- freely available
Sensors 2007, 7(7), 1216-1223; doi:10.3390/s7071216
Abstract: For the surface-optoelectronic study of Saxitoxin sensing, we fabricated self-assembled calixarene derivative monolayers as the recognition-functional interfaces on a gold surface. An interaction study between Saxitoxin and calixarene derivative monolayers were performed using surface plasmon resonance (SPR) spectroscopy. Among three calixarene derivatives, calixarene crown ether SAM showed the highest sensitivity to Saxitoxin. The detection limit of this system is three orders of magnitude lower than that of the mouse bioassay which is the current benchmark for Saxitoxin detection.
Harmful algal booms (red tides) produce a wide variety of secondary metabolites, but Saxitoxin (STX) is virtually isolated among them in being capable of causing human mortality by consumption of tainted shellfish. STX and its congeners are known as paralytic shellfish poisons (PSPs). Currently, governments of many countries monitor shellfish beds for the presence of saxitoxin by using one of several tests, the most reliable test is mouse bioassay. New approaches include insect bioassay, tissue biosensors, molecular pharmacology, neurophysiology, whole-cell bioassay, HPLC with postcolumn oxidation of the C4-C12 bond [1-4]. However, most methods have low sensitivity and many pre-required treatment steps for analysis.
STX is a bis-guanidinium dication and its molecular structure was shown in Figure 1. Several crystal structures of 18-crown-6 and its aza analogues with guanidinium ions show hydrogen bonding with crown heteroatoms. R.E. Gawley et al. fabricated Aza-18-crown-6 ether sensor that showed excellent fluorescence enhancement . calixarene and calixarene derivatives were found for their guanidinium ion selectivity based on CHEMFETs . However, to our knowledge, there has no report about STX sensing using calix[n]arene derivative receptor yet.
Surface plasmon resonance (SPR) allows the user to study the interaction between immobilized recognition molecular and analytes in solution without labeling of the analyte. Recently, a novel method based on SPR, immobilization by recognition monolayers on noble metals have been utilized to detect small molecules and to obtain better reproducibility [7, 8]. These approaches of small molecule measurement have been proved to increase the sensitivity compared to other techniques. The SPR technique and careful construction of appropriate SAM are very useful methods to the development of more efficient biosensing interface.
Our research purpose here is a SPR based fundamental study of STX binding properties on the calixarene derivative recognition monolayer system in order to make high sensitive detection of STX. For this purpose, three calixarene derivatives containing ethylester (1), carboxylic acid (2), and crownether (3) at the lower rim were applied to recognition layer for STX. Self-assembly technique was utilized to construct a well–characterized recognition monolayer. The characterization of calixarene SAMs has been showed detailly in our previous work . A molecular interaction study between calixarene derivative SAMs and STX was investigated by SPR to detect low concentration up to 1.0 × 10-9 M on 3 SAM. The refractive index of the surface interfacial layer was calculated by the fitting of SPR curve with the Fresnel equations, which was found increased derived by the binding of STX onto the 3 SAM.
2.1. Chemicals and Reagents
STX was bought from National Research Council in Canada. The syntheses of calixarene derivative 1 (ethylester) and 2 (carboxylic acid) were carried out from the literature with minor modification . calixarene crown ether derivative 3 was obtained from Proteogen Co. (Seoul, Korea). The structure of calixarene derivatives was shown in Figure 2. Other reagents were bought from Sigma Chemicals (St. Louis, Mo) and Aldrich Chemical (Milwaukee, WI, USA). Milli-Q grade (> 18.2 mQ / cm) water was utilized for the preparation of the buffer solution.
2.2 Formation of calixarene derivative SAMs on a gold chip
A microscope cover glass (18 × 18 × 0.15 mm, with a refractive index of 1.515, Matsunami, Japan) with gold layer was used as a substrate for the formation of calixarene derivative SAMs. The gold film (thickness ≈ 50 nm) was deposited on the cover glass by the sputter coating system (E5000, Polaron Co., U.K.) under conditions of 2.0×10-2 mbar and 20 mA for 135 s. The sputtered Au substrate was rinsed using distilled water, methanol and acetone, sequentially. Then, the gold chip was dried in a nitrogen stream softly and ready to use. calixarene derivative 1 and 3 were prepared to 0.1 mM in chloroform:methanol = 1:9 (v/v) solution mixture for the formation of their SAMs on gold surface. The SAM was formed by immersing the gold chips into solution for 6 h. After the immobilization process, the sensor chip was rinsed with chloroform-methanol solution mixture and methanol and then dried under N2 stream .
