Evaluation of the HemoCue ® WBC System as a Point of Care Diagnostic Tool for White Blood Cell Quantiﬁcation in Pinnipeds

: Point of care (POC) hematology testing can be valuable in triage and ﬁeld settings. We assessed the accuracy between the HemoCue ® WBC system and two comparative analyzers, as well as the precision of the HemoCue ® WBC system in ﬁve different pinniped species: Zalophus californianus , Arctocephalus townsendi , Callorhinus urcinus , Phoca vitulina , and Mirounga angustirostris for white blood cell (WBC) quantiﬁcation. In Zalophus ( n = 164; 106 from U.S. Navy Marine Mammal Program (Navy); 58 from The Marine Mammal Center (TMMC)), the HemoCue ® was compared to two hematology analyzers, Sysmex Xe-5000 and Vet ABC Plus. In Phoca ( n = 50; TMMC), Callorhinus ( n = 29; TMMC), Arctocephalus ( n = 17; TMMC), and Mirounga ( n = 67; TMMC), the HemoCue ® was compared to Vet ABC Plus only. Bland–Altman and Passing–Bablok agreement of HemoCue ® with Sysmex Xe-5000 and Vet ABC Plus analyzers were good for Zalophus , Arctocephalus , Phoca , and Mirounga but marginal with Callorhinus ; bias = 0.56 × 10 9 /L ( Zalophus ; Navy), − 2.13 × 10 9 /L ( Zalophus ; TMMC), − 1.59 × 10 9 /L ( Arctocephalus ), − 2.48 × 10 9 /L 0.917 ( Phoca ), − 0.01 × 10 9 /L ( Mirounga ), and − 6.05 × 10 9 /L ( Callorhinus ). The coefﬁcient of variation from triplicate runs of samples were within acceptable limits for all species (2.50% ± 1.63 ( Zalophus ; TMMC), 3.09% ± 2.14 ( Arctocephalus ), 2.47% ± 1.35 ( Callorhinus ), 2.88% ± 1.75 ( Phoca ), and 3.44% ± 2.53 ( Mirounga )), respectively. The presence of nucleated red blood cells (nRBC; 1–37 nRBC/100 WBC) did not signiﬁcantly interfere with WBC counts in Zalophus , Callorhinus , and Phoca at the population level, but their presence should be evaluated at the individual level. The HemoCue ® provides an accurate method for WBC quantiﬁcation with WBC counts up to 30 × 10 9 /L (upper limit of linearity of the analyzer) in Zalophus , Arctocephalus , Phoca , and Mirounga , but is less accurate in Callorhinus , and showed good precision in all species.


