Modeling the Error of the Medtronic Paradigm Veo Enlite Glucose Sensor
AbstractContinuous glucose monitors (CGMs) are prone to inaccuracy due to time lags, sensor drift, calibration errors, and measurement noise. The aim of this study is to derive the model of the error of the second generation Medtronic Paradigm Veo Enlite (ENL) sensor and compare it with the Dexcom SEVEN PLUS (7P), G4 PLATINUM (G4P), and advanced G4 for Artificial Pancreas studies (G4AP) systems. An enhanced methodology to a previously employed technique was utilized to dissect the sensor error into several components. The dataset used included 37 inpatient sessions in 10 subjects with type 1 diabetes (T1D), in which CGMs were worn in parallel and blood glucose (BG) samples were analyzed every 15 ± 5 min Calibration error and sensor drift of the ENL sensor was best described by a linear relationship related to the gain and offset. The mean time lag estimated by the model is 9.4 ± 6.5 min. The overall average mean absolute relative difference (MARD) of the ENL sensor was 11.68 ± 5.07% Calibration error had the highest contribution to total error in the ENL sensor. This was also reported in the 7P, G4P, and G4AP. The model of the ENL sensor error will be useful to test the in silico performance of CGM-based applications, i.e., the artificial pancreas, employing this kind of sensor. View Full-Text
Share & Cite This Article
Biagi, L.; Ramkissoon, C.M.; Facchinetti, A.; Leal, Y.; Vehi, J. Modeling the Error of the Medtronic Paradigm Veo Enlite Glucose Sensor. Sensors 2017, 17, 1361.
Biagi L, Ramkissoon CM, Facchinetti A, Leal Y, Vehi J. Modeling the Error of the Medtronic Paradigm Veo Enlite Glucose Sensor. Sensors. 2017; 17(6):1361.Chicago/Turabian Style
Biagi, Lyvia; Ramkissoon, Charrise M.; Facchinetti, Andrea; Leal, Yenny; Vehi, Josep. 2017. "Modeling the Error of the Medtronic Paradigm Veo Enlite Glucose Sensor." Sensors 17, no. 6: 1361.
Note that from the first issue of 2016, MDPI journals use article numbers instead of page numbers. See further details here.