- freely available
Sensors 2014, 14(4), 7120-7141; doi:10.3390/s140407120
Published: 22 April 2014
Abstract: The Spanish Ministry of Defense, through its Future Combatant program, has sought to develop technology aids with the aim of extending combatants' operational capabilities. Within this framework the ATREC project funded by the “Coincidente” program aims at analyzing diverse biometrics to assess by real time monitoring the stress levels of combatants. This project combines multidisciplinary disciplines and fields, including wearable instrumentation, textile technology, signal processing, pattern recognition and psychological analysis of the obtained information. In this work the ATREC project is described, including the different execution phases, the wearable biomedical measurement systems, the experimental setup, the biomedical signal analysis and speech processing performed. The preliminary results obtained from the data analysis collected during the first phase of the project are presented, indicating the good classification performance exhibited when using features obtained from electrocardiographic recordings and electrical bioimpedance measurements from the thorax. These results suggest that cardiac and respiration activity offer better biomarkers for assessment of stress than speech, galvanic skin response or skin temperature when recorded with wearable biomedical measurement systems.
The Spanish Future Combatant Program (ComFut) developed by the Spanish Ministry of Defense pursues the integration of technology aids for the combatant with the aim of including not only intelligence facilities, geographical information systems or improved ballistics, but also systems able to analyze the physical and psychological state of the individual. With the aim of extending operational capability, the combatant must be not only by physically ready, but also mentally trained, which encompasses from stress management to dream deprivation. In this issue, information supported by augmented reality systems and real time monitoring of biometrics can be used to improve the mental capabilities of the future combatant.
In this framework, this paper presents the ATREC project funded by the “Coincidente” Spanish program of the Ministry of Defense, which aims at analyzing the use of biometrics and real time monitoring of combatant stress levels. This is a novel and multidisciplinary objective, since it includes the combination of wearable sensors, signal processing, pattern recognition and psychological analysis of the obtained information.
1.1. The ATREC Project
ATREC is the acronym for the Spanish project name “Analisis en Tiempo Real del Estrés del Combatiente” (Real-Time Analysis of the Stress of the Combatant). The main objective of the project is to set up a complete system to assess in real time the emotional, physical and mental stress load of a combatant during combat. It is based on the use of wearable measurement systems for the acquisition of non-invasive physiological measurements and wireless communication. Figure 1 shows a diagram of the whole system.
The project is divided in two main stages or phases. In a first phase, the ATREC project aims at identifying indicators suitable for monitoring Autonomous Nervous System (ANS) activity that can feasibly be acquired by customized wearable instrumentation. In this stage a set of sensorized garments has been implemented for performing non-invasive recording of several physiological variables. These variables include galvanic skin response (GSR), body temperature, electrocardiogram (ECG), thoracic electrical bioimpedance (TEB), and voice recording for speech analysis. In order to carry out this stage, a system prototype has been designed and implemented, and it has been used in a set of controlled stress experiments with a wide number of subjects. The prototype system is based on several different measurement systems that combined with different sensorized garments has allowed changes in the position and modalities of the different sensors, generating a wide number of possible measurement configurations.
In a second phase, a final prototype has been implemented allowing the measurement of the selected most important measurements, with the final configuration of the sensors. It consists in a uniform vest that embeds all the electronics and signal processing algorithms to assess and represent the different stress levels in both the soldier and command displays.
This manuscript is mainly focused on the work done during the first phase, the one targeting the selection of the main biomedical parameters for assessment of mental stress and workload. The second phase addresses the construction of sensorized vest and, its validation and the usability study is out of the scope of this manuscript and it will be described elsewhere.
1.2. Wearable Biomedical Measurement Systems
Advances in the fields of electronics and instrumentation technology, together with novel textile materials and novel textile-electronic integration technics have boosted the development and implementation of sensorized garments for wearable biomedical measurement systems , specially in the last decade.
As a result of extensive and numerous research efforts, several wearable and textile-enable monitoring systems for the performance of non-invasive measurements have been designed and implemented like a sensorized chest strap for cardiopulmonary activity status during emergencies , a sensorized vest incorporating fully woven textile sensors for monitoring ECG and respiration rate  or the HeartCycle Sensorized vest for detection of cardiogenic pulmonary edema . A comprehensive listing of wearable biomedical measurement systems is available in .
Moreover, leaving aside research project prototypes, currently it is possible to find commercially available wearable sensing technologies for several different applications, ranging from simple wearable heart rate (HR) monitors for fitness, e.g., the Polar heart rate strap , to body worn multiparametric measurement systems, e.g., the EQ02 life monitor manufactured by Equivital  or the nECG shirt L1 developed by nuubo which is designed to measure ECG .
Available wearable biomedical measurement systems and the respective enabling technologies allow the development of solutions not only for personalized health monitoring applications e.g., management of chronic patients and home-care, but also for personal protective equipment for workers operating in dangerous and stressful conditions e.g., soldiers, firemen, bomb-squad members, etc.
1.3. Non-Invasive Assessment of Mental Stress
The brain and the spinal cord comprise the central nervous system (CNS), which through the Autonomic Nervous System (ANS) and peripheral innervation of organs and glands controls the heart's electrical activity, gland secretion, blood pressure, and respiration function among others to preserve the homeostasis of the organism, i.e., the human body in this case. The ANS is responsible for enabling and controlling the reaction of the body to external and internal stimuli. Such a control function is performed through the two branches of the ANS: The Sympathetic Nervous System (SNS) and the Parasympathetic Nervous System (PNS). Analysis of ANS activity regarding SNS and PNS is a common practice for assessing stress .
Evaluation of ANS activity can be performed by recording and analysing several physiological variables: heart rate , respiration rate (RespR) , ElectroEncephaloGraphical (EEG) activity , skin galvanic response (GSR) [12,13] and skin temperature .
