- freely available
Manganese(III) Porphyrin-based Potentiometric Sensors for Diclofenac Assay in Pharmaceutical Preparations
AbstractTwo manganese(III) porphyrins: manganese(III) tetraphenylporphyrin chloride and manganese(III)-tetrakis(3-hydroxyphenyl)porphyrin chloride were tested as ionophores for the construction of new diclofenac−selective electrodes. The electroactive material was incorporated either in PVC or a sol−gel matrix. The effect of different plasticizers and additives (anionic and cationic) on the potentiometric response was studied. The best results were obtained for the PVC membrane plasticized with dioctylphtalate and having sodium tetraphenylborate as a lipophilic anionic additive incorporated. The sensor response was linear in the concentration range 3 × 10−6 – 1 × 10−2 M with a slope of −59.7 mV/dec diclofenac, a detection limit of 1.5 × 10−6 M and very good selectivity coefficients. It was used for the determination of diclofenac in pharmaceutical preparations, by direct potentiometry. The results were compared with those obtained by the HPLC reference method and a good agreement was found between the two methods.
Share & Cite This Article
Vlascici, D.; Pruneanu, S.; Olenic, L.; Pogacean, F.; Ostafe, V.; Chiriac, V.; Pica, E.M.; Bolundut, L.C.; Nica, L.; Fagadar-Cosma, E. Manganese(III) Porphyrin-based Potentiometric Sensors for Diclofenac Assay in Pharmaceutical Preparations. Sensors 2010, 10, 8850-8864.View more citation formats
Vlascici D, Pruneanu S, Olenic L, Pogacean F, Ostafe V, Chiriac V, Pica EM, Bolundut LC, Nica L, Fagadar-Cosma E. Manganese(III) Porphyrin-based Potentiometric Sensors for Diclofenac Assay in Pharmaceutical Preparations. Sensors. 2010; 10(10):8850-8864.Chicago/Turabian Style
Vlascici, Dana; Pruneanu, Stela; Olenic, Liliana; Pogacean, Florina; Ostafe, Vasile; Chiriac, Vlad; Pica, Elena Maria; Bolundut, Liviu Calin; Nica, Luminita; Fagadar-Cosma, Eugenia. 2010. "Manganese(III) Porphyrin-based Potentiometric Sensors for Diclofenac Assay in Pharmaceutical Preparations." Sensors 10, no. 10: 8850-8864.
Notes: Multiple requests from the same IP address are counted as one view.