- freely available
Int. J. Mol. Sci. 2013, 14(4), 8603-8610; doi:10.3390/ijms14048603
Published: 22 April 2013
Abstract: Simple immuno-magnetic separation tandem fluorescent probes based on quantum dots-antibody (QDs-Ab) were developed to detect Salmonella with sensitivity of 500 cfu mL−1. With two monoclonal antibodies, which recognize different antigenic determinant on the surface of Salmonella, we prepared antibody-coated magnetic nanoparticles (MNPs) and conjugates of QDs-Ab. The immune-magnetic beads were verified with high enrichment efficiency for Salmonella (90%). A sandwich structure formed if the Salmonella solution was mixed together with immune-beads and QDs-Ab, and the fluorescent single from QDs was related to the amount of Salmonella. A linear response between fluorescence intensity and various concentrations of Salmonella (2.5 × 103 to 1.95 × 108 cfu mL−1) were observed with this proposed method. The total assay time for Salmonella was 30 min, and no cross-reaction to other microbial strains, such as Staphylococcus aureus, Escherichia coli (E. coli) and Escherichia coli O157:H7 (E. coli O157:H7), were found using this detection system. All our results showed that the simple homogeneous immunoassay could be applied in Salmonella screening without time-consuming extra-enrichment of bacteria.
Salmonella is a common food-borne pathogen that causes food contamination, which has resulted in higher economic losses and poses a significant threat to public health [1,2]. Salmonellosis is the most frequently reported disease worldwide, initiating into diarrhea, cramps, vomiting and fever symptoms . These infections cause significant morbidity and mortality that are particularly severe in infants, the elderly and immune-compromised patients [4–6]. Annually, an estimated 33 million people in the world suffer from typhoid fever and 600,000 deaths are reported among them [7,8]. Even in developed countries, such as the United States, the outbreak incidences of Salmonella infections have been increasing in recent decades. The most easily contaminated foods by Salmonella are meat; poultry and egg products . Hence, the development of Salmonella screening method is crucial to safeguard human health and identify their distribution for better management of food safety.
To date, some approaches for Salmonella detection have been developed, including conventional culture methods, immunoassays [10–17] and polymerase chain reaction (PCR) measures [18–22]. However, these measures either require a long-time pre-enrichment step or depend on instruments and professionals in laboratory.
In this paper, we present a novel immuno-sensor using quantum dots (QDs) as fluorescent label and magnetic nanoparticles (MNPs) as enrichment reagent. A pair of monoclonal antibody against Salmonella was used in our detection design. One antibody was conjugated to QDs for detection and the other was bound to MNPs for capturing the targeted bacteria in solution. A fluoresce single was quantitative based on the forming a sandwich structure (QDs-bacteria-MNPs).
2. Materials and Methods
All bacterial strains used in this study were purchased from the China Center of Industrial Culture Collection (Beijing, China), including Salmonella, Staphylococcus aureus, E. coli and E. coli O157:H7. The paired monoclonal antibodies, QDs and MNPs were all prepared in our lab. Two antibodies were obtained through hybridoma selection process. The affinity constants were calculated as 1.8 × 109 and 2.2 × 109, respectively, for these two anti- Salmonella antibody.
Bovine serum albumin (BSA) was obtained from Solarbio Science & Technology, Co, Ltd. (Beijing, China). N-hydroxysulfosuccinimide sodium salt (sulfo-NHS) was purchased from J&K Scientific, Ltd. (Beijing, China). 1-ethyl-3-(3-dimethylaminopropyl)-carbodiimide (EDC) was purchased from Aladdin Reagent (Shanghai, China). Biotin and streptavidin were purchased from Sigma-Aldrich (Shanghai, China). Other chemicals were purchased from the Shanghai Chemical Reagent Company (Shanghai, China). All water in this experiment was ultra-water from Millipore water purification system (18.2 MΩ.cm; EMD Millipore Corporation, Billerica, MA, USA).
2.2. Design of the Sandwich Immunoassay for Salmonella Detection
A schematic of the sensing principle was illustrated in Figure 1. Three critical materials, including MNPs, QDs and two anti-Salmonella antibodies, were used in this detection system. The anti-Salmonella antibodies were immobilized individually on the MNPs and QDs. The two antibodies recognize different antigenic determinants of Salmonella. For assay, the sample solution was sufficiently incubated with immune-beads and QD probes in a tube for a period. Then, the tube was put in a magnetic field for separating the MNPs-bacteria-QDs from solution. The isolated Salmonella coupled with the QDs was quantified using a fluorescence spectrophotometer (F-7000, Hitachi Ltd., Tokyo, Japan). The increasing Salmonella concentrations allow more QDs to be captured by magnetic beads resulting in more intensive fluorescence single.
