrm{W}\cdot {\mathrm{Hz}}^{-1/2}$, respectively. A special planar logarithmic spiral antenna [33] was optimized for the spectral range 0.1–6 THz. The cryostat windows of 20 mm diameter, designed to enter the pump radiation into the cooled chamber and to extract it out after passing the crystal, were equipped with ITO filters [34]. The bolometer was inside three shielding shells of the cryostat. Glass windows were installed on an external aluminum shell that was kept at room temperature. The next shell from a stainless steel was provided by open windows. Finally, a third brass shell was in contact with the cooled board. The ITO filters were installed into the windows in the third shell, and the bolometer sensitive element was attached to the board. All cryostat compartments were pumped out to a pressure of 10

^{−6}mbar. By this arrangement, the temperature of both ITO filters practically coincided with the temperature of the bolometer, and these cooled filters prevented outside thermal radiation at THz frequencies from entering the cryostat [5]. At the same time, not only direct pumping beam, but also optical radiation of Stokes and anti-Stokes frequencies, generated in LN at small angles, could leave the cryostat without any significant absorption [11]. In addition, IR filter (ZitexRG-106) protected the bolometer from pump radiation elastically scattered within the crystal.

_{3}crystal was excited in type-0 geometry. According to phase-matching conditions, the photons of ~1 THz frequency were most effectively generated in this sample at polar angles ${\theta}_{i0}\approx {60}^{\circ}$ with respect to pump direction [4,10]. THz part of PDC radiation was extracted from the crystal through its side face in immediate proximity to the pump beam [5]. A high-resistivity silicon 45° prism was used in order to avoid total internal reflection of THz waves at the output crystal surface. After passing the prism, a part of PDC radiation was selected by a band pass filter (Tydex BPF1.0-24/31) and focused by 5 mm Si hemispherical lens to the bolometer input window. The total spectrum of terahertz photons which can be generated under PDC in the sideways geometry with Mg:LiNbO

_{3}crystal occupies the range from 0.1 THz up to 7.2 THz [35]. A narrow band pass filter with a central frequency of 1 THz and a half-width of 0.15 THz transmitted a part of this wide spectrum. The energy bandgap in NbN is 4.5 meV [36], which corresponds to a frequency of 1.15 THz. However, as it was shown experimentally in [33], the spectral responsivity of our bolometer is the highest in the range 1–2.5 THz. So, the spectral range of THz photons incident on the bolometer in our set-up hits the range of its maximal responsivity. Electric signal from HEB detector was amplified by a broadband cryogenic amplifier associated with the bolometer and then directed to time-gated Boxcar integrator SRS 250. The integrator was gated by pulses from a pin-diode (PD) placed in the lateral pump beam outlet, the integration strobe window was 6 ns.

## 3. Results and Discussion

_{33}is the crystal nonlinear coefficient, $L$ is its length, $\Delta {\omega}_{i}$, $\Delta {\Omega}_{i}$, and $S$ are the spectral band, a solid angle, and an input area of the detection system, correspondingly. Planck’s factor $\langle {N}_{T}\rangle ={\left[\mathrm{exp}(\hslash {\omega}_{i}/{k}_{B}T)-1\right]}^{-1}$ accounts thermal field fluctuations at the crystal temperature $T$. Frequencies of quantum-correlated THz (${\omega}_{i}$) and optical (${\omega}_{s}$) photons are related with the pump frequency ${\omega}_{p}$ via the energy conversation law, $\hslash {\omega}_{s}+\hslash {\omega}_{i}=\hslash {\omega}_{p}$; ${n}_{p},{n}_{s},{n}_{i}$ are the crystal refractive indexes at corresponding frequencies, ${\mathsf{\theta}}_{s0},{\mathsf{\theta}}_{i0}$ are the propagation angles of optical and THz photons generated under exact phase-matching condition. Evidently, the mean THz wave power and, correspondingly, the averaged HEB readings should demonstrate the same dependence on pump power as is predicted by Equation (1). It was shown in our previous study of PDC process for the same LN crystal sample [38] that the numerical relation

## 4. Conclusions

_{3}crystal pumped by 10 ns 523 nm laser pulses. Together with the input elements and NbN film of each bolometer, the crystal was placed in a cooled He cryostat closed from thermal radiation from other external sources. It was shown that statistical distributions of bolometer readings per laser pulse can be approximated with high accuracy under the following assumptions:

- (1)
- any reading consists of a number of discrete single counts with Poisson statistical distribution,
- (2)
- contributions of individual counts (“single charges”) to each total reading are statistically distributed according to the normal law,
- (3)
- the average values of contributions (“mean single charges”) depend on the number of counts in each reading.

