- freely available
Sensors 2010, 10(8), 7434-7468; doi:10.3390/s100807434
Published: 9 August 2010
Abstract: Analytical applications of Laser Induced Breakdown Spectroscopy (LIBS), namely optical emission spectroscopy of laser-induced plasmas, have been constantly growing thanks to its intrinsic conceptual simplicity and versatility. Qualitative and quantitative analysis can be performed by LIBS both by drawing calibration lines and by using calibration-free methods and some of its features, so as fast multi-elemental response, micro-destructiveness, instrumentation portability, have rendered it particularly suitable for analytical applications in the field of environmental science, space exploration and cultural heritage. This review reports and discusses LIBS achievements in these areas and results obtained for soils and aqueous samples, meteorites and terrestrial samples simulating extraterrestrial planets, and cultural heritage samples, including buildings and objects of various kinds.
The very first experiments of laser induced spark emission were performed in the 1960s and involved a laser beam producing a vapour upon irradiation of a solid target and an electrical spark subsequently exciting the ablated plume . These experiments can be considered as early precursors of Laser Induced Breakdown Spectroscopy (LIBS) and since then LIBS has shown its huge potential as an effective technique for fast multi-elemental analysis.
Starting from the 1980s, significant achievements have been achieved in both laser and detector technology, which have made reliable and relatively inexpensive instruments available to research laboratories . This has promoted a huge development of the technique and its adaptation to the qualitative and quantitative analysis of a wide variety of solid samples [3,4]. These include: metal alloys for metallurgy and for jewellery [5–11], cultural heritage materials [12–16], soils, rocks and sediments [17–23], dried and fresh vegetables [24,25], explosives . Gas mixtures and aerosols have also been studied by LIBS [27–30], as well as solutions, for which single- and multi-pulse approaches have been tested [31–34].
What renders the range of applicability of LIBS so wide are primarily its several instrumental advantages, particularly its extremely flexible experimental set-up. Virtually any kind of substance can be analysed by LIBS with no sample pretreatment and no particular sampling procedures needed. Thus, the analysis can be performed in air , in vacuum, in fluids , and even under extreme conditions such as high temperature and pressure environments. This enables to design compact and portable LIBS instruments, fit for in situ [37,38] measurements and for applications such as space exploration [39–43], as well as potentially adaptable to on flight analysis of asteroids and to stand-off analyses . Besides, LIBS provides a fast response and is thus suitable for on-line process control , has a good sensitivity (generally of the order of ppm) and induces a limited and localized damage on the sample surface, being the removed amount on the order of tens or hundreds of nanograms.
LIBS can be easily coupled with other laser-induced spectroscopic techniques, primarily Raman spectroscopy and Laser-Induced Fluorescence (LIF) (see for example [46,47]). These unique and attractive features qualify LIBS as a technique highly competitive with other largely applied spectroscopic methods for elemental analysis, such as Inductively Coupled Plasma-Optical Emission Spectroscopy (ICP-OES) and Inductively Coupled Plasma-Mass Spectrometry (ICP-MS). In particular, it is often the technique of choice, in some cases coupled with other techniques such as Raman spectroscopy, for geochemical and cultural heritage applications, due to their particular demands for non-destructiveness, speed of analysis and possibility of performing in situ and remote analysis.
On the other hand, the main drawbacks of LIBS include: its strict dependence on the fulfilment of equilibrium conditions in the plasma; the possible saturation of the signal for high concentration elements due to self-absorption effects, even for high-energy lines, that makes their quantification difficult or impossible with LIBS; the onset of matrix effects when analysing samples of very different kinds or very inhomogeneous specimens; the large number of complex physical-chemical phenomena involved in the processes of ablation and plasma formation, evolution and interaction with the background ambient.
In the present paper, the Laser Induced Plasmas (LIP) fundamentals and LIBS analytical methods are outlined, and the recent achievements obtained by our research group in the field of elemental analysis for cultural heritage, environmental and space applications are reviewed and inserted in the frame of the main questions the LIBS community has being addressing in the 2000s [48–50].
2. Basic Principles of LIP and LIBS
When a short laser pulse (with typical duration from ns to fs), is focused on a portion of matter, a significant amount of energy is transferred to the latter, which can result in the formation of a plasma of the irradiated material, a phenomenon usually referred to as breakdown at the material surface. The breakdown can occur only if the pulse irradiance exceeds a threshold value which depends on the state of aggregation of the material ; an irradiance value of ∼1 GW/cm2 is generally considered as an appropriate reference value to yield a high-temperature and high-electron density plasma from virtually any kind of irradiated solid targets. Once ablated, the material expands at supersonic velocity (in vacuum the initial expansion velocity is ∼106 cm/s) in a direction orthogonal to the target (approximately, expansion along the radial component can be neglected in the early stages after ablation). LIP is not a stationary system and typical values for temperature and electron density of LIPs in air range, respectively, between 20,000 K and 1019 cm−3 in the early times after its formation, and 5,000–6,000 K and 1017 cm−3 at the late stages of its evolution.
Several phenomena can contribute, at different extents, to produce the breakdown and yield the plasma, depending on the features of both the laser (wavelength, pulse duration, irradiance) and the irradiated material (state of aggregation, physicochemical characteristics). During its finite lifetime (∼10 μs in air), the produced plasma plume emits radiation that can be detected and used to retrieve qualitative and quantitative information about the elemental composition of the irradiated target, provided that the ablation is stoichiometric, namely, that the composition of the luminous plasma is the same as that of the target. Due to its transient nature, LIP manifests departures from the thermodynamic equilibrium due to the escape of radiation, that is, to radiative dissipation of part of the internal energy of the system. Anyway, such energy loss is negligible with respect to that exchanged by processes between material particles, which are therefore locally and instantaneously in equilibrium and are established in a shorter time than that required for the plasma to expand. Such condition is referred to as Local Thermodynamic Equilibrium (LTE)  and its occurrence is the underlying premise for the use of LIBS as an analytical technique, as it will be better clarified in the following section.
An example of a typical experimental set-up for LIBS experiments is provided in Figure 1, where the instrumentation generally employed in our laboratory is sketched as a block diagram. It mainly consists of a laser source (in our case, a nanosecond Q-switched solid-state laser, Nd:YAG, with first harmonic at 1,064 nm and second harmonic at 532 nm), a system for the radiation detection that comprises a monochromator coupled with an Intensified Charge Coupled Device (ICCD), and a pulse generator to synchronize the plasma production and the emission spectra acquisition. The emitted light can be collected through an optical fibre, as depicted in Figure 1, or directly focused on the monochromator entrance slit. Acquisition parameters, including number of spectra accumulations and averages, starting delay time and gate width of ICCD aperture with respect to laser pulse, can be set via software by means of a computer interfaced with the spectrograph. It is important to note that in LIBS experiments the solid targets are usually mounted on a rotating sample holder, in order to prevent the sample from being deeply dug in the point where the laser beam is focused. Moreover, using this solution, spectra can be acquired from several points on the target surface and all contribute to the overall signal, thus accounting for possible surface inhomogeneity that may affect the accuracy of the analytical determination.
Another common configuration is the Double Pulse LIBS (DP-LIBS), which involves the use of two time-delayed laser pulses to enhance the emission spectra intensity and improve the limits of detection by using either a collinear or an orthogonal geometry. In the first case the two collinear pulses follow the same optical path and reach the target surface perpendicularly with a time delay that can be set according to the background environment . In the second case only one pulse ablates the target, while the other is directed parallel to its surface, with two possible temporal schemes. If the parallel pulse is shot before the ablative pulse, it can produce a preparative pre-spark in air . If it is shot after, it can re-heat the ablated material . In any of the mentioned double-pulse configurations the two lasers can have the same or different pulse widths or wavelengths.
A collinear double pulse configuration can also be used to induce the breakdown of liquid samples, which proceeds with plasma formation mechanisms that are completely different from those occurring in solids. In particular, a single laser pulse is not appropriate for the analysis of bulk liquid samples, because most laser energy is spent in the interaction with the surrounding liquid bulk, which rapidly quenches the plasma. Therefore, a double-pulse approach is necessary for the analysis of submerged targets and liquid bulks, most typically water solutions. The plasma produced by the first pulse rapidly extinguishes, producing a cavitation bubble within the aqueous bulk; the second pulse induces a plasma inside the bubble which evolves similarly to LIPs in gaseous background and can be therefore useful for analytical purposes.
A detailed discussion about the laser-water interaction and the plasma generation under water is beyond the scope of this paper, but the interested reader can refer to previous works of ours and of other groups for a comprehensive treatment [56–58]. In the following sections, short mentions will be given to some applications of analytical LIBS to liquids (aqueous matrices) and submerged solids.
3. Elemental Analysis by LIBS
The use of LIBS for elemental analysis relies on some fundamental assumptions that must be verified, especially when semi-quantitative and quantitative analysis of samples is pursued. First of all the conditions of optically thin plasma and LTE are always the underlying premises for laser induced plasma studies by emission spectroscopy, both for fundamental studies and for analytical applications.
The emission intensity Iul of a peak of frequency vul is related to the number density of excited emitters, Nu by:
Several methods can be found in the literature [60–62] to evaluate and correct self-absorption, and a good practice in analytical LIBS is to select accurately the emission lines to be used for analysis. With this regard, some line selection criteria have been proposed by our group  and are briefly recalled here:
- All spectroscopic lines involving the ground state should be excluded, because they are the most severely affected by self-absorption phenomena, which alter the spectral line profiles and thus hinder the quantitative determination of major elements. For these elements, lines corresponding to transitions with lower energy level below 6,000 cm−1 should also be possibly excluded.
- All transitions with spontaneous emission rate lower than 2 × 106 s−1 should not be considered for quantitative analysis because the corresponding emission times might be comparable with the time associated to the plasma variations .
- All emission lines with high relative intensity (generally >3,000) should be considered with care. In fact, when the emission of low intensity transitions is under the detection limit, e.g., close to the plasma border, these lines are still detectable, therefore they can be useful for the detection of trace elements but they have high probability to overestimate the population, especially for elements at high concentration.
Two further aspects to be considered in the use of analytical LIBS are stoichiometric ablation and plasma homogeneity. All the ablation-based analytical techniques, such as LA-ICP-OES or LA-ICP-MS, as well as LIBS itself, are affected by the stoichiometry of material removal from the target, which may be altered by matrix effects, such as preferential ablation, or fractionation, of some species. The fractionation issue has been addressed in a number of papers [2,64–67] due to the occurrence of non-linear calibration graphs of emission intensity as a function of analyte concentration for reference samples. Fractionation effects are often present in LIBS analysis, in dependence on the matrix under study. For some of them, for example copper-based alloys, the effect can be minimised to reach an acceptable uncertainty of concentration determination by choosing appropriate laser parameters, such as fluence, wavelength, pulse duration, and employing calibration techniques to improve the analytical performance, though they may never be truly overcome .
On the other hand, establishing plasma homogeneity is a more complex issue. As a matter of fact, spatial and temporal gradients of temperature and both electron and heavy species number densities arise in the plasma, due to its fast dynamics and its interaction with the background environment. The peripheral zones of the plasma are the most involved in phenomena such as recombination processes (between ions and electrons within the plasma and between species from the surrounding ambient and plasma species) and confinement by the shock wave contact wall. Consequently, differences may arise between plasma parameters in external zones and those in plasma core. Anyway, for diagnostic or analytical purposes, only the portion coming from the plasma core is usually employed. For this sampled portion, the homogeneity assumption holds, as demonstrated by several authors with spatially-resolved LIBS spectra (see  and references therein). The validity of this assumption can be further ensured by means of appropriate experimental approaches, such as choice of temporal and spatial regions suitable for spectra detection, and methods of data treatment for spatial resolution, such as the Abel inversion .
As for the LTE condition, it is common practice in the LIBS literature to consider that the elementary balances between material particles are in equilibrium, while the light-matter balance is not, if the Boltzmann distribution (see next section) and the McWhirter criterion hold. This criterion states the existence of a critical value for electron density Ne, that is:
4. LIBS Approaches to Quantitative Analysis
Two traditional approaches are possible for quantitative analysis by LIBS, a calibration line approach and a calibration free procedure. The first consists in drawing a calibration line with a set of standards of the same kind and of similar composition to that of the unknown sample. As shown in Equation (1), the intensity of a given emission line is proportional to the number density of emitters, which, in turn, is proportional to the concentration of the emitter (namely, the analyte) in the irradiated sample. Therefore, the emission intensity is linearly correlated to the concentration of a given species in the sample, IX = K[X]. In order to minimize the matrix effects, which could affect the calibration graph linearity, the emission intensity is usually normalized to that of the most abundant element in the sample, which is used as an internal standard. Another possible approach for the calibration line is to normalize the analyte signal to the intensity of the spectral background signal .