2.3 Functionalization of MNPs with Antibody
The MNPs used here was modified with carboxyl group on their surface (particle size 100 nm). EDC and sulfo-NHS were applied as linkers in this experiment and they were dissolved to phosphate-buffered saline (PBS, 0.1 M, pH 7.2) with final concentrations of 0.5 and 0.2 mg mL−1, respectively. One milliliter of the linker solution was mixed with equal volume of MNPs and held for 15 min at room temperature, then 100 μL aliquots of anti-Salmonella antibody (Ab-1) was added drop-wise into the solution. The reaction was carried out at 37 °C for 2 h, and then the MNP beads were isolated using magnet. After removal of the liquid in the tube, 1% BSA solution in PBS (g/v, 1 mL) was re-dispersed on MNPs with continuous stirring for 1 h at 37 °C to block the bareness sites on the surface of MNPs. Following another magnetic separation step, the residues were dissolved in 1 mL PBS and then stored at 4 °C for further use. The capture ability of Salmonella from immune-magnetic beads was evaluated using the counted Salmonella (1.16 × 105 cfu mL−1). After half an hour’s incubation, the immune-magnetic beads were recovered using magnetic separation.
The separated MNPs were washed 5 times and re-suspended in nutrient broth. Then, plate counts were performed to evaluate the enrichment efficiency of immune-magnetic beads.
2.3.1. Synthesis of QDs-Ab Conjugates
The anti-Salmonella antibody (Ab-2) and QDs were coupled by a biotin-streptavidin bridge. The conjugation method is the same as our previous reports [23,24]. The prepared QDs-Ab conjugates were dissolved in PBS (200 μL) and stored at 4 °C for the following use.
2.3.2. Salmonella Sample Analysis
The sample solution (1 mL) was added with 50 μL of immune-MNPs beads, and following this, 50 μL of QDs-Ab conjugates were added. The mixture solution was allowed for 30 min incubation forming the sandwich structure of immune-MNPs/Salmonella/QDs-Ab. With a magnetic separation, the residual QDs were removed from the solution. The Salmonella cells bound to the QDs and MNPs were re-suspended in 1 mL PBS for fluorescent measurement. Various concentrations (from 102 to 108 cfu/mL) of Salmonella in PBS buffer were tested with this proposed detection system. The values of the fluorescence intensity were plotted against the Salmonella to produce a linear curve. The specificity of this proposed immunoassay was tested with other bacteria, including Staphylococcus aureus, E. coli and E. coli O157:H7.
3. Results and Discussion
3.1. Characterization of QD-Ab
Both antibody and QD were functioned using biotin, and then two parts were bridged using streptavidin through the specific affinity reaction. The successful preparation of Ab-biotin complex was verified with UV detection. As it was shown in Figure 2A, the characteristic absorption peak of antibody was located at 215 nm and 278 nm, and the peak for biotin was at 260 nm. The product of biotin-Ab exhibited both typical absorption peaks of biotin and antibody, respectively, at 260 nm and 215 nm. The modification of biotin on the QD was assisted with dBSA, which was prepared with NaBH4 reduction beforehand . The UV spectra in Figure 2B confirmed the coupling between dBSA and biotin.
With the connecting agent of streptavidin, the antibody was successfully anchored on the surface of QDs. The original QD showed the typical fluorescence emission peak at 620 nm, while an obvious red shift of the conjugates of QD-dBSA-biotin (QD-dBSA, 625 nm) and QD-Ab (630 nm) were seen in Figure 2C.
3.2. Optimization of the Immunoassay System
With gradient dilutions and then plate counts, the absorbed Salmonella by immune beads was calculated as (1.05 ± 0.12) × 105 out of the total amount of 1.15 × 105 cfu mL−1, resulting in the average 90% binding efficiency.
In evaluation of the optimum incubation period for sample solution, immune-beads and QD probes in the assay buffer, various incubation times (5, 15, 30 and 60 min) were tested. The total amount of Salmonella was 1.16 × 105 cfu for optimization tests, and a blank sample (not containing Salmonella) was used as a control. The photoluminescence (PL) intensity from the formed sandwich products was determined. The results from Figure 2D indicated that the fluorescence intensity was enhanced with increasing incubation time from 5 to 30 min. However, no significant differences were observed if the incubation was raised to 60 min. Hence, the optimal incubation time was 30 min, which reflected the high affinity among immune-beads, QD probes and Salmonella.
3.3. Performance Assessment of Salmonella Detection
The assay sensitivity was evaluated using a series of Salmonella concentrations (500 to 1.95 × 108 cfu mL−1). As shown in Figure 3A, the photoluminescence (PL) intensity rose gradually with the increasing concentration of Salmonella. It demonstrated a good linear relationship when the bacteria concentration ranging from 2.5 × 103 to 1.95 × 108 cfu mL−1 with a correlation coefficient of 0.9936 (Figure 3B). A test sample was considered positive if the ratio (P/N) of the optical density (OD) value in the test well to that of the negative control well was ≥2.1. The sensitivity of this proposed assay was 500 cfu mL−1.