## Author Contributions

## Funding

## Institutional Review Board Statement

## Informed Consent Statement

## Data Availability Statement

## Conflicts of Interest

## References

- Kitaeva, G.K.; Yakunin, P.V.; Kornienko, V.V.; Penin, A.N. Absolute brightness measurements in the terahertz frequency range using vacuum and thermal fluctuations as references. Appl. Phys. B
**2014**, 116, 929–937. [Google Scholar] [CrossRef] - Riek, C.; Seletskiy, D.V.; Moskalenko, A.S.; Schmidt, J.F.; Krauspe, P.; Eckart, S.; Eggert, S.; Burkard, G.; Leitenstorfer, A. Direct sampling of electric-field vacuum fluctuations. Science
**2015**, 350, 420–423. [Google Scholar] [CrossRef][Green Version] - Kornienko, V.V.; Kitaeva, G.K.; Sedlmeir, F.; Leuchs, G.; Schwefel, H.G.L. Towards terahertz detection and calibration through spontaneous parametric down-conversion in the terahertz idler-frequency range generated by a 795 nm diode laser system. APL Photonics
**2018**, 3, 051704. [Google Scholar] [CrossRef] - Kitaeva, G.K.; Kornienko, V.V.; Leontyev, A.A.; Shepelev, A.V. Generation of optical signal and terahertz idler photons by spontaneous parametric down-conversion. Phys. Rev. A
**2018**, 98, 063844. [Google Scholar] [CrossRef] - Kitaeva, G.K.; Kornienko, V.V.; Kuznetsov, K.A.; Pentin, I.V.; Smirnov, K.V.; Vakhtomin, Y.B. Direct detection of the idler THz radiation generated by spontaneous parametric down-conversion. Opt. Lett.
**2019**, 44, 1198–1201. [Google Scholar] [CrossRef] - Haase, B.; Kutas, M.; Riexinger, F.; Bickert, P.; Keil, A.; Molter, D.; Bortz, M.; von Freymann, G. Spontaneous parametric down-conversion of photons at 660 nm to the terahertz and sub-terahertz frequency range. Opt. Express
**2019**, 27, 7458. [Google Scholar] [CrossRef] [PubMed] - Benea-Chelmus, I.-C.; Settembrini, F.F.; Scalari, G.; Faist, J. Electric field correlation measurements on the electromagnetic vacuum state. Nature
**2019**, 568, 202–206. [Google Scholar] [CrossRef] - Kuznetsov, K.A.; Malkova, E.I.; Zakharov, R.V.; Tikhonova, O.V.; Kitaeva, G.K. Nonlinear interference in strongly non-degenerate regime and Schmidt mode analysis. Phys. Rev. A
**2020**, 101, 053843. [Google Scholar] [CrossRef] - Kutas, M.; Haase, B.; Bickert, P.; Riexinger, F.; Molter, D.; von Freymann, G. Terahertz quantum sensing. Sci. Adv.
**2020**, 6, eaaz8065. [Google Scholar] [CrossRef][Green Version] - Kitaeva, G.K.; Leontyev, A.A.; Prudkovskii, P.A. Quantum correlation between optical and terahertz photons generated under multimode spontaneous parametric down-conversion. Phys. Rev. A
**2020**, 101, 053810. [Google Scholar] [CrossRef] - Novikova, T.I.; Kuznetsov, K.A.; Leontyev, A.A.; Kitaeva, G.K. Study of SPDC spectra to reveal temperature dependences for optical-terahertz biphotons. Appl. Phys. Lett.
**2020**, 116, 264003. [Google Scholar] [CrossRef] - Lemos, G.B.; Borish, V.; Cole, G.D.; Ramelow, S.; Lapkiewicz, R.; Zeilinger, A. Quantum imaging with undetected photons. Nature
**2014**, 512, 409. [Google Scholar] [CrossRef][Green Version] - Nomerotski, A.; Keach, M.; Stankus, P.; Svihra, P.; Vintskevich, S. Counting of Hong-Ou-Mandel bunched optical photons using a fast pixel camera. Sensors
**2020**, 20, 3475. [Google Scholar] [CrossRef] - Clark, A.S.; Chekhova, M.; Matthews, J.C.F.; Rarity, J.G.; Oulton, R.F. Special Topic: Quantum sensing with correlated light sources. Appl. Phys. Lett.