Furthermore, multivariate calibration techniques can be used for quantitative analysis by LIBS, such as Partial Least Squares regression (PLS) and Artificial Neural Networks (ANN), which have been drawing increasing attention from the LIBS community over the past few years (see for example [72,73]). Such advanced calibration techniques can reduce the complexity of spectra, enabling to extract the valuable information when the standard calibration curve method may fail. Moreover, chemometric methods such as Principal Components Analysis (PCA) are often coupled to calibration techniques in order to observe data clusters and discriminate outliers, prior to performing the actual quantitative determination [38,74–76].
An alternative to calibration approaches is the so-called Calibration Free (CF) or Self-Calibration method, which is based on the LTE assumption and was first proposed by Ciucci et al. . In LTE conditions, the number density of emitters is given by the Boltzmann distribution:
This method, commonly referred to as the Boltzmann plot method, enables the determination of the excitation temperature from the line slope and N0 from the intercept. N0 is proportional to the species weight percentage of species in the sample, which can be retrieved as described in  through a normalization procedure:
In order for the CF approach to be used for quantitative measurements, the above mentioned assumptions (stoichiometric ablation, LTE condition, plasma homogeneity, optical thin plasma) must be strictly fulfilled.
The CF algorithm represents the only possible approach when the preparation of matrix-matched standards is unfeasible or difficult (e.g., in extreme environments, such as nuclear reactors) and in attaining automated analytical procedures. However, the accuracy of CF quantitative measurements depends more critically on the necessary assumptions for analytical LIBS and may lead to only semi-quantitative results when their fulfilment is unsatisfactory [67,68]. The results that will be presented in the following sections have been achieved mostly by CF-LIBS or by calibration methods, either with the standard calibration line approach or with multivariate techniques.
5. LIBS in Cultural Heritage Science
The most appealing features of LIBS applications in the field of cultural heritage analysis are mainly the following: its micro-destructiveness, with ablated sample portions on the order of fractions of micrograms and induced damage virtually invisible to the naked eye; its potential for fast multi-elemental analysis and capability of simultaneous detection of major and trace elements; its equipment, easily compactable into portable instruments for in situ analyses of piece of arts that cannot be removed from museums or excavation sites, or of historical buildings and wall paintings.
LIBS appears the most suitable technique for dating and rapid classification purposes of metal objects that are among the most abundant archaeological findings, including tools, weapons and utensils, as well as jewellery and statues. Copper-based alloys were the most used in the ancient times and several LIBS analytical and fundamental studies have been dedicated to ancient bronzes, partly because of their suffering from fractionation effects [2,64–66] mainly due to the significantly different melting points of their main constituents. Fantoni et al. [2,68] performed a modelling study aimed at clarifying the fractionation effects that could be expected upon ablation of quaternary bronzes (Cu, Zn, Sn, Pb) with different lasers, namely, single ns-pulse with different (IR to UV) wavelengths, double ns-pulse, single fs-pulse. By comparing the concentrations in the plasma as results of simulations of the different ablation mechanisms with certified concentration values, these authors identified the possible deviations from the stoichiometric ablation and provided a method to optimize the experimental conditions.
A different approach, based on the experimental comparison between ns- and fs-ablation of copper-based alloys, was proposed by our group in , in order to establish if significant differences might arise between the analytical performances of laser pulses of different duration. In  a variant of the CF method was developed which was based on the assumption of partial LTE (pLTE) in the energy range covered by the observed emission lines in the fs and ns cases. The number densities of plasma species were determined by normalization of the line emission intensity to that of the continuum background, which was assumed to follow a Planck-like distribution. With this method, the concentrations of minor elements (Sn, Pb, Ni) in three bronze standards were calculated, while Cu concentration was not determined due to the significant self-absorption of Cu lines, and its weight percentage was just considered as the complementary species for the determined ones. The obtained results, summarized in Table 1, showed that a significant agreement exists between fs- and ns-CF-LIBS, in spite of the known marked differences in the ablation mechanisms of solid targets by ultra-short and short pulses [79–81]. Data in Table 1 show that a good agreement exists between the ns- and fs-LIBS analytical results, as well as between them and the certified values for the concentrations of some elements (Sn, Pb, Ni). It appears therefore reasonable to consider that fractionation was negligible in both cases and that the residual fractionation took place in a comparable degree for both pulse durations.
Once established the suitability of ns-LIBS for quantitative analysis of bronze samples, in a subsequent work , we analyzed by a ns Nd:YAG laser five VII b.C. century objects made of copper-based alloys, namely, a belt (A), a basin (B), a brooch or fibula (C), a piece of armour fragment (D) and a pendant (E), which were excavated from an archaeological site in Southern Italy (Minervino Murge, Apulia). First of all, a depth profiling of Pb and Sn was performed on a piece of scarce archaeological interest (a basin fragment, which was not included in the above selection), in order to estimate any possible discrepancy in their concentrations between the sample surface and bulk. Indeed, it is well known that the external layers of ancient objects strongly differ from the original material as a result of long-time corrosion, so that both morphology and composition can be altered significantly. In the case of bronze artefacts, this corrosion process is often referred to as “bronze disease” which, due to the object interaction with its burial environment (soil, water or sea water), causes it to be covered by a layer (patina) containing mainly Cu II salts and, in some cases, also significant amounts of Sn compounds .
The possibility of analysing multi-layered samples and performing depth profiling is virtually unique to LIBS, among the various techniques used for elemental analysis of archaeological objects. This is because the laser beam focused in a given spot removes successive layers thus different spectra can be observed as a consequence of the compositional changes. In the case of metal and stone handiworks, the incoming laser beam is able to locally remove the altered layers, so that in the irradiated spot the bulk materials, with composition close to the original one, can be analyzed. In particular, in the case of bronze artworks it is important to perform a depth profile analysis to establish how many laser shots are needed to reach the bulk sample, where the composition holds a constant value and can be considered reasonably close to the original one. The Sn and Pb concentrations of the basin fragment are plotted in Figure 2 as a function of the number of laser shots reaching the sample. In this experiment, 600 shots were necessary to remove the patina and reach the bulk material. The end point of the corrosion layer was also made visually evident by the appearance of the typical red-brownish bronze colour. It is interesting to note that, while Pb maintains an almost constant concentration across the sample section, Sn shows a marked depletion in the outer layers, possibly due to the formation of Sn compounds and their subsequent solubilisation in the presence of moisture in the burial soil.
For the quantitative analysis of the above-mentioned unknown samples, a classical calibration line approach was preferred, due to the mentioned difficulty of determining Cu concentration with a CF method. In this approach, Cu was used as the internal standard to draw calibration lines for the quantitative analysis of Sn, Pb, Zn, Fe, Ni, Sb and Si.
A useful element for dating studies is As, which was not present in the analyzed samples. Dating studies of this kind, aimed at distinguishing between arsenical bronzes, typical of Early Bronze Age, and tin bronzes, which were used from the Middle Bronze Age on, have been performed by Fortes et al. . These authors examined a set of 37 specimens from Southern Spain and, based on their As content, sorted them in Bronze Age and Iron Age objects.
In the case of the Italian bronzes analyzed in our laboratory in collaboration with a team of archaeologists, we drew some conclusions about the possible use of the various hand works, on the basis of their Sn content. For example, the piece D was found to have a high Sn content (17 ± 1 wt%), which suggested that it was a part of a piece of armour, probably a helm, because adding large amounts Sn to the alloy hardened the material and prevented it from being easily damaged by blows. On the other hand, unusually high concentrations of Sn and Pb were found in specimens C and E, lying outside the range covered by the available standards. Thus, for these samples only lower limits could be obtained for concentrations (Sn > 39 ± 3 wt%, Pb > 27 ± 2 wt% for the pendant; Sn > 28 ± 2 wt%, Pb 4.3 ± 0.3 wt% for the fibula), which were sufficient to suggest that they must be pieces of jewellery. Indeed, they are not made of common bronzes, but of slightly different copper-based alloys respectively named as bell metal and high leaded tin bronze, which, due to their high Sn percentage, were difficult to tool and used for precious objects. The B sample was a piece of basin, with a well conserved rim and showing typical tin bronze concentrations (Sn 7.6 ± 0.5, Pb 0.17 ± 0.03) which allowed an easy working of the piece, as required to obtain every day use objects of various shapes. The most interesting piece of the set was sample A, a belt composed of four pieces nailed together. For each piece, a depth profiling was performed and Sn and Pb were determined with increasing sampling depth. The obtained results were quite surprising and are reported in Figure 3(a,b): three of the four nailed pieces had typical tin bronze concentrations, while one had a significantly lower Pb content. This suggested that the fourth piece was probably added to the hand work in a later time, likely by a different craftsman, to repair the belt after it got damaged. Nails were also analyzed and they resulted composed of Fe and C, but these components were not quantified.
As an example of the depth profiling procedure, the six spectra of the added piece (Zone 2) are reported in Figure 3. Apparently, the spectra of the external layers are contaminated by high amounts of soil elements such as Si, Mg (see Figure 3(b), Zone 2(a)), while those of the internal layers, used for the quantitative analysis, only display the typical spectral features of bronze elements (see Figure 3(b), Zone 2(f)). These results clearly highlight the significant advantages of using LIBS for this kind of analysis. First of all, no piece was sacrificed, but only a negligible damage was caused, directly on the sample surfaces, with a spot diameter around 180 μm. Moreover, depth profiles were easily performed, which enabled to discriminate the corrosion layer and to make sure that the analyzed sample portion was representative of the original material, at the same time removing the insignificant or extraneous ones.
The spectra of removed layers can also provide useful information to conservators, helping them to identify the causes of object alteration and to plan tailored restoration and/or cleaning actions . Since laser cleaning of a wide variety of artworks has been strongly developing in the last decade, LIBS has become an effective method for on-line monitoring of the cleaning process itself. During contaminants removal from the object surface, the emitted spectra change due to change of elemental composition of the ablated layer. This change can be used to understand the causes of contamination and also to avoid undesired over-cleaning of the object itself. Laser cleaning, with the aid of on-line LIBS monitoring, has been employed to remove extraneous coatings of various kinds, including dark encrustations from marble, terracotta, stone and glass artworks due to exposure to air pollutants [16,83–85], protective and conservative coatings from canvas paintings and synthetic mimicking materials [86–90], corroded layers from metal objects [91,92], dirt (dust and pen and pencil traces) from historical paper documents [93,94], pencil and pen marks on alabaster and marble statues in a stand-off configuration , and patinas on surfaces of historical buildings .
Besides being a powerful technique to control the removal of layers due to aging and pollution of artworks, LIBS is among the few techniques able to provide multi-elemental depth-profiling of intrinsically multi-layered samples, such as ceramics, paintings and frescoes, resulting at the same time only micro-destructive. This potential has been exploited, often coupled with Raman spectroscopy that provides complementary molecular information for dating and provenance studies [97,98] and for pigment identification in a number of different painted artworks, including ceramics [99,100], icons and miniatures [101,102], painted plasters , polychromes on wood , artistic prints , wall paintings of historical buildings [106–107] and historical parchment .
In conclusion of this section, our studies on submerged bronze targets will be briefly mentioned . These studies were conducted in the frame of a project aimed at verifying the feasibility of LIBS analysis for drowned artefacts, such as sunken ships and their content, directly in situ (namely, under seawater). The possibility of obtaining some information about the findings would be of primary importance prior to planning recovery activities and taking decisions about whether the object should be moved or not. In our work, bronze standards immersed in seawater were analyzed in a home-made cuvette with two collinear laser pulses focused through the liquid bulk. The quantitative analysis was obtained with the calibration line method and gave satisfactory results, as compared to similar ones performed in air. The calibration lines for Sn and Pb under water had higher slope and correlation coefficient than those obtained in air, which indicated higher sensitivity and better accuracy. This result suggested that a better thermalization through electron impact took place under water, due to the strong confinement experienced by plasma species inside the bubble, which, together with a longer-lasting LTE condition and a more homogeneous plasma, led to a better analytical performance .