The analysis of a speaker's speech can provide information about cognitive and affective task workloads as well as performance levels [15,16]. Interpersonal oral communication includes both verbal and paraverbal information that allows the communication of emotions and feelings . Therefore, speech analysis if combined with the sensor-based detection process may improve the final performance of the stress detection system. In this line the aforesaid first phase of the ATREC project also includes the use of voice recordings for further speech analysis.
1.4. Sensorized Garments and Instrumentation for Assessment of ANS Activity
The available literature indicates that most of the wearable systems made for assessment of ANS activity have targeted the study of emotional, cognitive, physical arousal mental status and stress [12,18–28]. Nevertheless several wearable devices and sensorized garments have been implemented and used to record physiological variables to study the response of the ANS during stressful tasks in a non-invasive manner through, e.g., parameters from cardiac activity, GSR dynamics, skin temperature and RespR [10,13,29,30].
The sensorized garments manufactured for recording such relevant physiological variables are mostly chest straps [5,6,31–33] and t-shirts [7,20,21,34–40] aiming to record respiration and heart activity and gloves [20–23], glove-like devices [19,24,25] and wrist-bands [26–28,41] for recording GSR and temperature.
Since the turn of the century, sensorized gloves, gloves-like systems and wristbands have been used for obtaining GSR recordings and performing electrodermal response studies. With the passage of time the level of electronic integration has increased. Initially metallic [25–28] or Ag/AgCl external electrodes [20,21,24,41] were used, later a glove with an integrated electrodermal response sensor was manufactured , and an eventually gloves with textrodes integrated in the garment have been produced .
Several textile-enabled measurement devices and sensorized garments have been implemented for recording physiological variables that would allow to study the response of the ANS during stressful tasks in a non-invasive manner e.g., parameters from cardiac activity, GSR dynamics, skin temperature and RespR [10,13,29,30]. Some details relevant to the following Sections: Section 2.1 Measuring Devices and Section 2.2 Sensorized Garments have been published elsewhere [42,43] and validate the measurement performance of the wearable biomedical measurement system produced during the 1st phase of the ATREC project.
2.1. Wearable Biomedical Measurement Devices
For the 1st phase, two non-invasive physiological measurement devices have been built: One to measure GSR and skin temperature and a second device to record cardiogenic biopotentials and impedance thoracic measurements. Both are battery operated with 900 mAh lithium-ion batteries, and are provided with an internal 4GB SD memory card to allow long-term recording and off-line data analysis.
2.1.1. GSR and Body Temp
One of the devices enables skin impedance monitoring as well as ambient and body temperature monitoring. The reduced dimensions (50 × 35 × 15 mm) enable its integration in a wearable biomedical measurement system in a straightforward manner. In this case, this device is intended to be used with the sensorized glove and/or the arm-strap that will be explained in the following Sections, see Figure 2.
For the GSR measurements a Wheatstone bridge topology with constant voltage excitation of 0.5 V is implemented. The voltage is applied to the skin through a resistor of 2 KΩ that limits the maximum current. The Wheatstone bridge differential voltage is measured by an instrumentation amplifier that it is connected to the input of an Analog-to-Digital Converter (ADC) available in the microcontroller in order to obtain the GSR value. The GSR is measured with a sampling frequency of 250 Hz.
The temperature measurements are performed through a 1-wire DS1825 digital thermometer manufactured by Maxim Integrated (San Jose, CA, USA). The 1-Wire® communication protocol implemented in the DS1825 allows the use of only one data line and ground for the communications with the microprocessor as well as for the power supply. The DS1825 incorporates a hardware 4-bit identification code that allows the connection of several temperature sensors to the same data line. The GSR measuring unit is provided with two temperature sensors, one external temperature sensor that is in contact with the skin and one internal temperature sensor placed in the microcontroller board. A sampling frequency of 1 Hz is used to measure both temperatures.
2.1.2. ECG amplifier and EBI Plethysmographer
This device records both the ECG biopotential signal as well as Thoracic Electrical Bioimpedance at a single frequency of 50 kHz. Its reduced dimensions (50 × 35 × 20 mm), makes the device suitable for easy integration with the chest strap electrode system described in the following section. The intended use for this ECG/EBI sensing unit is to record cardiogenic biopotentials to compute the HR from the acquired ECG and measure the impedance change caused during breathing to extract the RespR from the recorded changing TEB signal. In Figure 3 it is possible to observe the device with a typical electrode connection for ECG and TEB.
The ECG signal is measured by a 1-lead instrumentation amplifier topology suggested by Merritt et al. , the output voltage is connected to an ADC input in the microcontroller and recorded with a sampling frequency of 250 Hz. The TEB measurement is obtained using an excitation frequency of 50 kHz and a sampling frequency of 100 Hz. The impedance estimation core is based in the SOC AD5933 manufactured by Analog Devices (Norwood, MA, USA), implementing an analog front-end customized for four-electrode measurements .
2.2. Sensorized Garments and Textrodes
The following garments have been confectioned with several types of textile materials including conductive fabrics used for the textrodes and electrical connections. The conductive fabric used is the Shieldex® Fabric P130+B manufactured by STATEX GmbH, (Bremen, Germany). All the sensorized garments and their different supporting elements required to operate with the textile-enabled instrumentation mentioned in the previous section are described in this section.
2.2.1. Glove for GSR and Skin Temperature
Two textile electrodes have been integrated inside of the glove in the proximal phalanx of the index and middle fingers on the inside of the glove for measuring the GSR, and a temperature sensor has been placed in the tip of the ring finger of the glove to sense the peripheral skin temperature, see Figure 4 for details.
The textrodes and the temperature sensors are connected through 4 cables with the measuring device that is fastened with Velcro to a wristband as shown in Figure 5.
