Evaluation of Urban Bioclimatic Measurements towards an Easier and more Affordable Method of Instrumental Monitoring †

: Thermal comfort is a key aspect of optimal conditions in urban public spaces. Air temperature, relative humidity, wind speed, and globe temperature measurements are critical components of bioclimatic research in the broader scientiﬁc ﬁeld of urban space quality assessment. The evaluation of thermal comfort in public spaces frequently requires ﬁeld measurements over long periods and at multiple sites at the same time. This can be challenging on a qualitative and quantitative level. Finding the most accurate way to collect such data in an accessible and manageable way is crucial in the context of an urban ﬁeld study. Data from various instruments were evaluated and statistically compared in order to assess possible instrument synergy or even similarities that would allow a transition to a simpliﬁed way of measuring these determinants of thermal comfort.


Introduction
Comfort, particularly thermal comfort, is a fundamental feature of urban public spaces that has a direct impact on the quality of life of city people.External thermal comfort in urban public places is strongly tied to people's well-being, especially in hot regions where heat stress conditions and the impact on residents' health are significant [1].Thermal comfort is frequently measured using a variety of bioclimatic parameters such as air temperature (Ta), relative humidity (RH), wind speed (WS), and globe temperature (Tg) [2][3][4].These data are critical not only for understanding and analyzing the thermal environment of public areas, but also for identifying infections and developing ways to improve thermal comfort for users.Given the process's complexity, both qualitatively and computationally, collecting bioclimatic data in metropolitan public places can be challenging [5,6].Field measurements may be necessary over long periods of time and at several sites at the same time, making data collecting an intensive and time-consuming process.Furthermore, the cost of the necessary equipment and regular maintenance might be prohibitively expensive for many research studies and urban environmental exploration and planning efforts, restricting the range and scale of such projects.As a result, an easier and more accessible method of instrumental monitoring of bioclimatic parameters in urban public places is required.In this study, we expect to identify possible synergies or similarities between different bioclimatic instruments of varying cost (which affects the economic viability of each project) and physical size (which affects the ease of implementing measurements) in order to create a simplified approach to data collection.The aim of this study is to contribute to the creation of a more accessible and viable technique of evaluating the bioclimatic determinants of thermal comfort, with the goal of enhancing the convenience and accessibility of researching the urban environment.

Methodology
In order to compare the various instruments and determine the possible use of more portable and affordable solutions for bioclimatological research, three different setups were employed.A micrometeorological station for reference, a portable heat stress tracker (Kestrel 5400) measuring Ta (TaK), RH (RHK), Tg (TgK) and WS (WSK) and a shielded portable thermohygrometer (HOBO MX2302A) measuring Ta (TaH) and RH (RHH).The reference station consisted of a Delta-T GP2 logger connected to a shielded ADCON TR1 thermohygrometer measuring Ta (TaS) and RH (RHS), an Atmos 22 sonic anemometer measuring WS (WSS) and a PT100 thermometer inserted into a PVC 40 mm diameter sphere painted grey (RAL 7001), measuring Tg (TgS).All sensors were installed 1.1 m from the ground in an open location (Figure 1) on the premises of AUA and measurements were logged in 10 min intervals from 9.00 AM to 17.00 PM on 8 May 2023.Specifications of instruments are presented in table (Table 1).Statistical analysis was conducted using SPSS 26 and Jamovi 2.3.21software.Correlation of measurements was estimated according to Spearman's rank-order and Pearson product-moment correlation methodologies.Descriptive statistics for all measurements plus the difference between the reference station and the two other instruments are presented in Figure 2 and Tables 2-5.

Methodology
In order to compare the various instruments and determine the possible use of more portable and affordable solutions for bioclimatological research, three different setups were employed.A micrometeorological station for reference, a portable heat stress tracker (Kestrel 5400) measuring Ta (TaK), RH (RHK), Tg (TgK) and WS (WSK) and a shielded portable thermohygrometer (HOBO MX2302A) measuring Ta (TaH) and RH (RHH).The reference station consisted of a Delta-T GP2 logger connected to a shielded ADCON TR1 thermohygrometer measuring Ta (TaS) and RH (RHS), an Atmos 22 sonic anemometer measuring WS (WSS) and a PT100 thermometer inserted into a PVC 40 mm diameter sphere painted grey (RAL 7001), measuring Tg (TgS).All sensors were installed 1.1 m from the ground in an open location (Figure 1) on the premises of AUA and measurements were logged in 10 min intervals from 9.00 AM to 17.00 PM on 8 May 2023.Specifications of instruments are presented in table (Table 1).Statistical analysis was conducted using SPSS 26 and Jamovi 2.3.21software.Correlation of measurements was estimated according to Spearman's rank-order and Pearson product-moment correlation methodologies.Descriptive statistics for all measurements plus the difference between the reference station and the two other instruments are presented in Figure 2 and Tables 2-5.

Results
The Ta values had a similar mean ∆Ta for TaH and TaK compared with the Ta values logged in the reference station (−0.92 K).The TaK values had a bigger range compared with the other two instruments.This could be due to the fact that the thermometer in Kestrel 5400 is exposed in contrast with the other two instruments.The RHK values followed the RHS values in greater detail compared with the RHH values.The mean WSK values were similar to the WSS values (mean ∆WS = 0.02%).The mean ∆Tg value between TgK and TgS was −4.48 K but the two instruments' responses to solar radiation appear correlated.Examining the TaS, TaK and TaH values, there was a statistically significant positive correlation.Between TaS and TaK; rs = 0.818 (p < 0.005), between TaS and TaH, rs = 0.982 (p < 0.005); and between TaK and TaH, rs = 0.833 (p < 0.005).Examining RHS, RHK and RHH values, there was a statistically significant positive correlation as follows: between RHS and RHK, rs = 0.858 (p < 0.005); between RHS and RHH, rs = 0.912 (p < 0.005); and between RHK and RHH, rs = 0.854 (p < 0.005).There was also a statistically significant positive correlation between WSS and WSK (rs = 0.6, p < 0.005) and between TgS and TgK (rs = 0.922, p < 0.005).
As a more affordable and portable solution compared to a full micrometeorological station, the HOBO MX2302A appeared to give better measurements for Ta and RH compared to Kestrel 5400.The WS values from Kestrel 5400 were comparable to the WS values logged in the reference station but must be used with caution, especially due to the lack of more detailed logging options in Kestrel 5400 and the different operational principal of the two anemometers (sonic vs. vane).However, the Tg values logged in Kestrel 5400 in comparison with Tg values from the reference station appeared to be significantly correlated.

Conclusions
In conclusion, the first findings indicate that the use of portable and less expensive instruments may be a feasible alternative to a full scientific micrometeorological station in bioclimatological research.However, further study is needed to properly comprehend the

Figure 1 .
Figure 1.Photograph captured in the measuring field.

Figure 1 .
Figure 1.Photograph captured in the measuring field.