In specificity tests, the control sample (no containing any bacteria) and three pathogens, including Staphylococcus aureus, E. coli and E. coli O157:H7, were detected. All tested bacteria were at the 105 level. As it was shown in Figure 3C, very slight PL intensity (<80 au) could be detected for the control solution and non-targeted bacteria, while a strong fluorescence signal was observed for Salmonella (~700 au). All the results indicated that the immune-MNPs tandem QDs-Ab system was very specific in Salmonella detection.
A sensitive and simple immunoassay was developed for Salmonella here. With the efficient enrichment of immune-MNPs, the sample could be completed in half an hour. Due to the predominant fluorescent character of QDs, quantitative analysis was possible for Salmonella with a linear curve. Without complex pre-culturing procedures, the proposed method could be directly detected samples with the sensitivity of 500 cfu mL−1 for Salmonella. This immunoassay was simple and fast, which makes it a great potential for being applied in surveillance of Salmonella contamination.
This work is financially supported by the National Natural Science Foundation of China (21071066, 91027038, 21101079, 21175034), the Key Programs from MOST (2012BAC01B07, 2012BAD29B05, 2012AA06A303, 2012BAD29B04, 2011BAK10B07, 2011BAK10B05, 2011BAK10B01, 2010AA06Z302, 2010DFB3047, 2011ZX08012-001, 2012BAK17B10, 2012BAK08B01, 2012YQ090194) and grants from the Natural Science Foundation of Jiangsu Province, MOF and MOE (NCET-12-0879, BE2011626, 201210036, 201310135, 311002) and the Foundation for the Author of National Excellent Doctoral Dissertation of China (No. 201187) and the Research Program of the State Key Laboratory of Food Science and Technology, Jiangnan University (SKLF- ZZA-201202).
- Newell, D.G.; Koopmans, M.; Verhoef, L.; Duizer, E.; Aidara-Kane, A.; Sprong, H.; Opsteegh, M.; Langelaar, M.; Threfall, J.; Scheutz, F.; et al. Food-borne diseases-The challenges of 20 years ago still persist while new ones continue to emerge. Int. J. Food Microbiol 2010, 139, S3–S15. [Google Scholar]
- Wong, T.L.; MacDiarmid, S.; Cook, R. Salmonella, Escherichia coli O157:H7 and E. coli biotype 1 in a pilot survey of imported and New Zealand pig meats. Food Microbiol 2009, 26, 177–182. [Google Scholar]
- Switt, A.I.M.; Soyer, Y.; Warnick, L.D.; Wiedmann, M. Emergence, distribution, and molecular and phenotypic characteristics of Salmonella enterica serotype 4,5,12:i:-. Foodborne Pathog. Dis 2009, 6, 407–415. [Google Scholar]
- Abdoel, T.H.; Pastoor, R.; Smits, H.L.; Hatta, M. Laboratory evaluation of a simple and rapid latex agglutination assay for the serodiagnosis of typhoid fever. Trans. R. Soc. Trop. Med. Hyg 2007, 101, 1032–1038. [Google Scholar]
- Fraser, A.; Paul, M.; Goldberg, E.; Acosta, C.J.; Leibovici, L. Typhoid fever vaccines: Systematic review and meta-analysis of randomised controlled trials. Vaccine 2007, 25, 7848–7857. [Google Scholar]
- Tauxe, R.V. Salmonella: A postmodern pathogen. J. Food Prot 1991, 54, 563–568. [Google Scholar]
- Ghenghesh, K.S.; Franka, E.; Tawil, K.; Wasfy, M.O.; Ahmed, S.F.; Rubino, S.; Klena, J.D. Enteric fever in Mediterranean north Africa. J. Infect. Dev. Ctries 2009, 3, 753–761. [Google Scholar]
- Srikantiah, P.; Girgis, F.Y.; Luby, S.P.; Jennings, G.; Wasfy, M.O.; Crump, J.A.; Hoekstra, R.M.; Anwer, M.; Mahoney, F.J. Population-based surveillance of typhoid fever in Egypt. Am. J. Trop. Med. Hyg 2006, 74, 114–119. [Google Scholar]
- Guard-Petter, J. The chicken, the egg and Salmonella enteritidis. Environ. Microbiol 2001, 3, 421–430. [Google Scholar]
- Bautista, D.; Elankumaran, S.; Arking, J.; Heckert, R. Evaluation of an immunochromatography strip assay for the detection of Salmonella sp. from poultry. J. Vet. Diagn. Invest 2002, 14, 427–430. [Google Scholar]