The monolayer of calixarene derivative 2 was converted from 1 SAM through the hydrolysis of ethylene group by treatment of LiOH for 3 h (Figure 2) . LiOH solution was prepared to 0.1 M in methanol:DW = 1:1 (v/v) mixture.
2.3 SPR spectroscopic measurement
The SPR spectroscopic measurements for the interactions between STX and the SAM were performed by homemade SPR system based on the traditional Kretschmann configuration (Figure 3) . A laser diode (LD, λmax = 675 nm) was used as the light source. The reflected intensity of light through the polarizer and the prism was measured with the photodiode detector (ANDO Electric Co. Ltd., AQ-1976, Kanagawa, Japan). The incident angle into the prism varied with the motorized rotary stage and its controller (Suruga Seiki, D80, Shizuoka, Japan). The signal from photodiode was converted through a signal process board (K-MAC Co., Spectra View 2000, Taejeon, Korea) and could be interfaced with a computer. The angle resolution of the SPR system that was determined by the resolution of motorized rotary stage was 0.004 degree.
Molecular interaction between the calixarene derivatives and the STX was measured by the batch method in a Teflon sample chamber. SPR spectra were recorded at concentration of STX ranging from 1.0 × 10 -10 to 1.0 × 10-5 M (pH =7.4, 0.01 M phosphate buffer, containing 0.15M NaCl).
3. Results and discussion
To compare binding properties, STX was injected onto different calixarene derivative SAMs modified chip surface. In case of calixarene 3 SAM, as the concentration of STX increased, the SPR angle shifts that resulted from molecular interaction between the calixarene 3 SAM and the STX gradually increased (Figure 4 (a)). However, in case of 1 and 2, no response was shown to STX, which indicates they have weak or no interactions with STX (Figure 4 (b)).
The calixarene binds with STX through the π–π and van der Waals interactions for all three derivatives. The upper rims of 1 (ethylester) and 2 (carboxylic acid) have very small binding affinity with STX, which results little refractive index change of interfacial layer. However, 3 SAM bound to STX may involve in the strong hydrogen-bonding of the crown-like loop . It was well known that calixarene crown ether binding to cations and completely encapsulated the cation . Gawley et al. presented the possible mode of binding between crown ether and STX, which shows the several hydrogen bonds between the guandinium of STX and crown ether oxygens.
SPR is a collective electron excitation that may exist at the interface of two media with different dielectric constants. Since the dipole moment of a molecule was a measuring factor of charge distribution of the overall molecule, the conformation of calixarene crown ether ionic complex with STX was able to affect its dipole moments. The dipole moment is intimately related to the dielectric constant κ through
The dipole moments of STX binding calixarene crown ether SAM derived variation of dielectric constant, consequently, resulting in distinguishable of RI change and a greater SPR angle shifts (eq. 1 and 2). To further study the RI change on the sensor surface, the RI was determined through four layer theoretical simulation by the Fresnel equation with experimental SPR data. We reported in our previous work on our SPR-based RI calculation method . Table 1 presents the RI changes via computer simulation when STX was bound on the 3 SAM. It was observed that the RI was linearly increasing by the gradual change of STX concentration. However, 1 and 2 SAM showed no RI change even when presenting the 1.0 × 10-5 M of STX (data not shown).
The SPR angle change for the increase in STX concentration from zero (buffer) to 1 × 10-5 M is 0.22 degree. The linear detection range using calixarene crown ether is found to be 1.0 × 10-9 - 1.0 × 10-5 M with r2 = 0.994 (Figure 5). With regard to the low detection limit of conventional mouse bioassay and chemosensor, which stays at the micromolar level, these experimental results are quite remarkable [1, 5].
We have constructed a calixarene derivative monolayers using a SAM method. To make the high sensitivity of STX detection, three different calixarene derivative monolayers were applied to a STX recognition system coupled with SPR spectroscopy. Among three derivatives, calixarene crown ether SAM has been shown to bind to STX, even at very low STX concentration. In binding, the RI change of the interfacial recognition layer induces the SPR angle shift, permitting a sensitive detection of STX at a range of 1.0 × 10-9 - 1.0 × 10-5 M. This RI increase may result from the conformation change of calixarene induced by the guanidinium ion bond on the low rim of crown ether. Consequently, these results reveal that a well-designed molecular recognition system based on SPR spectroscopy is very useful for the study of small molecular interaction. Studies are underway to elucidate the mechanism of binding, and to design a superior sensor.