6. LIBS in Environmental Science
The interest of LIBS in the broad field of environmental science lies mainly in its versatility and potential for in situ measurements with a very limited sample pretreatment or none at all. The discussion in this section will be focused on the analysis of soil in its various aspects. Soil is the environmental compartment most affected by heavy metal contamination, being constantly exposed to metal emissions from many anthropogenic sources , and thus potentially subject to long-term modifications, with consequences on the equilibrium of ecosystems. Other kinds of samples will also be included in the discussion, such as rocks and sediments, due to the wide variety of specimens studied by various authors especially during in situ and remote analyses. For a further discussion specifically focused on the LIBS analysis of rocks and minerals, the interested reader may refer to  and references therein. Finally, a short mention will be given in this section to the analysis of aqueous samples, namely seawater and wastewater from industrial plants, due to the strong need for rapid in situ chemical sensors in such field and the potential of LIBS to fulfil such need.
6.1. LIBS Analysis of Soils
Soil analysis is one of the most traditional applications of LIBS. Compared to LIBS, the techniques most typically used for this purpose (such as Atomic Absorption Spectrometry (AAS), X-Ray Fluorescence (XRF), ICP-MS, ICP-OES) and the chemical methods specific for the various analytes, generally regulated by national or international protocols (see for example [110,111]), usually provide better LODs but are time-consuming, need consistent sample manipulation prior to analysis, and require specific soil sampling procedures and accurate specimen selection. Further, and especially for the detection of contaminants in soils after accidental or systematic spill of toxic waste, the use of a real-time, in situ analytical technique such as LIBS would allow the assessment of their degree and kind of contamination, followed by immediate application of tailored remediation strategies, in order to avoid irreversible damages to environment and to human and animal health.
The detection and measurement of levels of toxic metals in soils are among the most important concerns of environmental science, due to the important implications that excessive depletion or enrichment of trace elements in soils can have on the health of living species [108,112]. This is by far the most studied environmental issue as far as LIBS is concerned. In the last decade, many research groups have been developing experimental strategies for this purpose, their common long-term aim being the optimization of reliable LIBS procedures for the assessment of soil degree of contamination and the development of fast remediation strategies. In this context, LIBS research proceeds along two parallel directions: on one hand, the laboratory optimization of LIBS experiments, in order to overcome the many problems of direct analysis of an untreated, exceptionally complex matrix such as soil; on the other hand, the development of high-performance instruments for on field and stand-off investigations.
6.1.1. Laboratory LIBS Analysis of Soils
Since the 1990s many papers have already appeared in the LIBS literature concerning the laboratory analysis of soil samples. Research in this domain is still very active and is mainly aimed at refining the technique and evaluating its capabilities in comparison with other well established techniques. Capitelli et al.  determined the concentration of Cr, Cu, Fe, Mn, Ni, Pb and Zn in a heterogeneous set of soil and sludge samples of various origin and nature and compared the LIBS performance to that of ICP-OES. These two techniques, which can be considered similar in that they are both based on the detection of light emitted by a high-temperature and high-density plasma, were also compared in a work of our group  where the contents of Pb, Zn, Cu, V, Cr were determined in five soil and sludge samples, finely grained and pressed in pellets with no further pretreatment. Calibration lines (reported in Figure 4) were drawn on the basis of ICP-OES measurements. For these lines the emission intensity of analytes was normalized to that of the spectral background, because no suitable internal standard was found in the analyzed set of samples. The good linearity of calibration lines was a clear confirmation of the feasibility of the adopted method and showed that limited matrix effect took place.
In order to compare the LIBS performance with that of ICP-OES, the following procedure was followed for each element. One sample, chosen from the central part of the concentration range covered, was excluded from the calibration curve of the considered element and treated as an unknown sample. A new calibration line was then drawn without the excluded sample, and the concentration of the element was determined consequently. The results of such comparisons are reported in Table 2 and show a good agreement between the two techniques.
In the same paper, we defined an anthropogenic index (A.I.) to rapidly evaluate the degree of heavy metals pollution of soils, in analogy to other enrichment factors proposed in the agrochemical literature . The A.I. was defined as the ratio between the concentration of an element in a polluted soil normalized to that of a reference element, and the corresponding value in an unpolluted soil of the same kind. Taking into account that for a given analyte a direct proportionality exists between line emission intensity and sample concentration (see Equation 1), a modified A.I., relying on LIBS intensity instead of concentration, was defined as:
Several further works have been published on this subject. For example, Hussain et al.  determined the concentration of Cr, Ca, Fe, Mg, Cu, Na, Ni, K and Ba in soil samples collected from a coastal area surrounding the Khursania site of Saudi Arabia along the Persian Gulf, which was involved in a large oil spill during the Gulf War. Another highly toxic contaminant, Cd, was the object of the investigation by Santos et al.  who evaluated its concentration in standard soil and sediment samples. Bousquet et al.  determined Cr in soils collected from industrial areas or prepared in the laboratory by comparing the results obtained with different emission lines and by LIBS and ICP-OES.
The allowed concentration of some heavy metals in unpolluted soil is very low (see for example the limits established by the Italian Ministry of Environment for Hg and Cd, respectively 1 mg/kg and 2 mg/kg in public parks and residential areas, which rise to 5 mg/kg and 15 mg/kg in industrial and commercial areas ). Therefore, some alternative approaches to the classical LIBS have been proposed to improve the sensitivity of the technique, including the use of double pulse irradiation [118,119].
Besides the assessment of the soil degree of pollution by heavy metals, LIBS has also been proposed for other soil-related studies, including soil classification by means of chemometric methods [116,120]; monitoring of the phytoremediation process of heavy metals-polluted soils [121–123]; establishing soil quality, both in terms of micronutrients content  and to aid the evaluation of soil organic matter (SOM) and fertility by means of total C and N quantitative determination [19,125–127].
6.1.2. In situ LIBS Analysis of Soils
As regards the development of LIBS portable systems for direct in situ analysis of soils, so far most efforts have been dedicated to preliminary laboratory testing and calibration of mobile instruments and to addressing the important matrix problems that can be expected during such analyses. For example, Ferreira et al.  have calibrated a portable instrument through laboratory measurements, using a chemometric method based on the use of Artificial Neural Network (ANN) for Cu determination in a heterogeneous set of Brazilian soils, though poor correlation was found between Cu intensity and concentration.
Bousquet et al.  designed and built a prototype of a mobile instrument, equipped with a Nd:YAG ns-laser with the fundamental harmonic and tested it in laboratory, addressing the most serious issue affecting the feasibility of in situ LIBS analysis of soils, namely, the influence of soil moisture on the analytical performance. They attempted two possible approaches to solve this problem. The first consisted in irradiating the samples with long IR pulses, in order to evaporate the excess water prior to focusing the ablative pulse. The second was the classical approach of pressing the samples into pellets to squeeze out water. Of these two strategies, only the latter provided satisfactory results. In all the experiments reviewed so far, matrix effects in quantitative analysis were limited by using either univariate calibration strategies, i.e., the usual calibration line approach, or multivariate ones, e.g., ANN and PLS.
Differently, Corsi et al.  used the CF algorithm (described in Section 4 of this paper), to overcome the calibration problem and also to reduce markedly the time analysis to 1 to 5 minutes, according to the complexity of the produced LIBS spectrum. Their experiments were carried out in the laboratory, in order to test the response of a mobile instrument, equipped so as to analyze either soil pellets or directly the “unsampled ground”. Moreover, they performed a comparative study between the plasma produced by a single laser pulse and by a sequence of two pulses. This study confirmed the DP-LIBS potential for lowering the LODs and increasing the emission intensity and the technique sensitivity for the considered elements. However, the main drawback of using the CF approach for soil analysis is that some of the main soil elements are also contained in the surrounding gas (i.e., air). As a consequence, their weight percentage in the ablated samples cannot be easily quantified, because the spectral signals from air and from the target can be distinguished only if high spatial resolution is applied in the spectra acquisition (see for example ).
Results of measurements actually performed on site have already been published. Wainner et al.  compared the sensitivity of a portable spectrometer and a laboratory instrument for the specific detection of Pb in soils and paint. In this experiment collected samples were analyzed in the laboratory, after doping them with variable amounts of Pb-containing powders or solutions. Then, the portable system was tested on areas of documented Pb contamination, namely, the grounds around a military installation in California used until the late 1950s for the demilitarization by burning of small arms ammunition and Pb-containing paints on the walls of an old military hospital in Colorado. In both cases, analyses performed on site were only qualitative, but allowed the authors to validate the portable system for the detection of Pb concentrations down to 0.01% in weight.
A different kind of study was carried out by Barbini et al.  by using a portable LIBS instrument mounted on board the R/V Italica ship during the XVI Antarctic campaign (2000–2001). Sediment slices of 2 cm diameter sampled from the 30 cm diameter and 50 cm height cylindrical sections extracted from the sea bed were oven-dried and then qualitatively analyzed to identify the presence and distribution of various elements in the different sediment horizons, with the aim of locating the most interesting sites with a fast on line method.
The quantitative analysis of two selected pressed sediments was performed with a method proposed by the same group in  and based on the evaluation of self absorption in the core zone of the plasma while neglecting that in the external zones. The obtained results were compared with certified values, resulting in a satisfactory agreement except for Al, which resulted severely underestimated by LIBS.
Recently, quantitative analysis of Pb in road sediments has been reported by Cuñat et al.  by means of a man-portable LIBS instrument, consisting of a backpack and a probe unit of overall weight of approximately 5 kg and a size of 45 cm × 27 cm × 15 cm. An on-field campaign was performed on soil samples in the Cerrado de Calderon tunnel in Malaga, Spain, to evaluate the Pb released by the road traffic. Depth-profiling of Pb and Ti concentrations, normalized to that of Ca, were performed on the tunnel walls every 2 m along the tunnel length. The comparison of results obtained by LIBS quantitative analysis with those of AAS laboratory analysis of samples collected in the gallery, indicated the existence of a good correlation between the two techniques and a satisfactory LOD which resulted lower than those demanded by most environmental organizations.
6.1.3. Stand-off LIBS Analysis of Soils
Stand-off LIBS applications have also been described for soils analysis. For example, López-Moreno et al.  investigated samples from a coastal site subjected to a high industrial activity and therefore to potentially significant heavy metal contamination. These samples, including rocks, soils, trees leaves and bark, stone and wall fragments from the factory or from surrounding buildings, were qualitatively and quantitatively analyzed in laboratory, at a stand-off distance of 12 m. The effects of the sample conditions (i.e., salinity, moisture) and characteristics (i.e., matrix composition, orientation with respect to the incoming laser, distance from the contamination source) were investigated. Further, depth-profiling of rocks and stone samples clearly revealed a surface contamination by Cr whose emission concentration was unexpectedly high on the samples surface and decreased until eventually disappearing in the bulk of the unpolluted sample.
In a later paper , the same authors described the design, construction and assessment of a field-deployable instrument for open path remote LIBS, which was optimized for acquisitions in a hundreds of metres range. The authors investigated which critical parameters should be taken into account in stand-off experiments by using acquisition tests on Ti as a reference sample and at increasing distances. An equation was retrieved for the LIBS signal dependence on the acquisition range (distance from the irradiated target), laser wavelength, beam divergence and dimensions of the receiving optical element, which are factors more critical in a stand-off configuration than in laboratory experiments. Further, the analyte emission was compared to that of air components, in order to evaluate any interference due to possible breakdown along the laser beam path.
6.2. LIBS Analysis of Aqueous Samples
From the environmental point of view, what renders LIBS in liquid environment particularly interesting with respect to well-established techniques for liquids analysis such as ICP-OES  is the possibility of planning on-site and remote applications, dedicated essentially to monitor the concentration of metals in wastewaters from industrial plants and in seawater. As briefly discussed in Section 2, focusing a single laser pulse through a liquid bulk does not usually provide suitable analytical signal, but low emission spectra mainly dominated by a continuum background radiation, where only few low-energy lines can be observed [56,57].
Though some works report results about the detection of various elements by focusing a laser in bulk aqueous samples (Koch et al. for Cr , De Giacomo et al. for Al, Na, Ca and Li in various water solutions ), over the past several years different Single-Pulse (SP) configurations have been tried to overcome the intrinsic drawbacks of the traditional configuration. For example, Hussain et al. in  determined the concentrations of Ca, Mg, P, Si, Fe, Na and K in wastewater samples from dairy products plants. The laser beam was focused on the liquid surface contained in a suitably designed cell so as to avoid splashing on the optical components, and high energy laser pulses (100 mJ) were used to enhance the emission intensity. This approach provided accurate analytical results that were in good agreement with those obtained with ICP-OES of the same samples.
