Wrist Pulse Rate Monitor Using Self-Injection-Locked Radar Technology
AbstractTo achieve sensitivity, comfort, and durability in vital sign monitoring, this study explores the use of radar technologies in wearable devices. The study first detected the respiratory rates and heart rates of a subject at a one-meter distance using a self-injection-locked (SIL) radar and a conventional continuous-wave (CW) radar to compare the sensitivity versus power consumption between the two radars. Then, a pulse rate monitor was constructed based on a bistatic SIL radar architecture. This monitor uses an active antenna that is composed of a SIL oscillator (SILO) and a patch antenna. When attached to a band worn on the subject’s wrist, the active antenna can monitor the pulse on the subject’s wrist by modulating the SILO with the associated Doppler signal. Subsequently, the SILO’s output signal is received and demodulated by a remote frequency discriminator to obtain the pulse rate information. View Full-Text
Scifeed alert for new publicationsNever miss any articles matching your research from any publisher
- Get alerts for new papers matching your research
- Find out the new papers from selected authors
- Updated daily for 49'000+ journals and 6000+ publishers
- Define your Scifeed now
Wang, F.-K.; Tang, M.-C.; Su, S.-C.; Horng, T.-S. Wrist Pulse Rate Monitor Using Self-Injection-Locked Radar Technology. Biosensors 2016, 6, 54.
Wang F-K, Tang M-C, Su S-C, Horng T-S. Wrist Pulse Rate Monitor Using Self-Injection-Locked Radar Technology. Biosensors. 2016; 6(4):54.Chicago/Turabian Style
Wang, Fu-Kang; Tang, Mu-Cyun; Su, Sheng-Chao; Horng, Tzyy-Sheng. 2016. "Wrist Pulse Rate Monitor Using Self-Injection-Locked Radar Technology." Biosensors 6, no. 4: 54.