2.2.2. Upper-Arm Strap for GSR and Superficial Temperature
An upper arm strap has been confectioned with two integrated textrodes to sense the galvanic skin response in the skin surface. A DS1825 sensor is also integrated in the inner lining of the strap in order to contact the skin and in this way measure the skin surface temperature. Figure 6 shows the design of the confectioned sensorized garment and Figure 7a shows the confectioned sensorized upper arm strap and Figure 7b shows the same arm strap connected to the measurement unit for GSR and body temp measurement while worn on the arm.
2.2.3. Chest Straps System for Cardiac and Respiration Recordings
A chest strap garment with repositionable textile electrodes to record 1-lead ECG from two textrodes and tetrapolar TEB measurements with another four textrodes has been developed. The possibility of placing the electrodes in any place along the horizontal and vertical straps enables to test different measurement configurations for ECG and TEB measurements, see Figure 8a. Depending on the placement of the textrodes around the surface of the thorax and abdomen, the TEB measurement will have more or less cardiac and respiratory components allowing us to perform a multi-parametric signal recording if required.
Each strap is made with a highly elastic band with 1 cm perforations every 2 mm, see Figure 8b. These perforations enable the reposition of the textrodes along the straps and fixation between straps using chef jacket buttons, see Figure 8c.
2.2.4. Repositionable Textrodes
The textile electrodes have an approximate surface of 25 cm2 and are confectioned with a textile structure that is folded over itself creating a clamp through a male-female pair of press-studs, as shown in Figure 9. Note that the connection between the measurement leads and the textrode is achieved through a third press-stud, male.
2.3. Voice Recording
In order to record the voice during the experiments of the first phase of the project, a smartphone was custom programmed for such purpose. The selected device was the BRVAR909 manufactured by Bravus Europe AMISL (Andorra), an Android 2.2 based rugger device, IP67 specifications complaint including water and shock resistant. An 8 GB SD card was used as main storage system.
A customized Android application was designed and programmed, to allow the synchronized recording of the speakers' speech at 22.5 kHz. The local clock of the smartphone was also registered in this way, once the clock was synchronized with the rest of the measurement devices, the recordings of the speech could use the same time frames.
3.1. Activities for Assessment of Stress
In order to evaluate the performance of the prototype designed in the first phase of the project, an experiment was designed with the purpose of generating different types of stress in a set of subjects. The main purpose of this experiment was the controlled generation of emotional, mental and physical stress, through a set of stages. A PC-based application was created to induce stress situations in a controlled manner. The main tools used to generate stress are described in the following sections for each different type of stress.
3.1.1. Emotional Stress
In order to generate emotional stress, a set of different videos was used. The videos used were selected to specifically generate several types of emotions, including neutral, sadness, anger and disgust. In a first step, the videos were viewed without the interaction of the subject, and in a second step, they were repeated and the subject had to click under the emotion perceived in any moment. For this purpose, a circle of emotions inspired in the Plutchick circle was used . Each video used in the experiment was based on a set of segments of movies with a total duration of approximated 10 min, and the movies used were Earth (2007), Life is Beautiful (1997) and Cannibal Holocaust (1980).
3.1.2. Mental Stress
In a different way, mental stress was generated through two different games. These games were selected in order to put the subject at the limits of mental calculus. The first game consists in making a set of quick mental sums. A set of digits to be added appears on the screen, and the subject has to click on the correct solution. The time available to solve each problem varies along the experiment, being decreased when the subject answers right, and increased when he/she answers wrong. The idea is to put the subject in a high level of mental activity, independently of his/her calculus ability. This game was played since 80 sums from two to five digits were carried out. The second game consists in a Tetris-based game, in which again the objective is to seek the limit of the mental ability of the subject. For this purpose, the speed of the game was controlled so the difficulty could be adapted during the exercise.
3.1.3. Physical Stress
Finally, physical stress was generated making the subject go up and down in the main stairs of a three level building during 5 min. This activity was selected due to its simplicity and since it quickly generates a high physical stress level.
3.1.4. Surveys and Questionnaires for Stress Assessment
Both socio-demographical and psychological computer-based surveys were included in the experiments. The socio-demographical survey aimed at registering all the relevant information from each subject, including personal data, main biometrics, sport activities, any toxic habits and details about what the subject did during the hour previous to the experiment.
Concerning the psychological surveys, their purpose was to monitor the stress level and typology during the different parts of the experiments. The Self-Assessment Manikin  and the Profile of Mood States  were carried out after the visualization of each video in the emotional stress stages.
Short computer-based control surveys were also carried out after every stage of the experiment, in order to determine if the subject was concentrated on each task or if, by the contrary, he/she got bored during the experiment. The aim was to determine the effectiveness of the stress situations along the experiment.
3.1.5. Baseline and Sequence of Experimental Setup
The total duration of the experiment was around 75 min. In a first step after starting the recording of the different devices of the prototype, the subject fills out a sociocultural questionnaire. The aim is to establish a baseline in the register. Once the sociocultural questionnaire is completed, the different stages of the experiment are carried out. It is important to highlight that several surveys are also carried out in order to monitor the real stress level of the subject during the experimental process. The sequence of the experiment is described as follows:
Neutral stage: neutral video.
First mental stage: sum game.
First emotional stage: sad video.
Second mental stage: Tetris game.
Second emotional stage: disgusting video.
Physical stage: stair running.
3.2. Operational Test of the Wearable Biomedical Measurement Devices and Sensorized Garments
A total number of 12 measurement devices and sensorized garments have been produced for each modality. The correct operation of the measuring devices and the corresponding sensorized garments has been validated at least on two different healthy volunteers obtaining at least 2 min of GSR recordings with the glove and the arm strap as well as ECG and respiration activity. In addition, to such test performed systematically prior to the initiation of each measurement session of 75 min, non-systematic test have been carried out all through the implementation of the project. e.g., battery life test, where the systems were used continuously for more than 2 h.