Rai et al.  used a liquid jet configuration to detect Cr in wastewater from electroplating industries. High energy pulses (120 mJ) were focused just below the surface of a solution stream from a Teflon nozzle, so to produce a plasma at its front surface. Good agreement was found between the LIBS and AAS results for the only analyzed unknown sample, provided that a suitable calibration line could be drawn prior to the LIBS experiment.
The liquid stream approach was also adopted by Cheung et al. [138,139], who used an ArF excimer laser to ablate the sample. They obtained good results for the analysis of water solutions, with LODs on the order of ppb, thanks to the VUV-laser wavelength, which provided a better signal-to-noise ratio and a shorter-lived continuum emission than those usually found with non-VUV-LIBS.
However, for bulk liquids and submerged targets the best improvements can be obtained by using a Double-Pulse (DP) approach. The cavitation bubble generated by the extinction of the first pulse offers a favourable microenvironment for chemical analysis and represents the only possible choice if liquid bulk analysis is pursued, as in the case of in situ analysis of seawater. In one of our works, previously cited for the analysis of submerged bronze samples, a laboratory analysis of tap water was also performed which showed significant improvements for both qualitative and quantitative analysis made possible by the DP approach .
An interesting application, proposed by Michel et al. [36,140], is the use of LIBS as a chemical sensor at high liquid pressures simulating oceanic depths, aimed at in situ identification of hydrothermal vents. Clearly, the effects of quenching and spectral interference by continuum radiation are even more severe at high pressures (up to hundreds of bars), which also limit the lifetime of the bubble itself and the temporal windows when the analysis can be pursued. However, it is important to underline that LIBS in the cavitation bubble appears as the only possible choice for the analysis of hydrothermal vents, whose emitted fluids can be observed only in situ, because irreversible changes would occur if they were moved from underwater to a common laboratory . The effect of some key experimental parameters such as delay between the laser pulses, background pressure, pulse energies and acquisition gate time were studied and optimized for the detection of five critical elements (Na, Ca, Mn, Mg, K).
7. LIBS in Space Science
All the mentioned advantages of LIBS as an analytical technique are particularly attractive for its use in space applications. LIBS feasibility as an in situ and remote technique renders it particularly appropriate for space exploration purposes, also because LIBS instruments can be integrated in rovers for planetary surface analysis and easily coupled with Raman spectrometers for simultaneous multi-elemental and molecular information. In particular, the next NASA and ESA missions to Mars (respectively Mars Science Laboratory, scheduled for launch in Fall 2011  and ExoMars, scheduled for launch in 2016 ) will include also a compact spectrometer for stand-off LIBS analysis on the respective rovers. Moreover, LIBS has a specific strong advantage in the similarity between LIPs and the ablation plasmas produced by falling meteoroids while crossing the Earth atmosphere [143–145]. Laser ablation techniques can be therefore a useful resource to simulate such spectra and contribute to asteroids and meteorites recognition. In particular, the mass of flying bodies can be determined by analysis of their luminous ablation plasmas and their elemental composition, which in turn allows predicting their trajectory and the effect of their impact [146,147]. Further, the optimization of LIBS experiments in laboratory conditions controlled is expected to promote future development of on-flight techniques for the direct analysis of flying meteoroids.
The elemental analysis of meteorites, which is employed for their classification, can be performed on a geological basis (petrological and mineralogical classifications) and/or on a chemical one , depending on the kind of meteorite. With respect to established techniques usually employed for this aim, including INAA (Instrumental Neutron Activation Analysis), PIXE (Proton-Induced X-ray Emission), ICP-MS and XRF, LIBS presents the obvious advantages of not requiring any analytical chamber or radioactive sources and of being adaptable also to extreme conditions and out of the laboratory, with no limitations of the atomic weight of the detectable elements. Therefore, research in the field of LIBS spatial applications proceeds dealing with both the elemental analysis of actual space objects and the development of experimental strategies and prototypes for planet exploration, mainly performed on terrestrial samples similar to Martian and Lunar ones.
Thompson et al. [43,149,150] have analyzed two Martian meteorites, Zagami and Dar al Gani 476, in a stand-off mode at distances of up to 8.3 m, aiming to test laboratory systems as similar as possible to the ChemCam instrument built for the NASA Mars Science Laboratory and to develop calibration strategies for improving the method reliability and reduce the matrix effects on the analytical performance. Calibration lines have also been drawn using terrestrial basalts to test the remote-LIBS capability of distinguishing between the two different Martian meteorites.
In a work of ours , a CF method was employed to analyze a set of four known meteorite samples of different classes, namely an L6 chondrite (Sahara 98222), a Lunar meteorite (Dhofar 461), a Martian meteorite (Dhofar 019) and an iron meteorite (Sikhote Alin). The weight percentages of the main elements were determined, while the trace elements were detected only qualitatively, and the LIBS results were compared with those in the literature, obtained by INAA and ICP-MS. Due to the difficulty in preparing a satisfactory set of matrix-matched standards for the analysis of meteorites of different classes and, above all, to the impossibility of applying calibration procedures in the case of completely unknown samples, we opted for the CF-LIBS approach. The obtained results were in good agreement with the weight percentages reported in the literature for most elements, although the CF accuracy was still not very good (around 15%).
Successively, we studied in detail two further iron meteorites, Campo del Cielo and Toluca  and for the very first time in LIBS literature, we obtained concentration profiles of the main elements, Fe and Ni, and of the most abundant trace element, Co, along the Widmanstätten structure, visible on the Toluca meteorite analyzed slab (see Figure 6(a)). This is a pattern of alternating taenite and kamacite lamellae, peculiar to iron meteorites of the Toluca class (medium octahedrites) and its detection, coupled with theoretical models, is an important method for the study of the parent body, the mechanism of formation and the age of meteorites . The space resolution achieved by the used LIBS experimental set-up (laser spot diameter 25 μm) was not comparable with that of techniques that use electron or ion microprobes, but enabled anyway the clear identification of seven lamellae and the determination of the main elements concentrations across the Widmanstätten pattern. The weight percentages of Ni, Fe and Co across one lamella are reported in Figure 6(b) as an example. It is noteworthy that the spatial resolution could be easily improved by using a micro-LIBS set-up, which consists of a microscope through which the laser beam can be focused, thus obtaining micrometric spots. The good agreement found with a simple cooling rate theoretical model  demonstrated the potential competitiveness of LIBS with the classical techniques for meteorites analysis. In , we also determined the oxygen content of the Martian meteorite Dhofar 019 by using a CF method and irradiating the sample in reduced pressure Ar atmosphere.
So far, most work of the LIBS community in the field of space applications has been dedicated to the analysis, mainly at short stand-off distance, of rocks and materials simulating surface and atmosphere of extraterrestrial planets. The purpose of these studies is the optimization of prototypes and methods for forthcoming planetary exploration missions, which must fulfil stringent requirements for the dimensions of instrumentation. To this aim many European and US groups are working, in the frame of the above mentioned ESA and NASA missions to Mars.
Colao et al.  performed CF-LIBS analysis of Martian rocks analogues (volcanic and sedimentary rock collected near the Etna Volcano in Italy and on Chilean Andes) in a special analytical chamber, in a rarefied gaseous environment (CO2:N2:O2 = 95:4:1, total pressure adjusted between 6 and 9 Torr) simulating composition and low pressure of the Martian atmosphere. Prior to Martian analogues investigation, these authors validated the CF method for the analysis of complex matrices by using several soil and rock standards, and found a substantial agreement between LIBS values and certified values. The most important deviations were observed for easily ionizable elements, especially Ca.
Similar preparatory studies have been carried out by several authors. Sallé et al.  tested a stand-off system for qualitative analysis of a set of metallic alloys and basalt samples in the range 2–12 m, and detected the major components up to 12 m and the trace elements up to 8 m. A different configuration, at a distance of 3 m, was used for quantitative analysis, particularly of Cl and S, two elements often difficult to detect by LIBS due to their high excitation energies (≥ 10 eV) but extremely important for the study of Martian geology. For this purpose, samples containing Cl and S including KCl, NaCl, pure S, FeS2, PbS, CuS, were analyzed, in order to produce a spectral database and calibration lines. Experiments were conducted in simulated Martian atmosphere, a feature that was found to increase the signal to background ratio of Cl and S emission lines and improve their detection limit.
Radziemski et al.  investigated the feasibility of acquiring spectra in the VUV region (particularly 100–200 nm) for the analysis of Cl, S and other important elements, such as As, Br, C, P, of difficult detection in the regions normally covered by most LIBS spectrometers (visible-near IR). Also in this case, soils, minerals and metal samples were analyzed in simulated Martian atmosphere and used to draw calibration lines. A good linearity was obtained for calibration lines and the precision was comparable with that of usual LIBS measurements at longer wavelengths, but the authors pointed out an intrinsic drawback. As CO2 absorbs in the region 130–160 nm, the feasibility of such method at long stand-off distances in Martian atmosphere may be limited, being however suitable on asteroids and on the Moon.
Further experiments simulating Mars and Moon atmospheres (respectively, 7 Torr CO2 and near vacuum conditions, 50 m Torr in air) were performed by Sallé et al. , who studied the influence of the ambient pressure on calibration curves prepared for elements commonly observed in terrestrial soils (e.g., Al, Ba, Ca, Cr, Fe, K, Li, Mg, Mn, Na, Ni, Si, Sr, Ti) by using certified soil and clay pellets as standard samples. In a later work , these authors compared four different analytical methods, i.e., multi-matrices calibration curves, external normalization with correction for total concentrations, internal normalization by a major element and CF approach, in order to select the one with the best performance in overcoming matrix effects. The analyzed samples were terrestrial rocks similar to Martian ones, i.e., mafic volcanic rocks, gabbro and limestone, which were irradiated in CO2 at 9 m bar, at an intermediate stand-off distance (3 m). The best accuracy was obtained using the external normalization method, which consists in the determination of a matrix factor accounting for the difference in the number of ablated atoms of a given species between the standard and the unknown samples. This approach, as the CF one, is useful in case no reference sample is available but may not be very accurate and requires the detection of all elements. When this is not possible, i.e., in the case only a limited portion of the spectrum can be acquired, the internal standardization method is considered a valid alternative, as it can be applied only to the detected analyte.
In addition to the atmosphere composition, another factor which has been taken into account by some groups to simulate Mars conditions is the presence of ice and the moisture degree of Martian rocks. The presence of water, as already discussed in the section dedicated to in situ analysis of soils, can affect the laser-target coupling and consequently the LIBS signal intensity. On the other hand, it can give a significant contribution to verify current or past presence of life on the Red Planet.
Arp et al.  simulated the analysis of polar layer deposits at Mars poles by acquiring at short and intermediate stand-off distance LIBS spectra at Martian atmospheric pressure of several kinds of samples, including pure water ice, mixtures of ice and synthetic silicate certified materials mimicking soils and mixtures of ice and powdered certified rocks of various types. The effects on LIBS spectra of adding increasing amounts of ice to soil samples was studied and the penetration properties of the laser pulse through frozen samples were determined. Furthermore, the LODs evaluated at short distance and in a stand-off mode for dry soils and ice/soil mixtures yielded comparable results.
Lazic et al. in  applied cooling and heating cycles to terrestrial soil and rock samples in a simulated Martian atmosphere, in order to investigate the effect of the presence of ice and water in the samples pores on the emission spectra, as a function of temperature in the range from 25 °C to −60 °C. The preparation of samples, that is, their surface roughness and pore dimensions, was found to significantly affect the emission spectra. In particular, these effects were ascribed to the phase transitions undergone by the supercooled water present in the sample pores and, in particular, to the decrease in the laser-sample coupling due to discontinuities in the physical properties of the coexisting liquid water and ice phases. In a later work  these authors further refined their investigations by performing measurements at pressures below and above the triple point of water, and found in the latter case a clear anti-correlation between H emission intensity (used as a marker of water vaporization) and the sum of the emission intensities of the main elements constituting the rock samples.
Finally, at least a short mention should be given here to the exploration of other extraterrestrial planets. Although Mars is the most studied, due to its resemblance to Earth and its relatively mild surface conditions, other planets have received some attention, such as for example Venus. Its exploration by lander missions is challenging due to its extremely harsh conditions (CO2 atmosphere, 90 atm pressure and 735 K surface temperature). The high pressure and temperature would limit the lifetime of any surface probes to no longer than few hours, therefore fast in situ analytical techniques such as LIBS would be required.