This work was supported for two years by Pusan National University Research Grant and partially by the Brain Korea 21 project in 2007.
- McElhiney, J.; Lawton, L. A.; Edwards, C.; Gallacher, S. Development of a bioassay employing the desert locust (Schistocerca gregaria) for the detection of saxitoxin and related compounds in cyanobacteria and shellfish. Toxicon 1998, 36, 417–420. [Google Scholar]
- Negri, A.; Llewellyn, L. Comparative analyses by HPLC and the sodium channel and saxiphilin 3H-saxitoxin receptor assays for paralytic shellfish toxins in crustaceans and molluscs from tropical North West Australia. Toxicon 1998, 36, 283–298. [Google Scholar]
- Lawrence, J. F.; Wong, B. Development of a manganese dioxide solid-phase reactor for oxidation of toxins associated with paralytic shellfish poisoning. J. Chromatogr. A 1996, 755, 227–233. [Google Scholar]
- Mirocha, C. J.; Cheong, W.; Mirza, U.; Kim, Y. B. Analysis of saxitixin in urine by continuous-flow fast-atom bombardment mass spectrometry. Rapid Commun. Mass Spectrom. 1992, 6, 128–134. [Google Scholar]
- Gawley, R. E.; Pinet, S.; Cardona, C.; Datta, P.; Ren, T.; Guida, W.; Nydick, J.; Leblanc, R. M. Chemosensors for the marine toxin saxitoxin. J. Am. Chem. Soc. 2002, 124, 13448–13453. [Google Scholar]Kele, P.; Orbulescu, J.; Calhoun, T. L.; Gawley, R. E.; Leblanc, R. M. Coumaryl crown ether based chemosensors: selective detection of saxitoxin in the presence of sodium and potassium ions. Tetrahedron Lett. 2002, 43, 4413–4416. [Google Scholar]
- Kremer, F. J. B.; Chiosis, G. C.; Engbersen, J. F. J.; Reinhoudt, D. N. Improved guanidinium ion-selectivity by novel calixaren and calixarene receptor molecules on CHEMFETs. J. Chem. Soc. Perkin Trans. 1994, 2, 677–681. [Google Scholar]
- Lee, M.; Kim, T. I.; Kim, K. H.; Kim, J. H.; Choi, M. S.; Choi, H. J.; Koh, K. Formation of a self-assembled phenylboronic acid monolayer and its application toward developing a surface plasmon resonance-based monosaccharide sensor. Analytical Biochemistry 2002, 310, 163–170. [Google Scholar]
- Hur, Y.; Ock, K.; Kim, K.; Jin, S.; Gal, Y.; Kim, J.; Kim, S.; Koh, K. Surface plasmon resonance study on enhanced refractive index change of an Ag+ ion-sensing membrane containing dithiosquarylium dye. Analytica Chimica Acta 2002, 460, 133–139. [Google Scholar]
- Chen, H.; Lee, M.; Choi, S.; Kim, J. H.; Choi, H. J.; Kim, S. H.; Lee, J.; Koh, K. Comparative study of protein immobilization properties on calixarene monolayers. Sensors 2007, 7, 1091–1107. [Google Scholar]
- Iki, N.; Narumi, F.; Fujimoto, T.; Morohashi, N.; Miyano, S. Selective synthesis of three different isomers of tetrakis[(ethoxycarbonyl)methoxy]thia-calixarene and their complexation properties towards alkali metal ions. J. Chem. Soc. Perkin Trans. 1998, 2, 2745–2750. [Google Scholar]
- Perret, F.; Morel, J. P.; Morel-desrosiers, N. Thermodynamics of the complexation of the p-sulfonatocalixarene with simple model guests in water: a microcalorimetric study. Supramolecular Chemistry 2003, 15, 199–206. [Google Scholar]
- Ghidini, E.; Ugozzoli, F.; Ungaro, R.; Harkema, S.; El-Fadl, A. A.; Reinhoudt, D. N. Complexation of alkali metal cations by conformationally rigid, stereoisomeric calixarene crown ethers: a quantitative evaluation of preorganization. J. Am. Chem. Soc. 1990, 112, 6979–6985. [Google Scholar]
|Table 1. The refractive index (RI) of the interfacial layer according to the interaction between 3 SAM and the different concentration of STX.|
|[STX] (M)||1.0× 10-10||1.0× 10-9||1.0× 10-8||1.0× 10-7||1.0× 10-6||1.0× 10-5|
© 2007 by MDPI ( http://www.mdpi.org). Reproduction is permitted for noncommercial purposes.