**2021**, 118, 060401. [Google Scholar] [CrossRef] - Moreau, P.-A.; Toninelli, E.; Gregory, T.; Padgett, M.J. Ghost Imaging Using Optical Correlations. Laser Photonics Rev.
**2018**, 12, 1700143. [Google Scholar] [CrossRef][Green Version] - Yang, J.-Z.; Li, M.-F.; Chen, X.-X.; Yu, W.-K.; Zhang, A.-N. Single-photon quantum imaging via single-photon illumination. Appl. Phys. Lett.
**2020**, 117, 214001. [Google Scholar] [CrossRef] - Kitaeva, G.K.; Penin, A.N. Spontaneous parametric down-conversion. JETP Lett.
**2005**, 82, 350–355. [Google Scholar] [CrossRef] - Kalashnikov, D.A.; Paterova, A.V.; Kulik, S.P.; Krivitsky, L.A. Infrared spectroscopy with visible light. Nat. Photonics
**2016**, 10, 98. [Google Scholar] [CrossRef][Green Version] - Polyakov, S.V.; Migdall, A.L. High accuracy verificatio;n of a correlated photon-based method for determining photoncounting detection efficiency. Opt. Express
**2007**, 15, 1390–1407. [Google Scholar] [CrossRef][Green Version] - Avella, A.; Ruo-Berchera, I.; Degiovanni, I.P.; Brida, G.; Genovese, M. Absolute calibration of an EMCCD camera by quantum correlation, linking photon counting to the analog regime. Opt. Lett.
**2016**, 41, 1841–1844. [Google Scholar] [CrossRef][Green Version] - Kuznetsov, K.A.; Kitaeva, G.K.; Kovalev, S.P.; Germansky, S.A.; Buryakov, A.M.; Tuchak, A.N.; Penin, A.N. Complex extraordinary dielectric function of Mg-doped lithium niobate crystals at terahertz frequencies . Appl. Phys. B
**2016**, 122, 223. [Google Scholar] [CrossRef] - Karasik, B.S.; Sergeev, A.V. THz hot-electron photon counter. IEEE Trans. Appl. Supercond.
**2005**, 15, 618–621. [Google Scholar] [CrossRef] - McKitterick, C.B.; Vora, H.; Du, X.; Karasik, B.S.; Prober, D.E. Graphene microbolometers with superconducting contacts for terahertz photon detection. J. Low Temp. Phys.
**2014**, 176, 291–298. [Google Scholar] [CrossRef][Green Version] - Williams, J.O.D.; Alexander-Webber, J.A.; Lapington, J.S.; Roy, M.; Hutchinson, I.B.; Sagade, A.A.; Martin, M.-B.; Braeuninger-Weimer, P.; Cabrero-Vilatela, A.; Wang, R.; et al. Towards a graphene-based low intensity photon counting photodetector. Sensors
**2016**, 16, 1351. [Google Scholar] [CrossRef] [PubMed][Green Version] - Echternach, P.M.; Pepper, B.J.; Reck, T.; Bradford, C.M. Single photon detection of 1 5 THz radiation with the quantum capacitance detector. Nat. Astron.
**2018**, 2, 90–97. [Google Scholar] [CrossRef] - Jahanmirinejad, S.; Frucci, G.; Mattioli, F.; Sahin, D.; Gaggero, A.; Leoni, R.; Fiore, A. Photon-number resolving detector based on a series array of superconducting nanowires. Appl. Phys. Lett.
**2012**, 101, 072602. [Google Scholar] [CrossRef][Green Version] - Burenkov, I.A.; Sharma, A.K.; Gerrits, T.; Harder, G.; Bartley, T.J.; Silberhorn, C.; Goldschmidt, E.A.; Polyakov, S.V. Full statistical mode reconstruction of a light field via a photon-number-resolved measurement. Phys. Rev. A
**2017**, 95, 053806. [Google Scholar] [CrossRef][Green Version] - Lolli, L.; Taralli, E.; Portesi, C.; Rajteri, M.; Monticone, E. Aluminum–titanium bilayer for near-infrared transition edge sensors. Sensors
**2016**, 16, 953. [Google Scholar] [CrossRef] [PubMed][Green Version] - Santavicca, D.F.; Reulet, B.; Karasik, B.S.; Pereverzev, S.V.; Olaya, D.; Gershenson, M.E.; Frunzio, L.; Prober, D.E. Energy resolution of terahertz single-photon-sensitive bolometric detectors. Appl. Phys. Lett.