A preliminary study of LIBS feasibility for Venus missions was reported by Arp et al. , who acquired LIBS spectra of basalt rock samples in a high pressure chamber filled with pure CO2 or CO2/N2 mixtures at high pressure. They investigated the effect of experimental features such as the temporal evolution of emission signals and the dependence on the pulse energy.
More recently, papers about coupled LIBS-Raman analysis of Venus-analogue rocks in Venus-like atmosphere have appeared , which demonstrate that some useful information can be retrieved, though LIBS spectra are complicated by the appearance of emission lines that are not typically observable under terrestrial conditions and are caused by the increased number of collisions between plasma species and the surrounding supercritical CO2. A primary concern for the use of emission spectroscopy in Venus conditions is represented by the quenching of the LIBS signal, due to pressure effects which decrease the plasma lifetime, the transmission properties of CO2 and the continuum background interference in Raman spectra. Specific approaches and experimental strategies are required to address such issues, which were first investigated by Lambert et al. , in view of future development of an on-flight instrument for LIBS-Raman analysis.
The present paper has reviewed LIBS applications to qualitative and quantitative elemental analysis in the broad fields of environmental science, space exploration and cultural heritage. LIBS is a multi-elemental technique consisting in the production of a luminous transient plasma by focusing an intense laser pulse (of ns or less duration for typical analytical applications) on a target and in the subsequent detection of the plasma emission spectra. The main approaches to quantitative analysis by LIBS have been discussed and examples have been provided of analytical results obtained using either the classical calibration line approach (with some variants according to the calibration strategy adopted) or the Calibration Free approach. The LIBS capability of providing the elemental composition of virtually any kind of samples with no sample pretreatment and its potential for in situ and remote analysis, even in harsh environments, together with its micro-destructiveness and comparatively low cost, renders this technique particularly suitable for the analysis of a wide range of samples.
This review has focused on the analysis of various kinds of artworks, with a special emphasis on archaeological findings made of copper-based alloys, soils and meteorites and space objects. Examples of investigations performed with portable instruments for in situ LIBS measurements and of stand-off analyses, especially in the environmental and space exploration fields, have also been briefly presented and discussed, together with the main problems associated to the use of these variants of the standard laboratory technique. A short mention has been finally given to the double-pulse LIBS technique, used for the analysis of submerged solid targets or of bulk liquids, together with results about its application in the cultural heritage and environmental fields.
- Brech, F; Cross, L. Optical Microemission Stimulated by A Ruby Laser. Appl. Spectrosc 1962, 16, 59. [Google Scholar]
- Radziemski, LJ. From LASER to LIBS, the Path of Technology Development. Spectrochim. Acta B 2002, 57, 1109–1113. [Google Scholar]
- Fantoni, R; Caneve, L; Colao, F; Fornarini, L; Lazic, V; Spizzichino, V. Methodologies for Laboratory Laser Induced Breakdown Spectroscopy semi-Quantitative and Quantitative Analysis—A Review. Spectrochim. Acta B 2008, 63, 1097–1108. [Google Scholar]
- Tognoni, E; Palleschi, V; Corsi, M; Cristoforetti, G. Quantitative micro-Analysis by Laser-induced Breakdown Spectroscopy: A Review of the Experimental Approaches. Spectrochim. Acta B 2002, 57, 1115–1130. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; De Pascale, O; Gaudiuso, R; Teghil, R; Santagata, A; Parisi, GP. ns- and fs-LIBS of Copper-based-Alloys: A Different Approach. Appl. Surf. Sci 2007, 253, 7677–7681. [Google Scholar]
- Elhassan, A; Giakoumaki, A; Anglos, D; Ingo, GM; Robbiola, L; Harith, MA. Nanosecond and Femtosecond Laser Induced Breakdown Spectroscopic Analysis of Bonze Alloys. Spectrochim. Acta B 2008, 63, 504–511. [Google Scholar]
- Noll, R; Bette, H; Brysch, A; Kraushaar, M; Mönch, I; Peter, L; Sturm, V. Laser-induced Breakdown Spectrometry—Applications for Production Control and Quality Assurance in the Steel Industry. Spectrochim. Acta B 2001, 56, 637–649. [Google Scholar]
- Galbacs, G; Jedlinszki, N; Cseh, G; Galbacs, Z; Túri, L. Accurate Quantitative Analysis of Gold Alloys using multi-Pulse Laser Induced Breakdown Spectroscopy and a Correlation-based Calibration Method. Spectrochim. Acta B 2008, 63, 591–597. [Google Scholar]
- Garcia-Ayuso, LE; Amador-Hernández, J; Fernández-Romero, JM; Luque de Castro, MD. Characterization of Jewellery Products by Laser-induced Breakdown Spectroscopy. Anal. Chim. Acta 2002, 457, 247–256. [Google Scholar]
- Jurado-Lopez, A; Luque de Castro, MD. Rank Correlation of Laser-induced Breakdown Spectroscopic Data for the Identification of Alloys used in Jewelry Manufacture. Spectrochim. Acta B 2003, 58, 1291–1299. [Google Scholar]
- Gruber, J; Heitz, J; Strasser, H; Bäuerle, D; Ramaseder, N. Rapid in-situ Analysis of Liquid Steel by Laser-induced Breakdown Spectroscopy. Spectrochim. Acta B 2001, 56, 685–693. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; De Pascale, O; Gaudiuso, R; Santagata, A; Teghil, R. Laser Induced Breakdown Spectroscopy Methodology for the Analysis of Copper-based-Alloys used in Ancient Artworks. Spectrochim. Acta B 2008, 63, 585–590. [Google Scholar]
- Giakoumaki, A; Melessanaki, K; Anglos, D. Laser-induced Breakdown Spectroscopy (LIBS) in Archaeological Science—Applications and Prospects. Anal. Bioanal. Chem 2007, 387, 749–760. [Google Scholar]
- Fortes, FJ; Cortés, M; Simón, MD; Cabalín, LM; Laserna, JJ. Chronocultural Sorting of Archaeological Bronze Objects using Laser-induced Breakdown Spectrometry. Anal. Chim. Acta 2005, 554, 136–143. [Google Scholar]
- Dolgin, B; Chen, Y; Bulatov, V; Schechter, I. Use of LIBS for Rapid Characterization of Parchment. Anal. Bioanal. Chem 2006, 386, 1535–1541. [Google Scholar]
- Klein, S; Hildenhagen, J; Dickmann, K; Stratoudaki, T; Zafiropulos, V. LIBS-Spectroscopy for Monitoring and Control of the Laser Cleaning Process of Stone and Medieval Glass. J. Cult. Herit 2000, 1, 287–292. [Google Scholar]
- Senesi, GS; Dell’Aglio, M; Gaudiuso, R; De Giacomo, A; Zaccone, C; De Pascale, O; Miano, TM; Capitelli, M. Heavy Metal Concentrations in Soils as determined by Laser-induced Breakdown Spectroscopy (LIBS), with Special Emphasis on Chromium. Environ. Res 2009, 109, 413–420. [Google Scholar]
- Lazic, V; Barbini, R; Colao, F; Fantoni, R; Palucci, A. Self-absorption Model in Quantitative Laser Induced Breakdown Spectroscopy Measurements on Soils and Sediments. Spectrochim. Acta B 2001, 56, 807–820. [Google Scholar]
- Belkov, MV; Burakov, VS; De Giacomo, A; Kiris, VV; Raikov, SN; Tarasenko, NV. Comparison of Two Laser-induced Breakdown Spectroscopy Techniques for Total Carbon Measurement in Soils. Spectrochim. Acta B 2009, 64, 899–904. [Google Scholar]
- Capitelli, F; Colao, F; Provenzano, MR; Fantoni, R; Brunetti, G; Senesi, N. Determination of Heavy Metals in Soils by Laser Induced Breakdown Spectroscopy. Geoderma 2002, 106, 45–62. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; De Pascale, O; Longo, S; Capitelli, M. Laser Induced Breakdown Spectroscopy on Meteorites. Spectrochim. Acta B 2007, 62, 1606–1611. [Google Scholar]
- Barbini, R; Colao, F; Lazic, V; Fantoni, R; Palucci, A; Angelone, M. On Board LIBS Analysis of Marine Aediments Collected during the XVI Italian Campaign in Antarctica. Spectrochim. Acta B 2002, 57, 1203–1218. [Google Scholar]
- Cunat, J; Fortes, FJ; Laserna, JJ. Real Time and in situ Determination of Lead in Road Sediments using a Man-portable Laser-induced Breakdown Spectroscopy Analyzer. Anal. Chim. Acta 2009, 633, 38–42. [Google Scholar]
- Lei, W; Motto-Ros, V; Boueri, M; Ma, Q; Zhang, D; Zheng, L; Zeng, H; Yu, J. Time-resolved Characterization of Laser-induced Plasma from Fresh Potatoes. Spectrochim. Acta B 2009, 64, 891–898. [Google Scholar]
- Juve, V; Portelli, R; Boueri, M; Baudelet, M; Yu, J. Space-resolved Analysis of Trace Elements in Fresh Vegetables using Ultraviolet Nanosecond Laser-induced Breakdown Spectroscopy. Spectrochim. Acta B 2008, 63, 1047–1053. [Google Scholar]
- Gottfried, JL; De Lucia, FC; Munson, CA, Jr; Miziolek, AW. Laser-induced Breakdown Spectroscopy for Detection of Explosives Residues: A Review of Recent Advances, Challenges, and Future Prospects. Anal. Bioanal. Chem 2009, 395, 283–300. [Google Scholar]
- Windom, BC; Diwakar, PK; Hahn, DW. Dual-Pulse Laser Induced Breakdown Spectroscopy for Analysis of Gaseous and Aerosol Systems: Plasma-Analyte Interactions. Spectrochim. Acta B 2006, 61, 788–796. [Google Scholar]
- Asgill, ME; Hahn, DW. Particle Size Limits for Quantitative Aerosol Analysis using Laser-induced Breakdown Spectroscopy: Temporal Considerations. Spectrochim. Acta B 2009, 64, 1153–1158. [Google Scholar]
- Carranza, JE; Fisher, BT; Yoder, GD; Hahn, DW. On-line Analysis of Ambient Air Aerosols using Laser-induced Breakdown Spectroscopy. Spectrochim. Acta B 2001, 56, 851–864. [Google Scholar]
- Zimmer, L; Tachibana, S. Laser Induced Plasma Spectroscopy for Local Equivalence Ratio Measurements in an Oscillating Combustion Environment. Proc. Combust. Inst 2007, 31, 737–745. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; De Pascale, O. Single Pulse-Laser Induced Breakdown Spectroscopy in Aqueous Solution. Appl. Phys. A 2004, 79, 1035–1038. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; Colao, F; Fantoni, R; Lazic, V. Double-Pulse LIBS in Bulk Water and on Submerged Bronze Samples. Appl. Surf. Sci 2005, 247, 157–162. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; De Pascale, O; Capitelli, M. From Single Pulse to Double Pulse ns-Laser Induced Breakdown Spectroscopy under Water: Elemental Analysis of Aqueous Solutions and Submerged Solid Samples. Spectrochim. Acta B 2007, 62, 721–737. [Google Scholar]
- Lawrence-Snyder, M; Scaffidi, J; Angel, SM; Michel, APM; Chave, AD. Sequential-Pulse Laser-induced Breakdown Spectroscopy of High-Pressure Bulk Aqueous Solutions. Appl. Spectrosc 2007, 61, 171–176. [Google Scholar]
- Noda, M; Deguchi, Y; Iwasaki, S; Yoshikawa, N. Detection of Carbon Content in a High-Temperature and High Pressure Environment using Laser-induced Breakdown Spectroscopy. Spectrochim. Acta B 2002, 57, 701–709. [Google Scholar]
- Michel, APM; Lawrence-Snyder, M; Angel, SM; Chave, AD. Laser-induced Breakdown Spectroscopy of Bulk Aqueous Solutions at Oceanic Pressures: Evaluation of Key Measurement Parameters. Appl. Opt 2007, 46, 2507–2515. [Google Scholar]
- Goujon, J; Giakoumaki, A; Pinon, V; Musset, O; Anglos, D; Georgiou, E; Boquillon, JP. A Compact and Portable Laser-induced Breakdown Spectroscopy Instrument for Single and Double Pulse Applications. Spectrochim. Acta B 2008, 63, 1091–1096. [Google Scholar]
- Bousquet, B; Travaille, G; Ismael, A; Canioni, L; Michel-Le Pierres, K; Brasseur, E; Roy, S; Le Hecho, I; Larregieu, M; Tellier, S; Potin-Gautier, M; Boriachon, T; Wazen, P; Diard, A; Belbeze, S. Development of A Mobile System based on Laser-induced Breakdown Spectroscopy and Dedicated to in situ Analysis of Polluted Soils. Spectrochim. Acta B 2008, 63, 1085–1090. [Google Scholar]
- Arp, ZA; Cremers, DA; Harris, RD; Oschwald, DM; Parker, GR, Jr; Wayne, DM. Feasibility of Generating a Useful Laser-induced Breakdown Spectroscopy Plasma on Rocks at High Pressure: Preliminary Study for A Venus Mission. Spectrochim. Acta B 2004, 59, 987–999. [Google Scholar]
- Lazic, V; Rauschenbach, I; Jovicevic, S; Jessberger, EK; Fantoni, R; Di Fino, M. Laser Induced Breakdown Spectroscopy of Soils, Rocks and Ice at Subzero Temperatures in Simulated Martian Conditions. Spectrochim. Acta B 2007, 62, 1546–1556. [Google Scholar]
- Rauschenbach, I; Lazic, V; Pavlov, SG; Hübers, H-W; Jessberger, EK. Laser Induced Breakdown Spectroscopy on Soils and Rocks: Influence of the Sample Temperature, Moisture and Roughness. Spectrochim. Acta B 2008, 63, 1205–1215. [Google Scholar]
- Salle, B; Cremers, DA; Maurice, S; Wiens, RC. Laser-induced Breakdown Spectroscopy for Space Exploration Applications: Influence of the Ambient Pressure on the Calibration Curves Prepared from Soil and Clay Samples. Spectrochim. Acta B 2005, 60, 479–490. [Google Scholar]
- Thompson, J; Wiens, R; Clegg, SM; Barefield, J; Vaniman, DT; Newsom, HE. Remote Laser Induced Breakdown Spectroscopy (LIBS) Analyses of DaG 476 and Zagami Martian Meteorites. Proceedings of the 37th Lunar and Planetary Science Conference, Leauge City, TX, USA, 13–17 March 2006.
