Open AccessThis article is
- freely available
Design and Development of Low Cost, Simple, Rapid and Safe, Modified Field Kits for the Visual Detection and Determination of Arsenic in Drinking Water Samples
Centre for Environment, Institute of Science &Technology, Jawaharlal Nehru Technological University, Kukatpally, Hyderabad-500072, India
* Author to whom correspondence should be addressed.
Received: 10 January 2005; Accepted: 10 April 2005 / Published: 14 August 2005
Abstract: Arsenic is naturally found in surface and ground waters and the inorganic forms of arsenic are the most toxic forms. The adverse health effects of arsenic may involve the respiratory, gastrointestinal, cardiovascular, nervous, and haematopoietic systems. Arsenic contamination in drinking water is a global problem widely seen in Bangladesh and West Bengal of the Indian sub continent. As there is a great demand for field test kits due to the anticipated reduction of the US EPA arsenic standard from 50ppb to 10ppb a field kit which offers rapid, simple and safe method for precise estimation of arsenic at 10ppb in drinking water samples is developed. Field methods, based on the mercuric-bromide-stain, consist of three different major parts, which are carried out stepwise. The first part of the procedure is to remove serious interference caused by hydrogen sulphide. In commercially available kits either the sulphide is oxidized to sulphate and the excess oxidizing reagent removed prior to the hydride generation step or, the hydrogen sulphide is filtered out by passing the gas stream through a filter impregnated with lead acetate during the hydride generation step. The present method employs cupric chloride in combination with ferric chloride or Fenton’s reagent for the removal of hydrogen sulphide, which is rapid, simple and more efficient. Other interferences at this step of the analyses are normally not expected for drinking water analysis. In the second step, the generation of the arsine gas involves the classical way of using zinc metal and hydrochloric acid, which produce the ‘nascent’ hydrogen, which is the actual reducing agent. Hydrochloric acid can be replaced by sulfamic acid, which is solid and avoids a major disadvantage of having to handle a corrosive liquid in the field. The arsine gas produces a yellowish spot on the reagent paper. Depending on the arsenic content, either, Yellow – H (HgBr) 2 As (10-50ppb), Brown – (HgBr) 3 As (50-100ppb) or Black – Hg3 As2 (>100ppb) are formed which can be precisely estimated by visual comparison with standard color chart. The results obtained by field kits agree well with the data obtained through I.C.P.AES methods. The most important characteristic for field measurement is that analytical results can be obtained on the site where the sample is taken with high precision and can be conveniently utilized for monitoring arsenic rapidly in a highly contaminated large geographical area.
Keywords: arsenic; field kit; and drinking water
Article StatisticsClick here to load and display the download statistics.
Notes: Multiple requests from the same IP address are counted as one view.
Cite This Article
MDPI and ACS Style
Cherukuri, J.; Anjaneyulu, Y. Design and Development of Low Cost, Simple, Rapid and Safe, Modified Field Kits for the Visual Detection and Determination of Arsenic in Drinking Water Samples. Int. J. Environ. Res. Public Health 2005, 2, 322-327.
Cherukuri J, Anjaneyulu Y. Design and Development of Low Cost, Simple, Rapid and Safe, Modified Field Kits for the Visual Detection and Determination of Arsenic in Drinking Water Samples. International Journal of Environmental Research and Public Health. 2005; 2(2):322-327.
Cherukuri, Jyotsna; Anjaneyulu, Y. 2005. "Design and Development of Low Cost, Simple, Rapid and Safe, Modified Field Kits for the Visual Detection and Determination of Arsenic in Drinking Water Samples." Int. J. Environ. Res. Public Health 2, no. 2: 322-327.