Introduction
Point of care (POC) analyzers offer rapid adjunct diagnostic testing to supplement standard hematology in both human and veterinary medicine. White blood cell (WBC) quantification is a valuable diagnostic marker for the assessment of inflammation and leukogram responses, and the use of POC WBC count analyzers has been implemented in human out-patient and critical care departments, facilitating patient flow, decreasing the length of stay in the emergency department, and aiding in triage [1][2][3]. Compared to commercial laboratory analyzers in human laboratory diagnostics, POC WBC count analyzers are cost-effective in resource-limited settings, and provide rapid, bed-side results allowing fast initiation of treatment [1].
In veterinary medicine, the majority of the information regarding POC tests revolves around biochemical analytes (e.g., glucose and lactate) and blood gas analysis rather than hematology [1,4,5]. There are many potential applications for POC WBC count analyzers in domestic and non-domestic species. In a managed care setting and particularly for patients requiring sedation or anesthesia for diagnostics, treatments, and handling, the ability to determine WBC counts patient-side facilitates rapid initiation of treatment, potentially requiring fewer sedation and/or anesthetic events, and can help guide clinical decisionmaking and management of critically ill patients. In field settings, many health assessment studies in free-ranging wildlife species aim to determine the impact of different diseases on populations but are limited by the lack of capacity to perform a WBC count to complement a WBC differential and often must rely on blood film evaluation only [6,7]. Additionally, a POC WBC count analyzer can provide individual patient-side automated hematology in rescue situations or managed care that may otherwise be impossible.
The HemoCue ® WBC system (HemoCue America, Brea, CA 92821, USA) is a lightweight, portable analyzer that is battery powered or can be plugged into an AC 6-volt adapter [8]. The HemoCue ® requires a small amount of whole blood (approximately 10 µL), uses a hemolyzing agent to lyse the red blood cells, and a staining agent to color the white blood cells. A photomicroscope is used to capture an image of the stained WBC and the number of cells is counted by image analysis. The WBC count is then displayed on the liquid crystal display screen after approximately three minutes [8]. The presence of platelets, platelet clumps, lipemia, reticulocytes, and immature leukocytes did not interfere with HemoCue ® results in humans [8]. However, samples with >2% nucleated red blood cells (nRBC) may give a falsely elevated WBC count [8]. The analyzer should be stored at temperatures between 0 and 50 • C (32-122 • F) with less than 90% non-condensing humidity and operated in temperatures between 15 and 35 • C (59-95 • F) and less than 90% non-condensing humidity [8]. The microcuvettes must be stored at temperatures between 15 and 35 • C (59-95 • F) and less than 90% non-condensing humidity [8]. The system requires very little technical skill and minimal specialized equipment to operate.
Validation studies of the HemoCue ® WBC and HemoCue ® WBC DIFF differential portable analyzers in humans have demonstrated accuracy for rapid diagnosis of abnormal WBC [3,9,10]. A single study in veterinary medicine showed promising results for the HemoCue ® WBC system as a rapid POC test to diagnose abnormal WBC concentrations in cats, horses, cows, and dogs [11]. This study aimed to assess the accuracy (or closeness of agreement) between the HemoCue ® WBC system and two comparative analyzers, as well as the precision of the HemoCue ® WBC system in five different pinniped species: California sea lions (Zalophus californianus), Guadalupe fur seals (Arctocephalus townsendi), Northern fur seals (Callorhinus urcinus), Pacific harbor seals (Phoca vitulina), and Northern elephant seals (Mirounga angustirostris).

Animals
Whole blood samples were collected from animals in two separate locations, the U.S. Navy Marine Mammal Program (Navy) and The Marine Mammal Center (TMMC). The Navy houses and cares for a population of Zalophus in San Diego Bay (CA, USA). The Marine Mammal Center is a rescue and rehabilitation facility for marine mammals in Sausalito (CA, USA).

U.S. Navy Marine Mammal Program
Zalophus samples were the only samples collected from the Navy and included a total of a 106 blood samples from 28 individual Zalophus ranging in age from 4 to 32 years old.

The Marine Mammal Center
Samples from stranded free-ranging Zalophus, Arctocephalus, Callorhinus, Phoca, and Mirounga were collected and analyzed at TMMC and age classes were determined based on dentition and body morphometrics in all species [12][13][14]. Total sample numbers and animals include 58 blood samples from 38 individual Zalophus ranging in age class from pup to adult, 17 blood samples from 10 individual Arctocephalus, all of which were pups, 29 blood samples from 16 individual Callorhinus, all of which were pups, 50 blood samples from 25 individual Phoca, all of which were pups, and 66 blood samples from 39 different Mirounga ranging in age class from pup to yearling. Samples collected from Navy Zalophus were collected opportunistically from an overall clinically healthy population of Zalophus, during routine care under the approved U.S. Code, Title 10, USC 7524. Blood was obtained either from the caudal gluteal vein directly into Becton Dickinson (BD) vacutainer tubes containing potassium-ethylenediaminetetraacetic acid (K-EDTA; BD, Franklin Lakes, NJ, USA) using a 20-gauge 1 2 inch BD vacutainer needle attached to a vacutainer holder or a 21-gauge 2 inch needle attached to a BD vacutainer Luer adapter and vacutainer holder (BD, Franklin Lakes, NJ, USA); or from the superficial antebrachial vein using a 21-gauge 3 4 inch butterfly needle and syringe, then placed immediately into K-EDTA vacutainer tubes. Gentle inversion of the filled vacutainer tubes was performed for a minimum of one minute prior to analysis for both methods.

