- freely available
High Precision UTDR Measurements by Sonic Velocity Compensation with Reference Transducer
AbstractAn ultrasonic sensor design with sonic velocity compensation is developed to improve the accuracy of distance measurement in membrane modules. High accuracy real-time distance measurements are needed in membrane fouling and compaction studies. The benefits of the sonic velocity compensation with a reference transducer are compared to the sonic velocity calculated with the measured temperature and pressure using the model by Belogol’skii, Sekoyan et al. In the experiments the temperature was changed from 25 to 60 °C at pressures of 0.1, 0.3 and 0.5 MPa. The set measurement distance was 17.8 mm. Distance measurements with sonic velocity compensation were over ten times more accurate than the ones calculated based on the model. Using the reference transducer measured sonic velocity, the standard deviations for the distance measurements varied from 0.6 to 2.0 µm, while using the calculated sonic velocity the standard deviations were 21–39 µm. In industrial liquors, not only the temperature and the pressure, which were studied in this paper, but also the properties of the filtered solution, such as solute concentration, density, viscosity, etc., may vary greatly, leading to inaccuracy in the use of the Belogol’skii, Sekoyan et al. model. Therefore, calibration of the sonic velocity with reference transducers is needed for accurate distance measurements.
Share & Cite This Article
Export to BibTeX | EndNote
MDPI and ACS Style
Stade, S.; Kallioinen, M.; Mänttäri, M.; Tuuva, T. High Precision UTDR Measurements by Sonic Velocity Compensation with Reference Transducer. Sensors 2014, 14, 11682-11690.View more citation formats
Stade S, Kallioinen M, Mänttäri M, Tuuva T. High Precision UTDR Measurements by Sonic Velocity Compensation with Reference Transducer. Sensors. 2014; 14(7):11682-11690.Chicago/Turabian Style
Stade, Sam; Kallioinen, Mari; Mänttäri, Mika; Tuuva, Tuure. 2014. "High Precision UTDR Measurements by Sonic Velocity Compensation with Reference Transducer." Sensors 14, no. 7: 11682-11690.