- Gehring, A.G.; Gerald, C.; Mazenko, R.S.; Van Houten, L.J.; Brewster, J.D. Enzyme-linked immunomagnetic electrochemical detection of Salmonella typhimurium. J. Immunol. Methods 1996, 195, 15–25. [Google Scholar]
- Jain, S.; Chattopadhyay, S.; Jackeray, R.; Abid, C.K.V.Z.; Kohli, G.S.; Singh, H. Highly sensitive detection of Salmonella typhi using surface aminated polycarbonate membrane enhanced-ELISA. Biosens. Bioelectron 2012, 31, 37–43. [Google Scholar]
- Jaradat, Z.W.; Bzikot, J.H.; Zawistowski, J.; Bhunia, A.K. Optimization of a rapid dot-blot immunoassay for detection of Salmonella enterica serovar Enteritidis in poultry products and environmental samples. Food Microbiol 2004, 21, 761–769. [Google Scholar]
- Magliulo, M.; Simoni, P.; Guardigli, M.; Michelini, E.; Luciani, M.; Lelli, R.; Roda, A. A rapid multiplexed chemiluminescent immunoassay for the detection of Escherichia coli O157: H7, Yersinia enterocolitica, Salmonella typhimurium, and Listeria monocytogenes pathogen bacteria. J. Agric. Food Chem 2007, 55, 4933–4939. [Google Scholar]
- Nicholas, R.A.; Cullen, G.A. Development and application of an ELISA for detecting antibodies to Salmonella enteritidis in chicken flocks. Vet. Rec 1991, 128, 74–76. [Google Scholar]
- Preechakasedkit, P.; Pinwattana, K.; Dungchai, W.; Siangproh, W.; Chaicumpa, W.; Tongtawe, P.; Chailapakul, O. Development of a one-step immunochromatographic strip test using gold nanoparticles for the rapid detection of Salmonella typhi in human serum. Biosens. Bioelectron 2012, 31, 562–566. [Google Scholar]
- Proux, K.; Houdayer, C.; Humbert, F.; Cariolet, R.; Rose, V.; Eveno, E.; Madec, F. Development of a complete ELISA using Salmonella lipopolysaccharides of various serogroups allowing to detect all infected pigs. Vet. Res 2000, 31, 481–490. [Google Scholar]
- Malorny, B.; Bunge, C.; Helmuth, R. A real-time PCR for the detection of Salmonella Enteritidis in poultry meat and consumption eggs. J. Microbiol. Methods 2007, 70, 245–251. [Google Scholar]
- Miller, N.D.; Davidson, P.M.; D’Souza, D.H. Real-time reverse-transcriptase PCR for Salmonella Typhimurium detection from lettuce and tomatoes. LWT-Food Sci. Technol 2011, 44, 1088–1097. [Google Scholar]
- Nam, H.M.; Srinivasan, V.; Gillespie, B.E.; Murinda, S.E.; Oliver, S.P. Application of SYBR green real-time PCR assay for specific detection of Salmonella spp. in dairy farm environmental samples. Int. J. Food Microbiol 2005, 102, 161–171. [Google Scholar]
- Saroj, S.D.; Shashidhar, R.; Karani, M.; Bandekar, J.R. Rapid, sensitive, and validated method for detection of Salmonella in food by an enrichment broth culture-nested PCR combination assay. Mol. Cell. Probes 2008, 22, 201–206. [Google Scholar]
- Verdoy, D.; Barrenetxea, Z.; Berganzo, J.; Agirregabiria, M.; Ruano-Lopez, J.M.; Marimon, J.M.; Olabarria, C. A novel real time micro PCR based Point-of-Care device for Salmonella detection in human clinical samples. Biosens. Bioelectron 2012, 32, 259–265. [Google Scholar]
- Chen, W.; Xu, D.; Liu, L.; Peng, C.; Zhu, Y.; Ma, W.; Bian, A.; Li, Z.; Yuan, Y.; Jin, Z.; et al. Ultrasensitive detection of trace protein by western blot based on POLY-quantum dot probes. Anal. Chem 2009, 81, 9194–9198. [Google Scholar]
- Peng, C.; Li, Z.; Zhu, Y.; Chen, W.; Yuan, Y.; Liu, L.; Li, Q.; Xu, D.; Qiao, R.; Wang, L.; et al. Simultaneous and sensitive determination of multiplex chemical residues based on multicolor quantum dot probes. Biosens. Bioelectron 2009, 24, 3657–3662. [Google Scholar]
© 2013 by the authors; licensee MDPI, Basel, Switzerland This article is an open access article distributed under the terms and conditions of the Creative Commons Attribution license (http://creativecommons.org/licenses/by/3.0/).