The Marine Mammal Center
Samples were collected from free-ranging stranded Zalophus, Arctocephalus, Callorhinus, Phoca, and Mirounga admitted for rehabilitation at TMMC during routine admission, followup, or release examinations under the Stranding Agreement between the National Oceanic and Atmospheric Administration's National Marine Fisheries Service West Coast Region and The Marine Mammal Center. Blood was collected from Zalophus, Arctocephalus, and Callorhinus via the caudal gluteal or subclavian vein directly into K-EDTA vacutainer tubes (Oakville, ON LH6 6R5, Canada) using a vacutainer adapter set-up as described above. Blood was collected from Phoca and Mirounga via the epidural intervertebral sinus directly into K-EDTA vacutainer tubes (Oakville, ON, Canada) using the same vacutainer adapter set-up. Gentle inversion of the filled vacutainer tubes was performed for a minimum of one minute immediately prior to analysis for both methods.

U.S. Navy Marine Mammal Program
Blood analysis was performed on well-mixed whole blood in K-EDTA via the Sysmex Xe-5000 (Sysmex Nordic APS Filial Sverige, Marios Gata 13, S-434 37 Kungsbacka, Sweden) comparative instrument at the Naval Medical Center San Diego in single replicate within five hours of collection. White blood cell counts were performed on whole blood samples in K-EDTA by the HemoCue ® WBC system candidate instrument in single replicate within one hour of collection. The time span between analyses on the Sysmex Xe-5000 and the HemoCue ® ranged from two to four hours for each sample. Blood films prepared from each sample were reviewed to assess the presence of nRBC, red blood cell morphology, and WBC differential and morphology. The WBC counts from both the Sysmex Xe-5000 analyzer and the HemoCue ® WBC system were corrected for nRBC when present. Briefly, the HemoCue ® analysis process included placing a single drop of whole blood on a sterile flat surface and was taken up by capillary action into the HemoCue ® microcuvette. Excess blood was gently wiped away from the cuvette surface prior to being placed into the HemoCue ® analyzer. Results were read immediately and recorded.

The Marine Mammal Center
Blood analysis was performed on well-mixed whole blood in K-EDTA by the Vet ABC Plus comparative analyzer (Vet ABC Plus; SCIL Vet America, Gurnee, IL 60031, USA) on the dog setting in single replicate within two hours of collection [15,16]. White blood cell counts were performed on whole blood in K-EDTA by the HemoCue ® WBC system candidate instrument in triple replicate for each species as described above within two hours of collection. The time span between analyses on the Vet ABC Plus and HemoCue ® was less than one hour for each sample. Blood films from each blood sample were reviewed to assess the presence of nRBC, red blood cell morphology, and WBC differential and morphology. White blood cell counts from the Vet ABC Plus analyzer and the HemoCue ® WBC system were corrected for nRBC when present.