3.3. Speech Analysis
Voice was recorded during the experiments with the aim of studying its use in order to assess emotional status. Several features were successfully calculated from the speech recordings, with the aim of implementing those more useful to determine the stress level. The features extracted in the voice analysis were statistical parameters from Mel Frequency Cepstral Coefficients, short time energy, pitch, pitch perturbation quotient, amplitude perturbation quotient, jitter, voice turbulence index, and soft phonation index.
3.4. Experimental Recordings Performed with the Wearable Biomedical Measurement Systems
Non-invasive recordings have been carried out with the 12 sets of wearable measurement devices and sensorized garments in 42 volunteers undergoing the battery of stress test described in previous sections. This way over 3,000 min of multichannel recordings have been produced.
3.5. Analysis of Multi-Sensorial Data
A total of 724 features from seven different signals were used to extract the information derived from the stress levels of the subjects. Time-Frequency and Statistical analysis has been performed on each of the different type of the signals. Several time configurations were tested in order to determine the best time frames for the stress analysis. For this purpose, two parameters were taken into account: the frame length and the frame overlap. Frame lengths of 10, 30, 60, 180 and 300 s were evaluated, and time overlaps of 0, 10, 30 and 60 were considered. From the preliminary results, we selected a frame length of 60 s and a time overlap of 10 s, so that a decision was taken every 10 s using information from the last 60 s of each signal. These values represented a tradeoff between time resolution and performance.
4.1. Validation of Sensorized and Wearable Biomedical Measurements Systems
This section presents the measurements recorded with the measuring devices described and the sensorized garments in Sections 2.1 and 2.2, respectively. More specifically, this section presents a set of measurements obtained during the validation test with the one measurement device and garment of each kind for a single subject. More detailed information has been presented in .
4.1.1. GSR and Temperature
Skin temperature and GSR were recorded in several subjects. While skin temperatures produced very stable measurements around 34 °C with both garments producing flat recordings, GSR measurements exhibited larger variance, see Figure 10. Note that the GSR data has been filtered with a moving average window of 5 s as in .
4.1.2. ECG Recordings
Surface cardiac biopotentials were recorded in several subjects with the wearable measurement system presented in Figure 8a. The ECG recording plotted in Figure 11a shows 60 s of an ECG record. As shown in the 4 second-long zoomed segment the recording is very fair, allowing for a straight detection of the R complex for heart rate assessment. The first 25–30 s show slow activity related with respiratory movement. In the plot it is possible to see that during the seconds 60 to 80 while the subject kept the breath, the ECG recording was completely stable.
4.1.3. Thoracic EBI Recordings
TEB recordings were obtained simultaneously to the ECG recording, see Figure 11, using the wearable measurement system presented in Figure 8a for several subjects. A 60-second segment of the TEB recording is shown in Figure 11b. The respiration activity is easily observable in this plot as the resistance increases and decreases according to the respiration cycle. Note that the changes stop between 60 and 80 s on the plot while the subject was holding his or her breath.
4.2. Voice Recordings
The recordings obtained with the BRVAR909 presented clear speech signals as shown in Figure 12, where a segment of 90 min is shown.
4.3. Measurement Database
The experimental test produced full recordings for 42 subjects from February 2013 to May 2013; generating more than 50 h of multisensorial data, see Table 1. All signals were stored in raw data, and later processed to extract features potentially useful to implement detection algorithms and classifiers for assessment of the user's mental stress.
4.4. Preliminary Analysis of Acquired Data
It is important to recall that the aim of the experiments during the first phase was to establish the potential usefulness of the different type of recordings measured with the implemented prototypes. Notice that, for such analysis work, the database was split in two: the first 25 subjects were used for design and training, and the remaining 17 subjects were used to testing purposes to validate the performance of the classification algorithms.
For this purpose, a 4-class classification scenario was designed, with the aim of determining in which stage was the subject: neutral stage, emotional stress stage, mental stress stage and physical stress stage. As previously indicated, a time window with a length of 60 s and a time overlap of 10 s was used, i.e., a decision was produced every 10 s using data from the last 60 s of each recording. These values represented a tradeoff between time resolution and performance.
From the 724 evaluated 192 were selected using a genetic algorithm  that were run through a Linear Discriminant Classifier (LDC).
Table 2 shows the results obtained with the implemented classifiers. Both the number of features selected by the genetic algorithm and the error rate obtained by the LDC are included. Results show the best performance of the electrocardiogram signals and the thoracic impedance signals, when compared to the other signals recorded in the experiments. The 10 features selected from the ECG recording and the 20 selected from the TEB measurement are enumerated in two lists in Appendix, respectively. It is important to highlight the performance of the speech analysis that resulted in a very high error rate. This is mainly caused by the absence of voice in most of the recorded data, making the speech-based classifier to achieve higher error rates.
5.1. Wearable Biomedical Measurement Devices and Sensorized Garments
Small sized measurement devices were made specifically to facilitate the integration with the sensorized garments. State-of-the-art techniques were used for textile-electronic integration like press-studs, which pressed through a conductive fabric is the most used approach to perform such type of textile-electronic interconnections and it can be found in commercial products [5,7] as well as in research prototypes [51,52]. Sewing with conductive yarn through conductive fabrics has been used before to establish an electrical connection . The sensorized garments have fulfill their intended purpose in Phase 1 allowing the acquisition of several different physiological measurements during exercises designed to specifically target assessment of stress. The performance of the measurement devices and the sensorized garments was previously validated in [42,43].
5.2. Preliminary Observations from Stress Experiments
It is clearly noticed from the error rate reported in Table 2, that the biosignals producing the best classification of mental status are the ECG, mainly heart rate and the respiration rate. Such results may be expected since both respiration and heart rate are highly mediated by the ANS and also it is relatively easier to obtain robust ECG and TEB measurements with tight garments on the thorax than GSR or temperature from fingers or arms. A detail description of the multiparametric and classification work performed on the database will be reported elsewhere.