- Salle, B; Mauchien, P; Maurice, S. Laser-induced Breakdown Spectroscopy in Open-Path Configuration for the Analysis of Distant Objects. Spectrochim. Acta B 2007, 62, 739–768. [Google Scholar]
- Boue-Bigne, F. Laser-induced Breakdown Spectroscopy Applications in the Steel Industry: Rapid Analysis of Segregation and Decarburization. Spectrochim. Acta B 2008, 63, 1122–1129. [Google Scholar]
- Hilbk-Kortenbruck, F; Noll, R; Wintjens, P; Falk, H; Becker, C. Analysis of Heavy Metals in Soils using Laser-induced Breakdown Spectrometry Combined with Laser-induced Fluorescence. Spectrochim. Acta B 2001, 56, 933–945. [Google Scholar]
- Dreyer, CB; Mungas, GS; Thanh, P; Radziszewski, JG. Study of sub-mJ-Excited Laser-induced Plasma Combined with Raman Spectroscopy under Mars Atmosphere-Simulated Conditions. Spectrochim. Acta B 2007, 62, 1448–1459. [Google Scholar]
- Handbook of Laser-induced Breakdown Spectroscopy; Cremers, David A, Radziemski, Leon J, Eds.; John Wiley & Sons: Hoboken, NJ, USA, 11 July 2006.
- Laser-induced Breakdown Spectroscopy; Singh, Jagdish P, Thakur, Surya Narayan, Eds.; Cambridge University Press: Cambridge, UK, 15 November 2007.
- Laser-induced Breakdown Spectroscopy; Miziolek, Andrzej W, Palleschi, Vincenzo, Schechter, Israel, Eds.; Cambridge University Press: Cambridge, UK, 28 August 2008.
- Laser-induced Plasma and Applications; Radziemski, LJ, Cremers, DA, Eds.; Marcel Dekker: New York, NY, USA, 1989.
- Van der Mullen, JAM. Excitation Equilibria in Plasmas: A Classification. Phys. Rep 1990, 2, 109–220. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; Bruno, D; Gaudiuso, R; De Pascale, O. Experimental and Theoretical Comparison of Single-Pulse and Double-Pulse Laser Induced Breakdown Spectroscopy on Metallic Samples. Spectrochim. Acta B 2008, 63, 805–816. [Google Scholar]
- Stratis, DN; Eland, KL; Angel, SM. Effect of Pulse Delay Time on A pre-Ablation Dual-Pulse LIBS Plasma. Appl. Spectrosc 2001, 55, 10. [Google Scholar]
- Santagata, A; De Giacomo, A; De Pascale, O; Dell’Aglio, M; Teghil, R; De Bonis, A; Corrente, M; Parisi, GP; Orlando, S. Femtosecond/Nanosecond Dual-Pulse Orthogonal Geometry Plasma Plume Reheating for Compositional Analysis of Ancient Copper-based-Alloy Artworks. J. Phys. Conf. Ser 2007, 59, 585–590. [Google Scholar]
- Casavola, A; De Giacomo, A; Dell’Aglio, M; Taccogna, F; Colonna, G; De Pascale, O; Longo, S. Experimental Investigation and Modelling of Double Pulse Laser Induced Plasma Spectroscopy under Water. Spectrochim. Acta B 2005, 60, 975–985. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; De Pascale, O; Capitelli, M. From Single Pulse to Double Pulse ns-Laser Induced Breakdown Spectroscopy under Water: Elemental Analysis of Aqueous Solutions and Submerged Solid Samples. Spectrochim. Acta B 2007, 62, 721–738. [Google Scholar]
- Kennedy, PK; Hammer, DX; Rockwell, BA. Laser Induced Plasma in Aqueous Media. Prog. Quantum Electron 1997, 21, 155–248. [Google Scholar]
- Fauchais, P; Coudert, JF; Sardelle, M. Diagnostic in Thermal Plasma. In Plasma Diagnostics; Academic Press: New York, NY, USA, 1989. [Google Scholar]
- Moon, H-Y; Herrera, KK; Omenetto, N; Smith, BW; Winefordner, JD. On the Usefulness of A Duplicating Mirror to Evaluate self-Absorption Effects in Laser Induced Breakdown Spectroscopy. Spectrochim. Acta B 2009, 64, 702–713. [Google Scholar]
- Amamou, H; Bois, A; Ferhat, B; Redon, R; Rossetto, B; Matheron, P. Correction of self-Absorption Spectral Line and Ratios of Transition Probabilities for Homogeneous and LTE Plasma. J. Quant. Spectrosc. Radiat 2002, 75, 747–763. [Google Scholar]
- Bulajic, D; Corsi, M; Cristoforetti, G; Legnaioli, S; Palleschi, V; Salvetti, A; Tognoni, E. A Procedure for Correcting self-Absorption in Calibration Free-Laser Induced Breakdown Spectroscopy. Spectrochim. Acta B 2002, 57, 339–353. [Google Scholar]
- De Giacomo, A. Experimental Characterization of Metallic Titanium-Laser Induced Plasma by Time and Space Resolved Optical Emission Spectroscopy. Spectrochim. Acta B 2003, 58, 71–83. [Google Scholar]
- Koch, J; Von Bohlen, A; Hergenroder, R; Niemax, K. Particle Size Distributions and Compositions of Aerosols Produced by near-IR femto- and Nanosecond Laser Ablation of Brass. J. Anal. At. Spectrom 2004, 19, 267–272. [Google Scholar]
- Borisov, OV; Mao, XL; Russo, RE. Effects of Crater Development on Fractionation and Signal Intensity during Laser Ablation Inductively Coupled Plasma Mass Spectrometry. Spectrochim. Acta B 2000, 55, 1693–1704. [Google Scholar]
- Margetic, V; Pakulev, A; Stockhaus, A; Bolshov, M; Niemax, K; Hergenroder, R. A Comparison of Nanosecond and Femtosecond Laser Induced Plasma Spectroscopy of Brass Samples. Spectrochim. Acta B 2000, 55, 1771–1785. [Google Scholar]
- Tognoni, E; Cristoforetti, G; Legnaioli, S; Palleschi, V. Calibration-Free Laser-induced Breakdown Spectroscopy: State of the Art. Spectrochim. Acta B 2010, 65, 1–14. [Google Scholar]
- Fornarini, L; Fantoni, R; Colao, F; Santagata, A; Teghil, R; Elhassan, A; Harith, M. A Modeling of Laser Ablation of Quaternary Bronze Alloys: Case Studies Comparing Femtosecond and Nanosecond LIBS Experimental Data. J. Phys. Chem. A 2009, 113, 14363–14374. [Google Scholar]
- Aragon, C; Aguilera, JA. Characterization of Laser Induced Plasmas by Optical Emission Spectroscopy: A Review of Experiments and Methods. Spectrochim. Acta B 2008, 63, 893–916. [Google Scholar]
- Capitelli, M; Capitelli, F; Eletskii, A. Non-Equilibrium and Equilibrium Problems in Laser-induced Plasmas. Spectrochim. Acta B 2000, 55, 559–574. [Google Scholar]
- Cristoforetti, G; De Giacomo, A; Dell’Aglio, M; Legnaioli, S; Tognoni, E; Palleschi, V; Omenetto, N. Local Thermodynamic Equilibrium in Laser-induced Breakdown Spectroscopy: beyond the McWhirter Criterion. Spectrochim. Acta B 2010, 65, 86–95. [Google Scholar]
- Sirven, J-B; Bousquet, B; Canioni, L; Sarger, L. Laser-induced Breakdown Spectroscopy of Composite Samples: Comparison of Advanced Chemometrics Methods. Anal. Chem 2006, 78, 1462–1469. [Google Scholar]
- Amador-Hernández, J; García-Ayuso, LE; Fernandez-Romero, JM; Luque de Castro, MD. Partial Least Squares Regression for Problem Solving in Precious Metal Analysis by Laser Induced Breakdown Spectrometry. J. Anal. At. Spectrom 2000, 15, 587–593. [Google Scholar]
- Munson, CA; De Lucia, FC; Piehler, T; McNesby, KL; Miziolek, AW. Investigation of Statistics Strategies for Improving the Discriminating Power of Laser-induced Breakdown Spectroscopy for Chemical and Biological Warfare Agent Atimulants. Spectrochim. Acta B 2005, 60, 1217–1224. [Google Scholar]
- Martin, MZ; Labbe, N; Rials, TG; Wullschlege, SD. Analysis of Preservative-Treated Wood by Multivariate Analysis of Laser-induced Breakdown Spectroscopy Spectra. Spectrochim. Acta B 2005, 60, 1179–1185. [Google Scholar]
- Tripathi, MM; Eseller, KE; Yueh, F-Y; Singh, JP. Multivariate Calibration of Spectra Obtained by Laser Induced Breakdown Spectroscopy of Plutonium Oxide Surrogate Residues. Spectrochim. Acta B 2009, 64, 1212–1218. [Google Scholar]
- Ciucci, A; Corsi, M; Palleschi, V; Rastelli, S; Salvetti, A; Tognoni, E. New Procedure for Quantitative Elemental Analysis by Laser Induced Plasma Spectroscopy. Appl. Spectrosc 1999, 53, 960–964. [Google Scholar]
- Tognoni, E; Cristoforetti, G; Legnaioli, S; Palleschi, V; Salvetti, A; Mueller, M; Panne, U; Gornushkin, I. A Numerical Study of Expected Accuracy and Precision in Calibration-Free Laser-induced Breakdown Spectroscopy in the Assumption of Ideal Analytical Plasma. Spectrochim. Acta B 2007, 62, 1287–1302. [Google Scholar]
- Stuart, BC; Feit, MD; Herman, S; Rubenchik, AM; Shore, BW; Perry, MD. Nanosecond-to-Femtosecond Laser-induced Breakdown in Dielectrics. Phys. Rev. B 1996, 53, 1749–1761. [Google Scholar]
- Nishihara, K; Amitani, H; Murakami, M; Bulanov, SV; Esirkepov, TZ. High Energy Ions Generated by Laser Driven Coulomb Explosion of Cluster. Nucl. Instrum. Methods Phys. Res. Sect. A 2001, 464, 98–102. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; Santagata, A; Teghil, R. Early Stage Emission Spectroscopy Study of Metallic Titanium Plasma Induced in Air by fs- and ns-Laser Pulse. Spectrochim. Acta B 2005, 60, 935–947. [Google Scholar]