Reporting and Statistical Analysis
All analyses were performed using R version 3.6.1 [17] and R Studio v 1.2.5019 [18] statistical analysis software. The HemoCue ® WBC system corrected WBC count was compared to the comparative instrument's corrected WBC count using Bland-Altman and Passing-Bablok methods and the R package mcr [19][20][21]. Normality of the distribution of the differences between the reference instrument WBC counts and the HemoCue ® WBC system WBC counts was confirmed prior to performing analyses. The Bland-Altman plot bias was calculated as the average difference between WBC counts produced by the comparative hematology analyzers and by the HemoCue ® WBC system. The Bland-Altman plot limits of agreement were calculated using the equation 1.96 x standard deviation of the average difference between each comparative method and HemoCue ® WBC system results in each species [19,22]. To rule out constant and proportional differences between the two methods, the Passing-Bablok intercept and slope 95% confidence intervals must include zero and one, respectively [20]. When interpreting Pearson's correlation coefficient r, a very strong association is considered between 0.8 and 0.9, a moderate association is considered between 0.6 and 0.7, a fair association is considered between 0.3 and 0.5, and a poor association is considered between 0.1 and 0.2 [23]. Of note, the upper limit of linearity of the HemoCue ® is 30 × 10 9 /L and a WBC count greater than or equal to 30 × 10 9 /L will yield a HemoCue ® result of "HHH" on the screen. Five blood samples from Mirounga yielded results of "HHH" on the HemoCue ® and all five samples had a WBC count greater than or equal to 30 × 10 9 /L via the Vet ABC Plus (WBC counts = 53.4, 39.8, 33.2, 30.2, and 80.0 × 10 9 /L) comparative instrument. These five samples were removed from the dataset. The lower limit of linearity of the HemoCue ® is 0.3 × 10 9 /L in which a WBC count less than or equal to 0.3 × 10 9 /L will yield a HemoCue ® result of "LLL" on the screen. There were no blood samples in this analysis that yielded results of "LLL".
After the first Bland-Altman and Passing-Bablok analyses were performed on all species, samples that contained nRBC were removed from the dataset and the Bland-Altman and Passing-Bablok analyses were performed again. To assess possible interference caused by nRBC, a Welch's two sample t-test was performed to compare significant differences in the biases. Nucleated red blood cells were present in samples from two Phoca and four Callorhinus at The Marine Mammal Center and eight Zalophus at the Navy.
The precision of the HemoCue ® WBC system was assessed for all five species at The Marine Mammal Center using triplicate tests performed on the same blood sample. The coefficient of variation was calculated for each triplicate group by dividing the standard deviation of the three measurements by the mean of the three measurements. The average coefficient of variation, standard deviation, and range were calculated. The coefficient of variation can be expressed as a percentage where <10% is considered to be very good, 10-20% is considered to be good, 20-30% is acceptable, and >30% is unacceptable [24]. In addition, the total observed error (TEobs) for the WBC count was calculated for all five species in this study using the equation: TEobs = absolute bias% + 2CV and compared to the total allowable error (TEa) recommended by guidelines set forth by the American Society for Veterinary Clinical Pathology (ASVCP) [25].

Results
Overall, the HemoCue ® performed well over a large range of WBC counts (2.5−30 × 10 9 /L) in four out of five different species of pinnipeds (Table 1).

Zalophus-Navy
Both the Bland-Altman analysis and Passing-Bablok regression analysis showed strong correlations between the comparative instrument and the HemoCue ® WBC system ( Table 1). The bias determined from the Bland-Altman plot was 0.56 × 10 9 /L. Of the 106 blood samples analyzed, seven data points fell outside of the upper and lower limits of agreement ( Figure 1A). There was no significant proportional bias between the two methods; however, a constant difference was observed. The HemoCue ® overall had WBC counts slightly lower than the Sysmex Xe-5000 WBC count (Figure 2A). Nine samples with nRBC ranging from 1 to 3 nRBC/100 WBC were identified by blood film evaluation. The WBC count from both the HemoCue ® WBC system and the Sysmex Xe-5000 were corrected for nRBC prior to performing the Bland-Altman and Passing-Bablok analyses. Additionally, the samples with nRBC were analyzed on an individual basis. The differences between the Sysmex Xe-5000 WBC counts and the HemoCue ® WBC counts for the nine samples that contained nRBC prior to adjusting the HemoCue ® WBC counts were: 0.7, 0.7, 0.5, 0.4, 0.8, 0.3, 0.9, 1.1, and 1.2 × 10 9 /L. After adjusting the HemoCue ® WBC counts for the presence of nRBC, the differences between the Sysmex Xe-5000 and the HemoCue ® WBC counts were approximately the same: 0.7, 0.7, 0.5, 0.4, 0.8, 0.4, 0.9, 1.1, and 1.2 × 10 9 /L, respectively. The Navy samples were analyzed in single runs, so a coefficient of variation was not able to be performed at this location.