The obtained error rates in detection of stress from heart rate and the respiration rate, 23.72% and 24.40%, respectively, are slightly better than 31% error rate obtained from HR variability analysis as shown in . The performance exhibit by heart rate and the respiration rate is also superior than the performance obtained from analysis of the electroculogram with an error rate >25% , while they are comparable to the performance obtained from ECG analysis  and significantly worse than the performance obtained from EEG analysis, with error rates as low as 6% .
5.3. Voice Recording
The classification error obtained with the analysis of the recorded speech (>48%) is slightly over what other authors have reported in  for automatic classification of stress from speech analysis (46% & 44%, using Hidden Markov Models and Vector Quantification, respectively). When comparing the 48% error rate obtained in this study with the error rates obtained with traditional Teager Energy Operator-based feature vector with a Hidden Markov Model-trained classifier (22.5% error), or using a weighted sub-band detection scheme (4.7%)  the differences are significant.
In either case the error rate obtained based on ECG or TEB features was significantly lower. The relatively larger error rate produced when using speech-based features might be due to the fact that while voice recordings were performed in every experiment for the duration of the experiment, unfortunately the test activities did not produce enough speech content to be able to characterize the different states properly. Hopefully during the final validation test planned for the end of the second phase, the voice recordings will contain more speech information useful for classification purpose.
5.4. Lessons Learned
Executing this project has given us the opportunity to obtain new experiences and learn from them. One issue that we find important is judging what is enough, otherwise the production of excess data, samples or data might increase the requirements of the project unnecessarily and can have undesired consequences. Considering the exploratory nature of the first phase, manufacturing 12 different sets of measurement devices and sensorized garments was completely unnecessary and not efficient since it required more financial and staff resources. The fact is that the experiments were never done simultaneously, which means that a few measurement sets would have been enough.
Another lesson learnt that had a significant influence on the realization of the experiments was the use of a SD card for data storage and also for data access, instead than an alternative method for data access like Bluetooth. The lack of this feature meant that prior to the execution of each experiment to perform a operational test, the SD card had to be removed and read in the PC and despite this fact, there were always doubts about whether the measurements were being recorded properly because the access to the recorded data could not be done until the experiment was finalized. This limitation has been overcome in the measurement device implemented during Phase 2.
6. Conclusions and Outlook
The wearable measurement systems and the voice recording system have allowed the execution of activities specifically targeted to evaluate the influence of stress on several physiological variables. Therefore the 1st phase of the ATREC project has been completed and the preliminary analysis from the experimental results suggest that for the 2nd phase, it is very likely that a sensorized vest or T-shirt with integrated textrodes for biopotential recordings and trans-thoracic bioimpedance measurements will be manufactured, similar to the one reported in .
The ATREC project is funded by the Spanish Ministry of Defense through the “COINCIDENTE” program (DGAM DNR 1003211003500) and it is executed under the supervision of the Metrology and Human Factors Department, of “La Marañosa” Institute of Technology.
F.S., J.F. and R.B. have implemented the sensorized garments and performed the functional tests. F.S. have conceived the Wearable Measurement Systems. F.G. has developed the measurement instrumentation. F.S. and J.F. have integrated electronic the measurement devices with the textile and wearable sensors. I.M.-H., D.A., C.L. and R.G.-P. have participated in the execution of the stress test and have performed the analysis of the recordings. F.S. and R.G.-P. have supervised both the technical and experimental activities. F.S. and R.G.-P. have written the initial manuscript and I.M.-H., J.F., L.A., R.B., D.A. and C.L. have participated in the revision of the manuscript supporting the writing process.
Conflicts of Interest
The authors declare no conflict of interest.
Features Selected for the Obtained Error Rates
A1. From Electrocardiogram
Mean value of the heart rate
Standard deviation of the heart rate
Third quartile of the heart rate
Logarithm of the spectrum in mid frequencies (0.08 Hz–0.15 Hz) of the heart rate
Ratio of the logarithm of the low frequency energy (0.04 Hz–0.08 Hz) vs. Logarithm of the high frequency energy (0.15 Hz–0.5 Hz) of the heart rate
First quartile of the heart rate variability
Harmonic mean of the heart rate variability
Logarithm of the standard deviation of the heart rate variability
Average number of consecutive RR intervals separated more than 50 ms (pNN50)
Mean value of the ultra-low frequency band (0 Hz–0.04 Hz) of the heart rate
A2. From Thoracic Electrical
First quartile of the low frequency component (0.04Hz–0.08Hz) of the TEB
Mode of the mid frequency component (0.08 Hz–0.15 Hz) of the TEB
Minimum value of the high frequency component (0.15 Hz–0.5 Hz) of the TEB
Mean value of the ultra-low frequency band (0 Hz–0.04 Hz) of the TEB
Mean value of the respiratory rate
Standard deviation of the respiratory rate
Third quartile of the respiratory rate
Kurtosis coefficient of the respiratory rate
Sample entropy of the second differentiation of the respiratory rate
Sample entropy of the third differentiation of the respiratory rate
Variance coefficient of the inspired respiratory flux
Mean value of the total respiratory time
Standard deviation of the total inspiration time
Harmonic mean of the respiratory rate
Trimmed mean of the total respiratory time
First LPC coefficient of the total respiratory volume
Mean value of the inspired respiratory flux
Variation coefficient of the inspired respiratory flux
Mean value of the ultra low frequency band (0–0.04 Hz) of the inspiration time
Mean value of the ultra low frequency band (0–0.04 Hz) of the total respiratory volume
- Chan, M.; Esteve, D.; Fourniols, J.Y.; Escriba, C.; Campo, E. Smart wearable systems: Current status and future challenges. Artif. Intell. Med. 2012, 56, 137–156. [Google Scholar]
- Lanata, A.; Scilingo, E.P.; de Rossi, D. A multimodal transducer for cardiopulmonary activity monitoring in emergency. IEEE Trans. Inf. Technol. Biomed. 2010, 14, 817–825. [Google Scholar]
- Di Rienzo, M.; Rizzo, F.; Meriggi, P.; Castiglioni, P.; Mazzoleni, P.; Ferrarin, M.; Ferratini, M. MagIC: A textile system for vital signs monitoring. Advancement in design and embedded intelligence for daily life applications. Proceedings of 29th Annual International Conference of the IEEE Engineering in Medicine and Biology Society, Lyon, France, 22–26 August 2007; pp. 3958–3961.
