Open AccessThis article is
- freely available
Atmospheric Correction and Vicarious Calibration of Oceansat-1 Ocean Color Monitor (OCM) Data in Coastal Case 2 Waters
Department of Biology, Jackson State University, Jackson, MS 39217, USA
Department of Oceanography and Coastal Sciences, Louisiana State University, Baton Rouge, LA 70803, USA
Department of Geosciences, Northern Gulf Institute, and Geosystems Research Institute, Mississippi State University, Mississippi State, MS 39762, USA
Code 7331, Naval Research Laboratory, Stennis Space Center, MS 39529, USA
* Author to whom correspondence should be addressed.
Received: 26 April 2012; in revised form: 11 May 2012 / Accepted: 31 May 2012 / Published: 8 June 2012
Abstract: The Ocean Color Monitor (OCM) provides radiance measurements in eight visible and near-infrared bands, similar to the Sea-viewing Wide Field-of-View Sensor (SeaWiFS) but with higher spatial resolution. For small- to moderate-sized coastal lakes and estuaries, where the 1 × 1 km spatial resolution of SeaWiFS is inadequate, the OCM provides a good alternative because of its higher spatial resolution (240 × 360 m) and an exact repeat coverage of every two days. This paper describes a detailed step-by-step atmospheric correction procedure for OCM data applicable to coastal Case 2 waters. This development was necessary as accurate results could not be obtained for our Case 2 water study area in coastal Louisiana with OCM data by using existing atmospheric correction software packages. In addition, since OCM-retrieved radiances were abnormally low in the blue wavelength region, a vicarious calibration procedure was developed. The results of our combined vicarious calibration and atmospheric correction procedure for OCM data were compared with the results from the SeaWiFS Data Analysis System (SeaDAS) software package outputs for SeaWiFS and OCM data. For Case 1 waters, our results matched closely with SeaDAS results. For Case 2 waters, our results demonstrated closure with in situ radiometric measurements, while SeaDAS produced negative normalized water leaving radiance (nLw) and remote sensing reflectance (Rrs). In summary, our procedure resulted in valid nLw and Rrs values for Case 2 waters using OCM data, providing a reliable method for retrieving useful nLw and Rrs values which can be used to develop ocean color algorithms for in-water substances (e.g., pigments, suspended sediments, chromophoric dissolved organic matter, etc.) at relatively high spatial resolution in regions where other software packages and sensors such as SeaWiFS and Moderate Resolution Imaging Spectrometer (MODIS) have proven unsuccessful. The method described here can be applied to other sensors such as OCM-2 or other Case 2 water areas.
Keywords: OCM; SeaWiFS; atmospheric correction; vicarious calibration; Case 2 water
Article StatisticsClick here to load and display the download statistics.
Notes: Multiple requests from the same IP address are counted as one view.
Cite This Article
MDPI and ACS Style
Dash, P.; Walker, N.; Mishra, D.; D’Sa, E.; Ladner, S. Atmospheric Correction and Vicarious Calibration of Oceansat-1 Ocean Color Monitor (OCM) Data in Coastal Case 2 Waters. Remote Sens. 2012, 4, 1716-1740.
Dash P, Walker N, Mishra D, D’Sa E, Ladner S. Atmospheric Correction and Vicarious Calibration of Oceansat-1 Ocean Color Monitor (OCM) Data in Coastal Case 2 Waters. Remote Sensing. 2012; 4(6):1716-1740.
Dash, Padmanava; Walker, Nan; Mishra, Deepak; D’Sa, Eurico; Ladner, Sherwin. 2012. "Atmospheric Correction and Vicarious Calibration of Oceansat-1 Ocean Color Monitor (OCM) Data in Coastal Case 2 Waters." Remote Sens. 4, no. 6: 1716-1740.