This article is
- freely available
Sensors 2005, 5(4), 209-219; doi:10.3390/s5040209
On Calibration of pH Meters
Department of Chemistry and Department of Physics, University of Missouri-Kansas City, 20045110 Rockhill Road, Kansas City, Missouri 64110 USA
* Author to whom correspondence should be addressed.
Received: 27 May 2004 / Accepted: 14 November 2004 / Published: 27 April 2005
(This article belongs to the Special Issue Papers presented at I3S2004, Nanjing)
Download PDF Full-Text [139 KB, uploaded 20 June 2008 16:51 CEST]
Abstract: The calibration of pH meters including the pH glass electrode, ISE electrodes,buffers, and the general background for calibration are reviewed. Understanding of basicconcepts of pH, pOH, and electrode mechanism is emphasized. New concepts of pH, pOH,as well as critical examination of activity, and activity coefficients are given. Theemergence of new solid state pH electrodes and replacement of the salt bridge with aconducting wire have opened up a new horizon for pH measurements. A pH buffer solutionwith a conducting wire may be used as a stable reference electrode. The misleadingunlimited linear Nernstian slope should be discarded. Calibration curves with 3 nonlinearportions for the entire 0—14 pH range due to the isoelectric point change effect areexplained. The potential measurement with stirring or unstirring and effects by double layer(DL) and triple layer (TL) will be discussed.
Keywords: electrochemistry; electrode; pH measurement
Article StatisticsClick here to load and display the download statistics.
Notes: Multiple requests from the same IP address are counted as one view.
MDPI and ACS Style
Cheng, K.L.; Zhu, D.-M. On Calibration of pH Meters. Sensors 2005, 5, 209-219.AMA Style
Cheng KL, Zhu D-M. On Calibration of pH Meters. Sensors. 2005; 5(4):209-219.Chicago/Turabian Style
Cheng, K. L.; Zhu, Da-Ming. 2005. "On Calibration of pH Meters." Sensors 5, no. 4: 209-219.