- Reiter, H.; Muehlsteff, J.; Sipila, A. Medical application and clinical validation for reliable and trustworthy physiological monitoring using functional textiles: Experience from the HeartCycle and MyHeart project. Proceedings of IEEE-EMBC, Boston, US, 30 August–3 September, 2011; pp. 3270–3273.
- Polar Electro Oy. Polar H6 Hear Rate Sensor. Available online: http://www.polarusa.com/us-en/products/accessories/H6 (accessed on 10 March 2014).
- Vivonoetics. The Equivital TnR Product Range. Available online: http://vivonoetics.com/wp-content/downloads/Brochures/GeneralTnRBrochureVivonoeticscontact.pdf (accessed on 10 March 2014).
- Nuubo. nECG shirt L1. Available online: http://www.nuubo.com/sites/default/themes/nuubo2/pdf/DATASHEETS_EN_shirt.pdf (accessed on 10 March 2014).
- Sloan, R.P.; Shapiro, P.A.; Bagiella, E.; Boni, S.M.; Paik, M.; Bigger, J.T., Jr.; Steinman, R.C.; Gorman, J.M. Effect of mental stress throughout the day on cardiac autonomic control. Biol. Psychol. 1994, 37, 89–99. [Google Scholar]
- Jovanov, E.; Lords, A.O.; Raskovic, D.; Cox, P.G.; Adhami, R.; Andrasik, F. Stress monitoring using a distributed wireless intelligent sensor system. IEEE Eng. Med. Biol. Mag. 2003, 22, 49–55. [Google Scholar]
- Bernardi, L.; Wdowczyk-Szulc, J.; Valenti, C.; Castoldi, S.; Passino, C.; Spadacini, G.; Sleight, P. Effects of controlled breathing, mental activity and mental stress with or without verbalization on heart rate variability. J. Am. Coll. Cardiol. 2000, 35, 1462–1469. [Google Scholar]
- Saidatul, A.; Paulraj, M.P.; Yaacob, S.; Mohamad, N.N.F. Automated system for stress evaluation based on EEG signal: A prospective review. Proceedings of IEEE 7th International Colloquium on Signal Processing and Its Applications, Penang, Malaysia, 4–6 March 2011; pp. 167–171.
- Setz, C.; Arnrich, B.; Schumm, J.; La Marca, R.; Troster, G.; Ehlert, U. Discriminating stress from cognitive load using a wearable EDA device. IEEE Trans. Inf. Technol. Biomed. 2010, 14, 410–417. [Google Scholar]
- Villarejo, M.V.; Zapirain, B.G.; Zorrilla, A.M. A stress sensor based on Galvanic Skin Response (GSR) controlled by ZigBee. Sensors 2012, 12, 6075–6101. [Google Scholar]
- Kataoka, H.; Kano, H.; Yoshida, H.; Saijo, A.; Yasuda, M.; Osumi, M. Development of a skin temperature measuring system for non-contact stress evaluation. Proceedings of the 20th Annual International Conference of the IEEE Engineering in Medicine and Biology Society, Hong Kong, China, 29 October–1 November 1998; pp. 940–943.
- Looije, R.; te Brake, G.; Neerincx, M. Geo-collaboration under stress. Prodeedings of Workshop on Mobile HCI for Emergencies, Singapore, 9 September 2007.
- Charfuelan, M.; Kruijff, G.-J. Analysis of Speech Under Stress and Cognitive Load in USAR Operations. In Natural Interaction with Robots, Knowbots and Smartphones; Mariani, J., Rosset, S., Garnier-Rizet, M., Devillers, L., Eds.; Springer: New York, NY, USA, 2014; pp. 275–281. [Google Scholar]
- Rothkrantz, L.J.; Wiggers, P.; Wees, J.-W.A.; Vark, R.J. Voice Stress Analysis. In Text, Speech and Dialogue; Sojka, P., Kopefcek, I., Pala, K., Eds.; Springer: Berlin/Heidelberg, Germany, 2004; Volume 3206, pp. 449–456. [Google Scholar]
- Ivonin, L.; Chang, H.-M.; Chen, W.; Rauterberg, M. Unconscious Emotions: Quantifying and Logging Something We Are not Aware of. Personal Ubiquitous Comput. 2012, 17, 663–673. [Google Scholar]
- Kimmy, A.; Chen, W.; Lindsay, B. Smart Photo Frame for Arousal Feedback—Wearable sensors and intelligent healthy work environment. Proceedings of Workshop on Smart Offices and Other Workplaces of the 7th International Conference on Intelligent Environments, Nottingham, UK, 25–28 July 2011; pp. 685–696.
- Axisa, F.; Dittmar, A.; Delhomme, G. Smart clothes for the monitoring in real time and conditions of physiological, emotional and sensorial reactions of human. Proceedings of the 25th Annual International Conference of the IEEE Engineering in Medicine and Biology Society, Valencia, Spain, 21–25 July 2003; pp. 3744–3747.
- Axisa, F.; Gehin, C.; Delhomme, G.; Collet, C.; Robin, O.; Dittmar, A. Wrist ambulatory monitoring system and smart glove for real time emotional, sensorial and physiological analysis. Proceedings of 26th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC2004), San Francisco, CA, USA, 1–5 September 2004; pp. 2161–2164.