- Robbiola, L; Blengino, J-M; Fiaud, C. Morphology and Mechanisms of Formation of Natural Patinas on Archaeological Cu-Sn Alloys. Corros. Sci 1998, 40, 2083–2111. [Google Scholar]
- Oujja, M; Rebollar, E; Castillejo, M; Domingo, C; Cirujano, C; Guerra-Librero, F. Laser Cleaning of Terracotta Decorations of the Portal of Palos of the Cathedral of Seville. J. Cult. Herit 2005, 6, 321–327. [Google Scholar]
- Maravelaki-Kalaitzaki, P; Zafiropulos, V; Pouli, P; Anglos, D; Balas, C; Salimbeni, R; Siano, S; Pini, R. Short Free Running Nd:YAG Laser to Clean Different Encrustations on Pentelic Marble: Procedure and Evaluation of the Effects. J. Cult. Herit 2003, 4, 77–82. [Google Scholar]
- Lazic, V; Fantoni, R; Colao, F; Santagata, A; Morone, A; Spizzichino, V. Quantitative Laser Induced Breakdown Spectroscopy Analysis of Ancient Marbles and Corrections for the Variability of Plasma Parameters and of Ablation Rate. J. Anal. At. Spectrom 2004, 19, 429–436. [Google Scholar]
- Scholten, JH; Teule, JM; Zafiropulos, V; Heeren, RMA. Controlled Laser Cleaning of Painted Artworks using Accurate Beam Manipulation and On-line LIBS-Detection. J. Cult. Herit 2000, 1, 215–220. [Google Scholar]
- Bracco, P; Lanterna, G; Matteini, M; Nakahara, K; Sartiani, O; de Cruz, A; Wolbarsht, ML; Adamkiewicz, E; Colombini, MP. Er:YAG Llaser: An Innovative Tool for Controlled Cleaning of Old Paintings: Testing and Evaluation. J. Cult. Herit 2003, 4, 202–208. [Google Scholar]
- Batishche, S; Englezis, A; Gorovets, T; Kouzmouk, A; Pilipenka, U; Pouli, P; Tatur, H; Totou, G; Ukhau, V. Nd:YAG Laser Double Wavelength Ablation of Pollution Encrustation on Marble and Bonding Glues on Duplicated Painting Canvas. Appl. Surf. Sci 2005, 248, 264–269. [Google Scholar]
- Pouli, P; Nevin, A; Andreotti, A; Colombini, P; Georgiou, S; Fotakis, C. Laser Assisted Removal of Synthetic Painting-Conservation Materials using UV Radiation of ns and fs Pulse Duration: Morphological Studies on Model Samples. Appl. Surf. Sci 2009, 255, 4955–4960. [Google Scholar]
- Teule, R; Scholten, H; van den Brink, OF; Heeren, RMA; Zafiropulos, V; Hesterman, R; Castillejo, M; Martin, M; Ullenius, U; Larsson, I; Guerra-Librero, F; Silva, A; Gouveia, H; Albuquerque, M-B. Controlled UV Laser Cleaning of Painted Artworks: A Systematic Effect Study on Egg Tempera Paint Samples. J. Cult. Herit 2003, 4, 209–215. [Google Scholar]
- Colao, F; Fantoni, R; Lazic, V; Caneve, L; Giardini, A; Spizzichino, V. LIBS as A Diagnostic Tool during the Laser Cleaning of Copper based Alloys: Experimental Results. J. Anal. At. Spectrom 2004, 19, 502–504. [Google Scholar]
- Fortes, FJ; Cabalín, LM; Laserna, JJ. The Potential of Laser-induced Breakdown Spectrometry for Real Time Monitoring the Laser Cleaning of Archaeometallurgical Objects. Spectrochim. Acta B 2008, 63, 1191–1197. [Google Scholar]
- Ochocińska, K; Kamińska, A; Sliwinski, G. Experimental Investigations of Stained Paper Documents Cleaned by the Nd:YAG Laser Pulses. J. Cult. Herit 2003, 4, 188–193. [Google Scholar]
- Kaminska, A; Sawczak, M; Komar, K; Sliwinski, G. Application of the Laser Ablation for Conservation of Historical Paper Documents. Appl. Surf. Sci 2007, 253, 7860–7864. [Google Scholar]
- Gronlund, R; Lunqvist, M; Svanberg, S. Remote Imaging Laser-induced Breakdown Spectroscopy and Remote Cultural Heritage Ablative Cleaning. Opt. Lett 2005, 30, 2882–2884. [Google Scholar]
- Vazquez-Calvo, C; Giakoumaki, A; Anglos, D; Alvarez de Buergo, M; Fort, R. Classification of Patinas Found on Surfaces of Historical Buildings by Means of Laser-induced Breakdown Spectroscopy. In Lasers in the Conservation of Artworks; Springer: Berlin, Germany, 2007; Volume 116, pp. 415–420. [Google Scholar]
- Lopez, AJ; Nicolas, G; Mateo, MP; Pinon, V; Tobar, MJ; Ramil, A. Compositional Analysis of Hispanic Terra Sigillata by Laser-induced Breakdown Spectroscopy. Spectrochim. Acta B 2005, 60, 1149–1154. [Google Scholar]
- Lopez, AJ; Nicolas, G; Mateo, MP; Ramil, A; Pinon, V; Yanez, A. LIPS and Linear Correlation Analysis Applied to the Classification of Roman Pottery Terra Sigillata. Appl. Phys. A 2006, 83, 695–698. [Google Scholar]
- Colao, F; Fantoni, R; Lazic, V; Spizzichino, V. Laser-induced Breakdown Spectroscopy for semi-Quantitative and Quantitative Analyses of Artworks—Application on multi-Layered Ceramics and Copper based Alloys. Spectrochim. Acta B 2002, 57, 1219–1234. [Google Scholar]
- Osticioli, I; Mendes, NFC; Porcinai, S; Cagnini, A; Castellucci, E. Spectroscopic Analysis of Works of Art using A Single LIBS and Pulsed Raman Setup. Anal. Bioanal. Chem 2009, 394, 1033–1041. [Google Scholar]
- Burgio, L; Melessanaki, K; Doulgeridis, M; Clark, RJH; Anglos, D. Pigment Identification in Paintings Employing Laser Induced Breakdown Spectroscopy and Raman Microscopy. Spectrochim. Acta B 2001, 56, 905–913. [Google Scholar]
- Clark, RJH. Pigment Identification by Spectroscopic Means: An Arts/Science Interface. C. R. Chim 2002, 5, 7–20. [Google Scholar]
- Brysbaert, A; Melessanaki, K; Anglos, D. Pigment Analysis in Bronze Age Aegean and Eastern Mediterranean Painted Plaster by Laser-induced Breakdown Spectroscopy (LIBS). J. Archaeol. Sci 2006, 33, 1095–1104. [Google Scholar]
- Castillejo, M; Martin, M; Silva, D; Stratoudaki, T; Anglos, D; Burgio, L; Clark, RJH. Analysis of Pigments in Polychromes by Use of Laser Induced Breakdown Spectroscopy and Raman Microscopy. J. Mol. Struct 2000, 550, 191–198. [Google Scholar]
- Oujja, M; Vila, A; Rebollar, E; Garcia, JF; Castillejo, M. Identification of Inks and Structural Characterization of Contemporary Artistic Prints by Laser-induced Breakdown Spectroscopy. Spectrochim. Acta B 2005, 60, 1140–1148. [Google Scholar]
- Bruder, R; L’Hermite, D; Semerok, A; Salmon, L; Detalle, V. Near-Crater Discoloration of White Lead in Wall Paintings During Laser Induced Breakdown Spectroscopy Analysis. Spectrochim. Acta B 2007, 62, 1590–1596. [Google Scholar]
- Bruder, R; Detalle, V; Coupry, C. An Example of the Complementarity of Laser-induced Breakdown Spectroscopy and Raman Microscopy for Wall Painting Pigments Analysis. J. Raman Spectrosc 2007, 38, 909–915. [Google Scholar]
- Senesi, GS; Baldassarre, G; Senesi, N; Radina, B. Trace Elements Inputs into Soils by Anthropogenic Activities and Implications for Human Health. Chemosphere 1999, 39, 343–377. [Google Scholar]
- Harmon, RS; Remus, J; McMillan, NJ; McManus, C; Collins, L; Gottfried, JL, Jr; DeLucia, FC; Miziolek, AW. LIBS Analysis of Geomaterials: Geochemical Fingerprinting for the Rapid Analysis and Discrimination of Minerals. Appl. Geochem 2009, 24, 1125–1141. [Google Scholar]
- G.U. 285, DM 99/248 Approvazione dei Metodi ufficiali di analisi chimica del suolo. Italian Ministry of Agriculture, Food and Forest Affairs, Number 20; Rome, Italy..
- Website of Ministry of Agriculture, Food and Rural Affairs of Ontario. Available online: http://www.omafra.gov.on.ca/english/nm/regs/sampro/sampro04_09.htm/ (accessed on 4 February 2010)..
- Steinnes, E. Soils and Geomedicine. Environ. Geochem. Health 2009, 31, 523–535. [Google Scholar]
- Zaccone, C; Cocozza, C; Cheburkin, AK; Shotyk, W; Miano, TM. Enrichment and Depletion of Major and Trace Elements, and Radionuclides in Ombrotrophic Raw Peat and Corresponding Humic Acids. Geoderma 2007, 141, 235–246. [Google Scholar]
- Hussain, T; Gondal, MA. Monitoring and Assessment of Toxic Metals in Gulf War Oil Spill Contaminated Soil using Laser-induced Breakdown Spectroscopy. Environ. Monit. Assess 2008, 136, 391–399. [Google Scholar]
- Santos, D, Jr; Nunes, LC; Trevizan, LC; Godoi, Q; Leme, FO; Braga, JWB; Krug, FJ. Evaluation of Laser Induced Breakdown Spectroscopy for Cadmium Determination in Soils. Spectrochim. Acta B 2009, 64, 1073–1078. [Google Scholar]
- Bousquet, B; Sirven, J-B; Canioni, L. Towards Quantitative Laser-induced Breakdown Spectroscopy Analysis of Soil Samples. Spectrochim. Acta B 2007, 62, 1582–1589. [Google Scholar]
- Decree of the Italian Ministry of Environment. Number 471; Rome, Italy, 25 November 1999..