**2010**, 96, 083505. [Google Scholar] [CrossRef][Green Version] - Shurakov, A.; Lobanov, Y.; Goltsman, G. Superconducting hot-electron bolometer: From the discovery of hot-electron phenomena to practical application. Supercond. Sci. Technol.
**2015**, 29, 023001. [Google Scholar] [CrossRef] - Pentin, I.; Vakhtomin, Y.; Seleznev, V.; Smirnov, K. Hot electron energy relaxation time in vanadium nitride superconducting film structures under THz and IR radiation. Sci. Rep.
**2020**, 10, 16819. [Google Scholar] [CrossRef] - Seliverstov, S.; Maslennikov, S.; Ryabchun, S.; Finkel, M.; Klapwijk, T.; Kaurova, N.; Vakhtomin, Y.; Smirnov, K.; Voronov, B.; Goltsman, G. Fast and sensitive terahertz direct detector based on superconducting antenna-coupled hot electron bolometer. IEEE Trans. Appl. Supercond.
**2015**, 25, 2300304. [Google Scholar] [CrossRef] - Shurakov, A.; Seliverstov, S.; Kaurova, N.; Finkel, M.; Voronov, B.; Goltsman, G. Input bandwidth of hot electron bolometer with spiral antenna. IEEE Trans. Terahertz Sci. Technol.
**2012**, 2, 400. [Google Scholar] [CrossRef] - Wang, T.; Zalkovskij, M.; Iwaszczuk, K.; Lavrinenko, A.V.; Naik, G.V.; Kim, J.; Boltasseva, A.; Jepsen, P.U. Ultrabroadband terahertz conductivity of highly doped ZnO and ITO. Opt. Mat. Express
**2015**, 5, 566. [Google Scholar] [CrossRef] - Suizu, K.; Koketsu, K.; Shibuya, T.; Tsutsui, T.; Akiba, T.; Kawase, K. Extremely frequency-widened terahertz wave generation using Cherenkov-type radiation. Opt. Express
**2009**, 17, 6676. [Google Scholar] [CrossRef] - Komenou, K.; Yamashita, T.; Onodera, Y. Energy gap measurement of niobium nitride. Phys. Lett. A
**1968**, 28, 335. [Google Scholar] [CrossRef] - Klyshko, D.N. Photons and Nonlinear Optics; Gordon and Breach: New York, NY, USA, 1988. [Google Scholar]
- Sultanov, V.D.; Kuznetsov, K.A.; Leontyev, A.A.; Kitaeva, G.K. Generation of optical–terahertz biphotons and detection of their terahertz component under frequency-nondegenerate parametric down-conversion. JETP Lett.
**2020**, 112, 269–273. [Google Scholar] [CrossRef] - Barends, R.; Hajenius, M.; Gao, J.R.; Klapwijk, T.M. Current-induced vortex unbinding in bolometer mixers. Appl. Phys. Lett.
**2005**, 87, 263506. [Google Scholar] [CrossRef] - Miao, W.; Delorme, Y.; Feret, A.; Lefevre, R.; Lecomte, B.; Dauplay, F.; Krieg, J.-M.; Beaudin, G.; Zhang, W.; Ren, Y.; et al. Comparison between hot spot modeling and measurement of a superconducting hot electron bolometer mixer at submillimeter wavelengths. J. Appl. Phys.
**2009**, 106, 10390. [Google Scholar] [CrossRef] - Miao, W.; Zhang, W.; Zhong, J.Q.; Shi, S.C.; Delorme, Y.; Lefevre, R.; Feret, A.; Vacelet, T. Non-uniform absorption of terahertz radiation on superconducting hot electron bolometer microbridges. Appl. Phys. Lett.
**2014**, 104, 052605. [Google Scholar] [CrossRef] - Scully, M.O.; Lamb, V.E. Quantum Theory of an Optical Maser. III. Theory of Photoelectron Counting Statistics. Phys. Rev.
**1969**, 179, 368–373. [Google Scholar] [CrossRef] - Tatarskii, V.I. Corrections to the theory of photocounting. Phys. Lett. A
**1990**, 144, 491. [Google Scholar] [CrossRef]

