Design, Construction, and Application of an Inexpensive, High-Resolution Water Sampler
AbstractThe cost of high-resolution water sampling devices for ecological studies and water quality analyses can be prohibitive. Moreover, the potential for operator error in the use of complicated sampling equipment can lead to inaccuracies. Here we describe the construction and operation of an inexpensive and easy-to-use water sampler that achieves a water column sampling resolution of approximately 1 cm. The device is driven by a peristaltic pump and is constructed entirely of non-corrosive and non-reactive materials. The sampler has no moving parts and was completely reliable in fieldwork on temperate and Antarctic lakes. The device is especially suited for the collection of water samples from calm or stagnant surface waters, such as lakes, ponds, reservoirs, and deep swamps or other wetlands. In addition, because its components are unaffected by corrosive salts and sulfides, the device is suitable for sampling calm inlet waters, including shallow bays and estuaries. Because of its low cost, simple construction, compact design, and precision performance, this water sampler is an excellent option for studying and monitoring shallow to moderately deep (<50 m) natural waters. View Full-Text
Scifeed alert for new publicationsNever miss any articles matching your research from any publisher
- Get alerts for new papers matching your research
- Find out the new papers from selected authors
- Updated daily for 49'000+ journals and 6000+ publishers
- Define your Scifeed now
Sattley, W.M.; Burchell, B.M.; Conrad, S.D.; Madigan, M.T. Design, Construction, and Application of an Inexpensive, High-Resolution Water Sampler. Water 2017, 9, 578.
Sattley WM, Burchell BM, Conrad SD, Madigan MT. Design, Construction, and Application of an Inexpensive, High-Resolution Water Sampler. Water. 2017; 9(8):578.Chicago/Turabian Style
Sattley, W. M.; Burchell, Brad M.; Conrad, Stephen D.; Madigan, Michael T. 2017. "Design, Construction, and Application of an Inexpensive, High-Resolution Water Sampler." Water 9, no. 8: 578.
Note that from the first issue of 2016, MDPI journals use article numbers instead of page numbers. See further details here.