- Valenza, G.; Lanata, A.; Scilingo, E.P.; de Rossi, D. Towards a smart glove: Arousal recognition based on textile electrodermal response. Proceedings of Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC2010), Buenos Aires, Argentina, 31 August–4 September 2010; pp. 3598–601.
- Katsis, C.D.; Katertsidis, N.; Ganiatsas, G.; Fotiadis, D.I. Toward emotion recognition in car-racing drivers: A biosignal processing approach. IEEE Trans. Syst. Man Cybern. Part A Syst. Hum. 2008, 38, 502–512. [Google Scholar]
- Strauss, M.; Reynolds, C.; Hughes, S.; Park, K.; McDarby, G.; Picard, R.W. The handwave bluetooth skin conductance sensor. In Affective Computing and Intelligent Interaction; Springer Berlin: Heidelberg: Beijing, China October 22–24, 2005. ; pp. 699–706.
- Scheirer, J.; Picard, R. The Galvactivator: A Glove that Senses and Communicates Skin Conductivity. Proceedings of the 9th International Conference on Human-Computer Interaction, LA, USA, 5–10 August 2001.
- Fletcher, R.R.; Dobson, K.; Goodwin, M.S.; Eydgahi, H.; Wilder-Smith, O.; Fernholz, D.; Kuboyama, Y.; Hedman, E.B.; Poh, M.-Z.; Picard, R.W. iCalm: Wearable sensor and network architecture for wirelessly communicating and logging autonomic activity. IEEE Trans. Inf. Technol. Biomed. 2010, 14, 215–223. [Google Scholar]
- Poh, M.-Z.; Swenson, N.C.; Picard, R.W. A wearable sensor for unobtrusive, long-term assessment of electrodermal activity. IEEE Trans. Biomed. Eng. 2010, 57, 1243–1252. [Google Scholar]
- Ouwerkerk, M. Unobtrusive Emotions Sensing in Daily Life. In Sensing Emotions; Westerink, J., Krans, M., Ouwerkerk, M., Eds.; Springer Netherlands: Eindovhen, Netherlands, 2011; Volume 12, pp. 21–39. [Google Scholar]
- De Santos Sierra, A.; Avila, C.S.; Guerra, C.J.; del Pozo, G.B. A Stress-Detection System Based on Physiological Signals and Fuzzy Logic. IEEE Trans. Ind. Electron. 2011, 58, 4857–4865. [Google Scholar]
- Zhai, J.; Barreto, A.B.; Chin, C.; Chao, L. Realization of stress detection using psychophysiological signals for improvement of human-computer interactions. Proceedings of IEEE SoutheastCon 2005, Ft. Lauderdale, FL, USA, 8–10 April 2005; pp. 415–420.
- Liu, G.-Z.; Huang, B.-Y.; Wang, L. A wearable respiratory biofeedback system based on generalized body sensor network. Telemed. e-Health 2011, 17, 348–357. [Google Scholar]
- Braecklein, M.; Tchoudovski, I.; Moor, C.; Egorouchkina, K.; Pang, L.; Bolz, A. Wireless telecardiological monitoring system for the homecare area. Proceedings of 27th Annual International Conference of the IEEE Engineering in Medicine and Biology Society (EMBC2005), Shanghai, China, 1 April 2005; pp. 3793–3795.
- Gargiulo, G.; Bifulco, P.; Cesarelli, M.; Ruffo, M.; Romano, M.; Calvo, R.A.; Jin, C.; van Schaik, A. An ultra-high input impedance ECG amplifier for long-term monitoring of athletes. Med. Devices 2010, 3, 1–9. [Google Scholar]
- De Rossi, D.; Carpi, F.; Lorussi, F.; Mazzoldi, A.; Paradiso, R.; Scilingo, E.P.; Tognetti, A. Electroactive fabrics and wearable biomonitoring devices. Autex Res. J. 2003, 3, 180–185. [Google Scholar]
- Grossman, P. The LifeShirt: A multi-function ambulatory system monitoring health, disease, and medical intervention in the real world. Stud. Health Technol. Inform. 2004, 108, 133–141. [Google Scholar]
- Weber, J.; Blanc, D.; Dittmar, A.; Comet, B.; Corroy, C.; Noury, N.; Baghai, R.; Vaysse, S.; Blinowska, A. Telemonitoring of vital parameters with newly designed biomedical clothing. Stud. Health Technol. Inform. 2004, 108, 260–265. [Google Scholar]
- Zhang, Y.-T.; Poon, C.C.; Chan, C.-H.; Tsang, M.W.; Wu, K.-F. A health-shirt using e-textile materials for the continuous and cuffless monitoring of arterial blood pressure. Proceedings of 3rd IEEE/EMBS International Summer School on Medical Devices and Biosensors, Cambridge, MA, USA, 5–6 September 2006; pp. 86–89.
- Yoo, J.; Yan, L.; Lee, S.; Kim, H.; Yoo, H.-J. A wearable ECG acquisition system with compact planar-fashionable circuit board-based shirt. IEEE Trans. Inf. Technol. Biomed. 2009, 13, 897–902. [Google Scholar]
- Linz, T.; Kallmayer, C.; Aschenbrenner, R.; Reichl, H. Fully untegrated EKG shirt based on embroidered electrical interconnections with conductive yarn and miniaturized flexible electronics. Proceedings of International Workshop on Wearable and Implantable Body Sensor Networks (BSN 2006), Boston, MA, USA, 3–5 April 2006; pp. 26–30.