**Figure 1.**Schematics of the experimental set-up. The pumping radiation for PDC is generated using a series of the pulsed laser (Nd

^{3+}:YLF), the frequency doubling crystal (KTP), and the 1st harmonic cut-off filter (HPF). A small part of the pump beam is diverted by a mirror to a PIN diode (PD) which triggers integrator and boxcar averager modules SR250 (Boxcar integrator) at the beginning of each laser pulse. PDC originates in Mg-doped lithium niobate crystal (Mg:LiNbO

_{3}) equipped by a prism from high-resistivity silicon (Si prism). The crystal and a system for detecting idler THz radiation are placed inside a He cryostat. The detection system contains a pump cut-off filter film (Zitex G-106), a narrow THz band pass filter Tydex BPF1.0-24/31 (TF), a focusing Si hemispherical lens (Si lens), and the NbN superconducting bolometer SCONTEL (HEB). The optical photons (signal radiation) are generated at small angles to the pump beam and leave the cryostat via the same windows as the pump radiation passes (marked by the blue color). In each window there is an external glass filter and an internal ITO filter. The latter is kept at the same temperature as the crystal and HEB.

**Figure 2.**Histograms of the HEB detector readings, recorded experimentally under different laser pump powers (circles), and results of their numerical modeling (curves). The time-averaged pump power was 29.6 mW (red circles), 53.4 mW (magenta circles), 76.6 mW (blue circles), and 92.8 mW (green circles). The transport current of HEB was 64 µA (a) and 69 μA (

**b**). Solid curves: numerical approximation by Equation (7). A dashed curve (

**a**): approximation by Equation (6); the inset shows a plot section in an enlarged scale.

**Figure 3.**The mean number of the bolometer photo counts per pulse (in absolute unites, dots), and the theoretical prediction for the mean number of incident photons (in arbitrary units, a solid curve), both versus the time-averaged pump power. Open dots: results of approximation of experimental histograms by Equation (6), filled dots: by Equation (7). Black circular dots correspond to experimental data obtained under the HEB transport current of 64 µA, red triangular dots correspond to the data under the HEB transport current of 69 μA.

**Table 1.**Results of approximation by two probability distributions, ${P}_{c}(z)$ from Equation (6) and $P(z)$ from Equation (7). The uncertainties are the root-mean-square errors obtained under approximating the experimental dependences by the least squares method.

Bolometer Current | Pump Power | Number of Counts <m> | Mean Single Current <e_{1}>/τ, arb.un. | Single-Current Deviation σ, arb.un. | |||
---|---|---|---|---|---|---|---|

Equation (6) | Equation (7) | Equation (6) | Equation (7) | Equation (6) | Equation (7) | ||

64 μA | 29.6 mW | 7.04 ± 0.14 | 4.58 ± 0.12 | 22.9 ± 0.5 | 46.4 ± 1.9 | 44.9 ± 0.2 | 56.6 ± 1.0 |

53.4 mW | 8.95 ± 0.7 | 9.22 ± 0.06 | 43.8 ± 3.6 | 35.1 ± 0.4 | 37.7 ± 2.6 | 16.4 ± 0.5 | |

76.6 mW | 11.9 ± 5.7 | 13.82 ± 0.03 | 45 ± 23 | 29.7 ± 0.15 | 35 ± 20 | 11.3 ± 0.2 | |

69 μA | 53.4 mW | 8.9 ± 1.2 | 9.07 ± 0.06 | 80 ± 12 | 66.5 ± 0.5 | 66.8 ± 9.2 | 30.3 ± 1.2 |

76.6 mW | 12.4 ± 8.8 | 14.3 ± 0.2 | 71 ± 53 | 48.3 ± 1.0 | 54.7 ± 48.3 | 19.7 ± 1.6 | |

92.8 mW | 17.91 ± 0.03 | 16.1 ± 0.8 | 87.9 ± 0.2 | 156 ± 4 | 19.7 ± 0.6 | 115 ± 4 |

Publisher’s Note: MDPI stays neutral with regard to jurisdictional claims in published maps and institutional affiliations. |

© 2021 by the authors. Licensee MDPI, Basel, Switzerland. This article is an open access article distributed under the terms and conditions of the Creative Commons Attribution (CC BY) license (https://creativecommons.org/licenses/by/4.0/).

## Share and Cite

**MDPI and ACS Style**

Prudkovskii, P.; Leontyev, A.; Kuznetsov, K.; Kitaeva, G.
Towards Measuring Terahertz Photon Statistics by a Superconducting Bolometer. *Sensors* **2021**, *21*, 4964.
https://doi.org/10.3390/s21154964

**AMA Style**

Prudkovskii P, Leontyev A, Kuznetsov K, Kitaeva G.
Towards Measuring Terahertz Photon Statistics by a Superconducting Bolometer. *Sensors*. 2021; 21(15):4964.
https://doi.org/10.3390/s21154964

**Chicago/Turabian Style**

Prudkovskii, Pavel, Andrey Leontyev, Kirill Kuznetsov, and Galiya Kitaeva.
2021. "Towards Measuring Terahertz Photon Statistics by a Superconducting Bolometer" *Sensors* 21, no. 15: 4964.
https://doi.org/10.3390/s21154964