Zalophus-The Marine Mammal Center
A strong correlation between the Vet ABC Plus analyzer and the HemoCue ® was observed ( Table 1). The bias determined from the Bland-Altman plot was −2.13 × 10 9 /L. Of the 58 blood samples analyzed, two data points fell outside of the upper and lower limits of agreement ( Figure 1B). There was no significant constant difference between the two methods; however, a proportional bias was observed. As the magnitude of WBC counts increased, the difference between the Vet ABC Plus analyzer and the HemoCue ® increased ( Figure 2B). Nucleated RBC were absent in all samples from this group.
All 58 Zalophus blood samples were run in triplicate to assess the precision of the HemoCue ® . The HemoCue ® was precise in Zalophus with a coefficient of variation of 2.5%. The TEobs for Zalophus was 13.66%.

Arctocephalus-The Marine Mammal Center
High agreement between the HemoCue ® and the Vet ABC Plus was observed in Arctocephalus ( Table 1). The bias determined from the Bland-Altman plot was −1.59 × 10 9 /L. All 17 blood sample data points were within the upper and lower limits of agreement ( Figure 1C). There was no significant constant difference between the two methods; however, a proportional bias was observed. Similar to what was observed in Zalophus, as the magnitude of WBC counts increased, the difference between the Vet ABC Plus and the HemoCue ® increased ( Figure 2C). The sample size for Arctocephalus was the smallest of all five species and likely contributed to the wide CI observed for the Passing-Bablok intercept. Nucleated RBC were absent in all samples from this group.
Five Arctocephalus blood samples were run in triplicate to assess the precision of the HemoCue ® . The HemoCue ® was precise in Arctocephalus with a coefficient of variation of 3.09% and the TEobs for Arctocephalus was 18.48%.  • Third bullet.
Numbered lists can be added as follows: 1. First item; 2. Second item; 3. Third item.
The text continues here.

Callorhinus-The Marine Mammal Center
Marginal agreement between the HemoCue ® and the Vet ABC Plus was observed in Callorhinus ( Table 1). The bias determined from the Bland-Altman plot was −5.67 × 10 9 /L. Of the 29 blood samples, two data points fell outside of the upper and lower limits of agreement ( Figure 1D). There was no significant constant difference between the two methods; however, a proportional difference was observed. Similar to what was observed in Zalophus and Arctocephalus, as the magnitude of WBC counts increased, the difference between the Vet ABC Plus and the HemoCue ® increased ( Figure 2D). Four samples with nRBC ranging from 5 to 37 nRBC/100 WBC were identified on blood film. The WBC count from both the HemoCue ® WBC system and the Vet ABC Plus were corrected for nRBC prior to performing the Bland-Altman and Passing-Bablok analyses. Additionally, the samples with nRBC were analyzed on an individual basis. The differences between the Vet ABC Plus WBC counts and the HemoCue ® WBC counts for the four samples that contained nRBC prior to adjusting the HemoCue ® WBC counts were: 2.0, 10.4, 11.1, and 10.3 × 10 9 /L. After adjusting the HemoCue ® WBC counts for the presence of nRBC, the differences between the Vet ABC Plus and the HemoCue ® WBC counts were slightly improved, but overall, still relatively large in magnitude at: 4.9, 8.3, 9.8, and 9.4 × 10 9 /L, respectively.
Twenty-eight Callorhinus blood samples were run in triplicate to assess the precision of the HemoCue ® . The HemoCue ® was precise in Callorhinus with a coefficient of variation of 2.47%. Total observed error for Callorhinus was 41.40%.