- Paradiso, R.; Loriga, G.; Taccini, N.; Gemignani, A.; Ghelarducci, B. Wealthy, a wearable health-care system: new frontier on etextile. J. Telecommun. Inf. Technol. 2005, 4, 105–113. [Google Scholar]
- Picard, R.W. Future affective technology for autism and emotion communication. Phil. Trans. Roy. Soc. B Biol. Sci. 2009, 364, 3575–3584. [Google Scholar]
- Seoane, F.; Ferreira, J.; Alvarez, L.; Buendia, R.; Ayllón, D.; Llerena, C.; Gil-Pita, R. Sensorized Garments and Textrode-Enabled Measurement Instrumentation for Ambulatory Assessment of the Autonomic Nervous System Response in the ATREC Project. Sensors 2013, 13, 8997–9015. [Google Scholar]
- Ferreira, J.; Álvarez, L.; Buendía, R.; Ayllón, D.; Llerena, C.; Gil-Pita, R.; Seoane, F. Bioimpedance-Based Wearable Measurement Instrumentation for Studying the Autonomic Nerve System Response to Stressful Working Conditions. J. Phys. Conf. Ser. 2013, 434, 012015. [Google Scholar]
- Merritt, C.R.; Nagle, H.T.; Grant, E. Fabric-based active electrode design and fabrication for health monitoring clothing. IEEE Trans. Inf Technol Biomed. 2009, 13, 274–80. [Google Scholar]
- Seoane, F.; Ferreira, J.; Sanchéz, J.J.; Bragós, R. Analog Front-End Enables Electrical Impedance Spectroscopy System On-Chip for Biomedical Applications. Physiol. Meas. 2008, 29, S267–S278. [Google Scholar]
- Plutchik, R.; Kellerman, H.; Conte, H.R. A structural theory of ego defenses and emotions. In Emotions in personality and psychopathopathology; Izard, C.E., Ed.; Springer US: New York, NY, USA, 1979; pp. 227–257. [Google Scholar]
- Bradley, M.M.; Lang, P.J. Measuring emotion: The Self-Assessment Manikin and the Semantic Differential. J. Behav. Ther. Exp. Psychiat. 1994, 25, 49–59. [Google Scholar]
- McNair, D.; Lorr, M.; Droppleman, L. Manual for the Profile of mood states (POMS); Educational and Industrial Testing Service: San Diego, CA, USA, 1971; pp. 1–27. [Google Scholar]
- Wu, S.; Lin, T. Exploring the use of physiology in adaptive game design. Proceedings of International Conference on Consumer Electronics, Communications and Networks (CECNet), 16–18 April 2011; pp. 1280–1283.
- Raymer, M.L.; Punch, W.F.; Goodman, E.D.; Kuhn, L.A.; Jain, A.K. Dimensionality reduction using genetic algorithms. IEEE Trans. Evol. Comput. 2000, 4, 164–171. [Google Scholar]
- Marquez, J.C.; Seoane, F.; Välimäki, E.; Lindecrantz, K. Comparison of Dry-Textile Electrodes for Electrical Bioimpedance Spectroscopy Measurements. J. Phys. Conf. Ser. 2010, 224, 012140. [Google Scholar]
- Marquez, J.C.; Seoane, F.; Lindecrantz, K. Textrode functional straps for bioimpedance measurements—Experimental results for body composition analysis. Eur. J. Clin. Nutr. 2013, 67 (Suppl 1). [Google Scholar] [CrossRef]
- Bouwstra, S.; Wei, C.; Feijs, L.; Oetomo, S.B. Smart Jacket Design for Neonatal Monitoring with Wearable Sensors. Proceedings of Sixth International Workshop on Wearable and Implantable Body Sensor Networks (BSN2009), Berkeley, CA, USA, 3–5 June 2009; pp. 162–167.
- Jongyoon, C.; Gutierrez-Osuna, R. Using Heart Rate Monitors to Detect Mental Stress. Proceedings of Sixth International Workshop on Wearable and Implantable Body Sensor Networks (BSN2009), Berkeley, CA, USA, 3–5 June 2009; pp. 219–223.
- Laurent, F.; Valderrama, M.; Besserve, M.; Guillard, M.; Lachaux, J.-P.; Martinerie, J.; Florence, G. Multimodal information improves the rapid detection of mental fatigue. Biomed. Signal. Process. Control 2013, 8, 400–408. [Google Scholar]
- Namdeo, P.R.; Mishra, A.K. Speech analysis under stress. World J. Sci. Technol. 2012, 2, 174–177. [Google Scholar]
- Hansen, J.; Kim, W.; Rahurkar, M.; Ruzanski, E.; Meyerhoff, J. Robust Emotional Stressed Speech Detection Using Weighted Frequency Subbands. Eur. J. Adv. Signal Process. 2011, 2011, 906789. [Google Scholar]
- Marquez, J.C.; Rempfler, M.; Seoane, F.; Lindecrantz, K. Textrode-enabled Transthoracic Electrical Bioimpedance Measurements. Towards Wearable Applications of Impedance Cardiography. J. Electr. Bioimpedance 2013, 4, 45–50. [Google Scholar]
|Table 1. Signals recorded on the database.|
|Signal||Site or Configuration|
|Electrocardiogram||In thoracic region, in an axial plane|
|Thoracic Electrical Bioimpedance||Transthoracic Respiration|
|Galvanic Skin Response||Between Middle and Index Fingers||Back of Upper Arm|
|Temperature||Tip of Ring Finger||Back of Upper Arm|
|Speech||Built-in Microphone on the Smartphone|
|Table 2. Signals recorded on the database.|
|Signal||Nr of Extracted Features||Selected Features||Error Rate|
|Thoracic Electrical Bioimpedance||244||20||24.40%|
|Galvanic Skin Response (Fingers)||80||40||48.48%|
|Galvanic Skin Response (Arm)||80||20||57.85%|
Note: Ordered according to classification performance.
© 2014 by the authors; licensee MDPI, Basel, Switzerland. This article is an open access article distributed under the terms and conditions of the Creative Commons Attribution license ( http://creativecommons.org/licenses/by/3.0/).