- Kwak, J; Lenth, C; Salb, C; Ko, E-J; Kim, K-W; Park, K. Quantitative Analysis of Arsenic in Mine Tailing Soils using Double Pulse-Laser Induced Breakdown Spectroscopy. Spectrochim. Acta B 2009, 64, 1105–1110. [Google Scholar]
- Burakov, VS; Tarasenko, NV; Nedelko, MI; Kononov, VA; Vasilev, NN; Isakov, SN. Analysis of Lead and Sulfur in Environmental Samples by Double Pulse Laser Induced Breakdown Spectroscopy. Spectrochim. Acta B 2009, 64, 141–146. [Google Scholar]
- Coelho Pontes, MJ; Cortez, J; Harrop Galvao, RK; Pasquini, C; Araujo, MCU; Marques Coelho, R; Chiba, MK; Ferreira de Abreu, M; Emoke Madari, B. Classification of Brazilian Soils by using LIBS and Variable Selection in the Wavelet Domain. Anal. Chim. Acta 2009, 642, 12–18. [Google Scholar]
- Hassan, M; Sighicelli, M; Lai, A; Colao, F; Hanafy Ahmed, AH; Fantoni, R; Harith, MA. Studying the Enhanced Phytoremediation of Lead Contaminated Soils via Laser Induced Breakdown Spectroscopy. Spectrochim. Acta B 2008, 63, 1225–1229. [Google Scholar]
- Galiova, M; Kaiser, J; Novotny, K; Samek, O; Reale, L; Malina, R; Palenikova, K; Liska, M; Cudek, V; Kanicky, V; Otruba, V; Poma, A; Tucci, A. Utilization of Laser Induced Breakdown Spectroscopy for Investigation of the Metal Accumulation in Vegetal Tissues. Spectrochim. Acta B 2007, 62, 1597–1605. [Google Scholar]
- Kaiser, J; Galiova, M; Novotny, K; Cervenka, R; Reale, L; Novotny, J; Liska, M; Samek, O; Kanicky, V; Hrdlicka, A; Stejskal, K; Adam, VM; Kizek, R. Mapping of Lead, Magnesium and Copper Accumulation in Plant Tissues by Laser-induced Breakdown Spectroscopy and Laser-Ablation Inductively Coupled Plasma Mass Spectrometry. Spectrochim. Acta B 2009, 64, 67–73. [Google Scholar]
- Hussain, T; Gondal, MA; Yamani, ZH; Baig, MA. Measurement of Nutrients in Green House Soil with Laser Induced Breakdown Spectroscopy. Environ. Monit. Assess 2007, 124, 131–139. [Google Scholar]
- Cremers, DA; Ebinger, MH; Breshears, DD; Unkefer, PJ; Kammerdiener, SAM; Ferris, J; Catlett, KM; Brown, JR. Measuring Total Soil Carbon with Laser-induced Breakdown Spectroscopy (LIBS). J. Environ. Qual 2001, 30, 2202–2206. [Google Scholar]
- Martin, MZ; Wullschleger, SD; Garten, CT, Jr; Palumbo, AV. Laser-induced Breakdown Spectroscopy for the Environmental Determination of Total Carbon and Nitrogen in Soils. Appl. Opt 2003, 42, 2072–2077. [Google Scholar]
- da Silva, RM; Milori, DMBP; Ferreira, EC; Ferreira, EJ; Krug, FJ; Martin-Neto, L. Total Carbon Measurement in Whole Tropical Soil Sample. Spectrochim. Acta B 2008, 63, 1221–1224. [Google Scholar]
- Ferreira, EC; Milori, DMBP; Ferreira, EJ; Da Silva, RM; Martin-Neto, L. Artificial Neural Network for Cu Quantitative Determination in Soil using A Portable Laser Induced Breakdown Spectroscopy System. Spectrochim. Acta B 2008, 63, 1216–1220. [Google Scholar]
- Corsi, M; Cristoforetti, G; Hidalgo, M; Legnaioli, S; Palleschi, V; Salvetti, A; Tognoni, E; Vallebona, C. Double Pulse, Calibration-Free Laser-induced Breakdown Spectroscopy: A New Technique for in situ Standard-Less Analysis of Polluted Soils. Appl. Geochem 2006, 21, 748–755. [Google Scholar]
- De Giacomo, A; Dell’Aglio, M; Gaudiuso, R; Cristoforetti, G; Legnaioli, S; Palleschi, V; Tognoni, E. Spatial Distribution of Hydrogen and Other Emitters in Aluminum Laser-induced Plasma in Air and Consequences on Spatially Integrated Laser-induced Breakdown Spectroscopy Measurements. Spectrochim. Acta B 2008, 63, 980–987. [Google Scholar]
- Wainner, RT; Harmon, RS; Miziolek, AW; McNesby, KL; French, PD. Analysis of Environmental Lead Contamination: Comparison of LIBS Field and Laboratory Instruments. Spectrochim. Acta B 2001, 56, 777–793. [Google Scholar]
- Lopez-Moreno, C; Palanco, S; Laserna, JJ. Remote Laser-induced Plasma Spectrometry for Elemental Analysis of Samples of Environmental Interest. J. Anal. At. Spectrom 2004, 19, 1479–1484. [Google Scholar]
- Palanco, S; Lopez-Moreno, C; Laserna, JJ. Design, Construction and Assessment of A Field-Deployable Laser-induced Breakdown Spectrometer for Remote Elemental Sensing. Spectrochim. Acta B 2006, 61, 88–95. [Google Scholar]
- Fichet, P; Tabarant, M; Salle, B; Gautier, C. Comparisons between LIBS and ICP/OES. Anal. Bioanal. Chem 2006, 385, 338–344. [Google Scholar]
- Koch, S; Garen, W; Müller, M; Neu, W. Detection of Chromium in Liquids by Laser Induced Breakdown Spectroscopy (LIBS). Appl. Phys. A 2004, 79, 1071–1073. [Google Scholar]
- Hussain, T; Gondal, MA. Detection of Toxic Metals in Waste Water from Dairy Products Plant using Laser Induced Breakdown Spectroscopy. Bull. Environ. Contam. Toxicol 2008, 80, 561–565. [Google Scholar]
- Rai, NK; Rai, AK. LIBS-An Efficient Approach for the Determination of Cr in Industrial Wastewater. J. Hazard. Mater 2008, 150, 835–838. [Google Scholar]
- Ho, WF; Ng, CW; Cheung, NH. Spectrochemical Analysis of Liquids using Laser-induced Plasma Emissions: Effects of Laser Wavelength. Appl. Spectrosc 1997, 51, 87–91. [Google Scholar]
- Lo, KM; Cheung, NH. ArF Laser-Induced Plasma Spectroscopy for Part-per-Billion Analysis of Metal Ions in Aqueous Solutions. Appl. Spectrosc 2002, 56, 682–688. [Google Scholar]
- Michel, APM; Chave, AD. Double Pulse Laser-induced Breakdown Spectroscopy of Bulk Aqueous Solutions at Oceanic Pressures: Interrelationship of Gate Delay, Pulse Energies, Interpulse Delay, and Pressure. Appl. Opt 2008, 47, 131–143. [Google Scholar]
- MSL Science Corner Website. Available online: http://msl-scicorner.jpl.nasa.gov/ (accessed on 4 February 2010)..
- ESA-ExoMars Website. Available online: http://www.esa.int/esaMI/ExoMars/index.html/ (accessed on 4 February 2010)..
- Borovicka, J; Betlem, H. Spectral Analysis of Two Perseid Meteors. Planet. Space Sci 1997, 45, 563–575. [Google Scholar]
- Borovicka, J; Jenniskens, P. Time Resolved Spectroscopy of A Leonid Fireball Afterglow. Earth Moon Planet 2000, 82, 399–428. [Google Scholar]
- Trigo-Rodriguez, JM; Llorca, J; Borovicka, J; Fabregat, J. Chemical Abundances Determined from Meteor Spectra-I, Ratio of the Main Chemical Elements. Meteorit. Planet. Sci 2003, 38, 1283–1294. [Google Scholar]
- Milley, EP; Hawkes, RL; Ehrman, JM. Meteor Luminosity Simulation Through Laser Ablation of Meteorite. Mon. Not. R. Astron. Soc 2007, 382, 67–71. [Google Scholar]
- Hawkes, RL; Milley, EP; Ehrman, JM; Woods, RM; Hoyland, JD; Pettipas, CL; Tokaryk, DW. What Can We Learn about Atmospheric Meteor Ablation and Light Production from Laser Ablation. Earth Moon Planet 2008, 102, 331–336. [Google Scholar]
- Norton, OR. The Cambridge Encyclopedia of Meteorites; Cambridge University Press: Cambridge, UK, 2002. [Google Scholar]
- Thompson, JR; Wiens, RC; Barefield, JE; Vaniman, DT; Newsom, HE; Clegg, SM. Remote Laser Induced Breakdown Spectroscopy (LIBS) Analyses of DaG 476 and Zagami Martian Meteorites. J Geophys Res 2006, 111, 6:1–6:9. [Google Scholar]
- Clegg, SM; Sklute, EC; Dyar, MD; Barefield, JE; Wiens, RC. Quantitative Analysis of Samples with Variable Composition by Remote Laser-induced Breakdown Spectroscopy. Proceedings of the Seventh International Conference on Mars, Pasadena, CA, USA, 9–13 July 2007.
- Dell’Aglio, M; De Giacomo, A; Gaudiuso, R; De Pascale, O; Senesi, GS; Longo, S. LIBS Applications to Meteorites: Chemical Analysis and Composition Profiles. Geochim Cosmochim Acta. in press..
- Gaudiuso, R; D’Alessandro, C; De Giacomo, A; Dell’Aglio, M; Longo, S. Laser Induced Breakdown Spectroscopy for the Elemental Analysis and Classification of Meteorites. Proceedings of the EMSLIBS 2007, Paris, France, 10–13 September 2007.
- Colao, F; Fantoni, R; Lazic, V; Paolini, A; Fabbri, F; Ori, GG; Marinangeli, L; Baliva, A. Investigation of LIBS Feasibility for in situ Planetary Exploration: An Analysis on Martian Rock Analogues. Planet. Space Sci 2004, 52, 117–123. [Google Scholar]
- Salle, B; Lacour, J-L; Vors, E; Fichet, P; Maurice, S; Cremers, DA; Wiens, RC. Laser-induced Breakdown Spectroscopy for Mars Surface Analysis: Capabilities at Stand-off Distances and Detection of Chlorine and Sulfur Elements. Spectrochim. Acta B 2004, 59, 1413–1422. [Google Scholar]
- Radziemski, L; Cremers, DA; Benelli, K; Khoo, C; Harris, RD. Use of the Vacuum Ultraviolet Spectral Region for Laser-induced Breakdown Spectroscopy-based Martian Geology and Exploration. Spectrochim. Acta B 2005, 60, 237–248. [Google Scholar]
- Salle, B; Lacour, J-L; Mauchien, P; Fichet, P; Maurice, S; Manhes, G. Comparative Study of Different Methodologies for Quantitative Rock Analysis by Laser-induced Breakdown Spectroscopy in a Simulated Martian Atmosphere. Spectrochim. Acta B 2006, 61, 301–313. [Google Scholar]
- Arp, ZA; Cremers, DA; Wiens, RC; Wayne, DM; Salle, B; Maurice, S. Analysis of Water/Ice and Water Ice/Soil Mixtures using Laser Induced Breakdown Spectroscopy: Application to Mars Polar Exploration. Appl. Spectrosc 2004, 58, 897–909. [Google Scholar]
- Rauschenbach, I; Lazic, V; Pavlov, SG; Hubers, H-W; Jessberger, EK. Laser Induced Breakdown Spectroscopy on Soils and Rocks: Influence of the Sample Temperature, Moisture and Roughness. Spectrochim. Acta B 2008, 63, 1205–1215. [Google Scholar]
- Clegg, SM; Barefield, JE; Wiens, RC; Quick, CR; Sharma, SK; Misra, AK; Dyar, MD; McCanta, MC; Elkins-Tanton, L. Venus Geochemical Analysis by Remote Raman-Laser Induced Breakdwon Spectroscopy (Raman-LIBS). Proceedings of Venus Geochemistry: Progress Prospects, and New Missions, Houston, TX, USA, 27–29 February 2009.
- Lambert, J; Morookian, J; Roberts, T; Polk, J; Smrekar, S; Clegg, SM; Wiens, RC; Dyar, MD; Treiman, A. Standoff LIBS and Raman Spectroscopy under Venus Conditions. Proceedings of the 41st Lunar and Planetary Science Conference, The Woodlands, TX, USA, 1–5 March 2010.
|Table 1. Comparison between the CF-LIBS results for three bronze standards upon irradiation by ns- and fs-lasers. The experimental uncertainty was estimated to range between 10–15%.|
|Sample||Certified w%||fs-CF-LIBS w%||ns-CF-LIBS w%|
|B4||Cu 84.00||Cu −||Cu −|
|Sn 11.05||Sn 9.32||Sn 9.70|
|Pb 2.50||Pb 3.99||Pb 4.11|
|Ni 0.57||Ni 0.81||Ni 0.31|
|B21||Cu 83.05||Cu −||Cu −|
|Sn 5.13||Sn 5.73||Sn 4.02|
|Pb 3.79||Pb 3.30||Pb 4.46|
|Ni 1.21||Ni 1.10||Ni 1.65|
|B22||Cu 82.75||Cu −||Cu −|
|Sn 3.85||Sn 4.15||Sn 3.25|
|Pb 6.12||Pb 6.76||Pb 6.38|
|Ni 2.56||Ni 1.62||Ni 2.90|
|Table 2. Comparison between concentrations of heavy metals in soil samples determined by ICP-OES and by LIBS (calibration line method).|
|Element||Unknown sample||ICP-OES concentration (mg kg−1)||LIBS concentration (mg kg−1)|
|Cr||S3||550 ± 82||698 ± 144|
|Cu||S4||100 ± 15||88 ± 21|
|Zn||S2||897 ± 135||846 ± 37|
|Pb||S1||180 ± 27||168 ± 18|
|V||S1||84 ± 12||80 ± 7|
© 2010 by the authors; licensee MDPI, Basel, Switzerland. This article is an open access article distributed under the terms and conditions of the Creative Commons Attribution license (http://creativecommons.org/licenses/by/3.0/).