Phoca-The Marine Mammal Center
High agreement between the HemoCue ® and the Vet ABC Plus was observed in Phoca ( Table 1). The bias determined from the Bland-Altman plot was −2.34 × 10 9 /L. Of the 50 blood samples, three data points fell outside of the upper and lower limits of agreement ( Figure 1E). There were both constant and proportional differences between the two testing methods in Phoca. Similar to what was observed in Zalophus, Arctocephalus, and Callorhinus, as the magnitude of WBC counts increased, the difference between the Vet ABC Plus and the HemoCue ® increased ( Figure 2E). Two samples with nRBC, 8 and 28 nRBC/100 WBC, were identified on blood film. The WBC count from both the HemoCue ® WBC system and the Vet ABC Plus were corrected for nRBC prior to performing the Bland-Altman and Passing-Bablok analyses. Additionally, the samples with nRBC were analyzed on an individual basis. The differences between the Vet ABC Plus WBC counts and the HemoCue ® WBC counts for the two samples that contained nRBC prior to adjusting the HemoCue ® WBC counts were 5.7 and 8.9 × 10 9 /L. After adjusting the HemoCue ® WBC counts for the presence of nRBC, the differences between the Vet ABC Plus and the HemoCue ® WBC counts were improved at 1.1 and 6.7 × 10 9 /L, respectively.
Thirty-two Phoca blood samples were run in triplicate to assess the precision of the HemoCue ® . The HemoCue ® was precise in Phoca with a coefficient of variation of 2.88% and TEobs for Phoca was 17.35%.

Mirounga-The Marine Mammal Center
High agreement between the HemoCue ® and the Vet ABC Plus was observed in Mirounga ( Table 1). The bias determined from the Bland-Altman plot was −0.01 × 10 9 /L. Of the 66 blood samples analyzed, two data points fell outside of the upper and lower limits of agreement ( Figure 1F). In Mirounga, there was a constant but no proportional bias between the Vet ABC Plus and the HemoCue ® ( Figure 2F) and likely due to a single outlier in which the difference between the comparative instrument and the HemoCue ® was 16.56 × 10 9 /L. Nucleated RBC were absent in all samples from this group.
Fifty-six Mirounga blood samples were run in triplicate to assess the precision of the HemoCue ® for Mirounga samples. The HemoCue ® was precise in Mirounga with a coefficient of variation of 3.44%. The TEobs for Mirounga was 6.79%.

Discussion
This study demonstrates the accuracy and precision of the HemoCue ® WBC system in five species of pinnipeds: Zalophus, Arctocephalus, Callorhinus, Phoca, and Mirounga over a wide range of WBC counts up to 30 × 10 9 /L in comparison to two comparative instruments (Vet ABC Plus and Sysmex Xe-5000). Based on the results of this study, this portable and easy-to-use POC WBC quantification device offers an additional diagnostic tool and potential applications to zoological and wildlife medicine, including but not limited to health assessments in the field, patient-side diagnosis of abnormal WBC counts leading to more rapid and guided treatment decisions, and in clinical facilities limited by financial resources.
When compared to the Vet ABC Plus, the HemoCue ® had marginal to good accuracy in all five species with the best accuracy observed in Mirounga and the lowest accuracy observed in Callorhinus. Very good accuracy was observed in Arctocephalus; however, a very wide confidence interval in the Passing-Bablok regression was observed, likely due to the small sample size. When compared to the Sysmex Xe-5000 in Zalophus only, the HemoCue ® also demonstrated very good accuracy. The variability observed between the two comparative analyzers and the HemoCue ® in this study was similar to what was observed by Riond et al. (2012), where WBC counts for feline, equine, and bovine species were more accurate than for dogs which showed marginal accuracy [11]. In this study, the HemoCue ® demonstrated very good precision in all five species of pinnipeds.
It is important to note that the HemoCue ® WBC system, despite showing good accuracy, did show constant and/or proportional differences in comparison to both comparative instruments in all five species in this study. On average, the HemoCue ® tended to underestimate WBC concentrations compared to the Sysmex Xe-5000 and overestimate WBC counts compared to the Vet ABC Plus. Results observed in rehabilitating Zalophus at TMMC in which overall WBC counts were higher compared to the overall healthy Navy Zalophus, showed that as the magnitude of WBC counts increased, the HemoCue ® became less accurate when compared to the Vet ABC Plus. Results from Arctocephalus and Callorhinus at TMMC showed similar results to Zalophus in which an increase in magnitude of WBC counts corresponded to less accurate HemoCue ® results as compared to the Vet ABC Plus. Results from Phoca and Mirounga at TMMC showed a constant difference. In Phoca, the HemoCue ® on average estimated the WBC count 2.5 × 10 9 /L greater than the results of the Vet ABC Plus. However, the magnitude of WBC counts did not affect the accuracy of the HemoCue ® in Phoca as it did in Zalophus, Arctocephalus, and Callorhinus. In Mirounga, the constant bias was likely due to a single outlier in which the difference between the Vet ABC Plus analyzer and the HemoCue ® was 16.56 × 10 9 /L, a difference that was three times greater than any other analyzed sample and from a patient with a markedly elevated WBC count of 43.6 × 10 9 /L. The ASVCP Quality Assurance and Laboratory Standards (QALS) accepts TEa for WBC counts at a reference laboratory of 15% and at an in-clinic laboratory of 20% [25]. For Zalophus, Arctocephalus, Phoca, and Mirounga, the TEobs was less than the TEa for in-clinic laboratories (13.66%, 18.48%, 17.35%, and 6.79%, respectively), complementing observations by Bland-Altman plots and Passing-Bablok analyses. However, for Callorhinus, TEobs (41.40%) was greater than the recommended TEa for in-clinic laboratories, complementing what was observed in the Bland-Altman plots and Passing-Bablok analysis and indicating the HemoCue ® requires more testing before it can confidently be used as a POC WBC device for Callorhinus. The reason for the decreased HemoCue ® accuracy observed in this study in Callorhinus is unknown but may be due in part to the small sample size analyzed. The TEobs was unable to be calculated for Zalophus in comparison to the Sysmex Xe-5000 as the HemoCue ® samples were performed in single runs and a coefficient of variation was unable to be calculated.
A study performed in humans found that the presence of more than 2% circulating nRBC can cause inaccurate HemoCue ® results [3]. However, in the current study at the population level, the presence of nRBC (maximum nRBC/100 WBC = 37) did not change the observed constant or proportional biases. Overall, the sample sizes for each species were relatively small with few animals having nRBC present in each dataset. The small number of animals with nRBC may have been the reason why nRBC did not have a significant effect on the HemoCue ® accuracy in this study rather than a true lack of interference. At the individual level, after adjusting the HemoCue ® WBC counts for the presence of nRBC, the magnitude of the difference between the comparative analyzers and the HemoCue ® results improved, most notably for Phoca, but were still relatively large overall. Blood film review to identify nRBC and correct the WBC count should always be performed, especially with unexpected HemoCue ® results.
The HemoCue ® was demonstrated to be an accurate and precise diagnostic test for WBC quantification in Zalophus, Arctocephalus, Phoca, and Mirounga, and showed marginal accuracy but good precision in Callorhinus. In all cases, the WBC counts must be clinically interpreted in light of overall patient health status, findings from physical examination, and additional diagnostic tests, including blood film review (e.g., WBC estimate as control measure, white blood cell differential, and evaluation of blood cell morphology). If an unexpected result is obtained by the HemoCue ® , we recommend in addition to blood film review that at least one other method of hematological analysis be performed to corroborate the HemoCue ® , e.g., repeat HemoCue ® test or submission to a reference laboratory. Additionally, confirmation of results is highly recommended for animals that are documented or suspected to have an increased number of nRBC.
Based on the results observed in this study, the HemoCue ® WBC system is a promising tool that has the potential to estimate WBC concentrations accurately and precisely in Zalophus, Arctocephalus, Phoca, and Mirounga, but more testing is recommended for Callorhinus. The HemoCue ® WBC system unit costs approximately USD 2000.00 and each cuvette costs approximately USD 2.85. When combined with other analyses that are readily performed at low cost (e.g., packed cell volume, total protein, and manual differential), the HemoCue ® allows for a fairly comprehensive, rapid, and low-cost hematological assessment for four out of five pinniped species analyzed in this study.