Screen-Printed Electrodes: Promising Paper and Wearable Transducers for (Bio)Sensing

Screen-printing technology has revolutionized many fields, including that of electrochemical biosensing. Due to their current relevance, this review, unlike other papers, discusses the relevant aspects of electrochemical biosensors manufactured using this technology in connection to both paper substrates and wearable formats. The main trends, advances, and opportunities provided by these types of devices, with particular attention to the environmental and biomedical fields, are addressed along with illustrative fundamentals and applications of selected representative approaches from the recent literature. The main challenges and future directions to tackle in this research area are also pointed out.

, anions [13], and gases [14,15]. For instance, the electrocatalytic activity of polyoxymetalates (POMs) toward the electrochemical reduction of chlorate was employed to prepare a vanadiumcontaining POM ([PMO11VO40] 5− ) for the determination of ClO3 − in soils. Just like in other paper-based designs, the SPCE brought the electrodes and the paper together to create the appropriate cell volume for the electrolyte solution, resulting in a thin layer cell for electrochemical detection [10]. Moreover, in this particular application, paper also acted as a filter for soil analysis. Using chronocoulometry, a LOD of 0.31 mg mL −1 ClO3 − was achieved. Using a similar configuration, a disposable gas-sensing paper-based device (gPAD) was fabricated in origami design, integrating in a single device activated carbon as the gas adsorbent and the electrochemical detection consisting of a screen-printed graphene electrode modified with copper nanoparticles. Both NO and NO2 (as NOx) were detected with the same current responses measured by differential pulse voltammetry (DPV) achieving LODs of 0.23 vppm and 0.03 vppm with exposure times of 25 min and 1 h, respectively. Relative standard deviation (RSD) values less than 5.1% (n=7 devices) for 25, 75, and 125 vppm NO2 were reported, and the gPAD was applied to detect NOx in air and exhaust gases from cars [14].
The nerve agent VX gas is prohibited as a chemical warfare agent. Since it cannot be used in research experiments, dimethyl methylphosphonate (DMMP) is utilized as a model. A conductive paper prepared with poly(aniline) (PANI) nanofiber and graphene sheet was used to detect DMMP at parts per billion within few seconds. The intermolecular forces between the cellulosic paper and the conductive additives were improved by using the copolymer poly(vinylbutyral-co-vinyl-alcoholco-vinyl acetate) ( Figure 1). The resulting sensor exhibited a detectable level of 3 ppb and a response time of 2 s [15]. In this field, strategies consisting of electrodes screen-printed onto a filter paper support allows enzymes and other reagents to be pre-loaded into the cellulose network. An illustrative example is the paper-based wearable electrode constructed for the detection of mustard agent, one of the most dangerous chemical warfare agents (CWAs). It is an origami-like device where the detection is based on the inhibitory effects of the analyte toward the enzyme choline oxidase. The amperometric responses were measured at a carbon black/Prussian blue nanocomposite distributed on the electrode surface and profiting its electrocatalytic activity for H2O2 reduction. A Keithley 2400 current source meter (Keithley Co., Cleveland, OH, USA) was used to measure the electrical properties and sensing performance of the conductive papers. The LOD attained in the aerosol phase was 0.019 g min m −3 [11]. A carbon-nanotubes-based ink prepared with sodium dodecyl sulfate (SDS) and chitosan (CS) absorbed onto cellulose fibers was used to prepare a paper electrode for the determination of Pb 2+ trace levels in water samples. Square-wave anodic stripping voltammetry (SWASV) using a bismuth A carbon-nanotubes-based ink prepared with sodium dodecyl sulfate (SDS) and chitosan (CS) absorbed onto cellulose fibers was used to prepare a paper electrode for the determination of Pb 2+ trace levels in water samples. Square-wave anodic stripping voltammetry (SWASV) using a bismuth film prepared by in situ plating of Bi into the CNT-CS-SDS paper electrodes made the determination of Pb 2+ in the presence of Bi (10-200 ppb) with a LOD of 6.74 ppb possible [12]. Furthermore, a paper-based, disposable electrochemical platform was developed for the determination of nitrite. Graphene nanosheets and gold nanoparticles were assembled to form a three-dimensional structure onto mixed cellulose ester (MCE) filter papers, leading to thin layer rather than planar diffusion behavior of nitrite at the paper-based electrode. The resulting platform provided larger currents compared to conventional gold or glassy carbon electrodes and, consequently, allowed an improved sensitivity. Importantly, this design effectively avoided the fouling arising from the adsorption of oxidation products thus allowing the determination of nitrite in environmental samples such as waters and industrial sewage. The calibration curve at +0.74 V covered a wide concentration range of 0.3-720 µM, and the LOD was 0.1 µM (S/N = 3) [13].
Contamination of water by microbial pathogens leading to water-borne diseases requires strict controls in drinking water resources, particularly in poor regions, to reduce mortality incidence. Although not included in Table 1, various methods related to the detection of microorganisms should be highlighted. For instance, a simple low-cost paper-based impedimetric sensor for the detection of bacteria in water was prepared using carbon electrodes screen-printed with a conductive ink onto a commercial hydrophobic paper. Concanavalin A covalently immobilized onto the carboxylated electrode surface was used as the biorecognition element due to its ability to selectively interact with mono-and oligo-saccharides on bacteria. In this method, the hydrophobicity of the cellulose paper used as a substrate prevented any unspecific adsorption. The biosensor was applied to bacterial cultures from sewage sludge that were grown in synthetic water, then filtered and enumerated for defining the stock solution. The calibration plot showed an increase in the charge transfer resistance (R CT ) over the 10 3 to 10 6 colony forming unit (CFU) mL −1 range, with an estimated LOD of 1.9×10 3 CFU mL −1 [16]. A fast-flow paper-based electrochemical sensor was developed by Channon et al. for the label-free detection of virus particles [17]. West Nile viruses were detected by electrochemical impedance spectroscopy using antibody functionalized Au microwires, achieving a LOD of 10.2 particles in 50 µL of cell culture media. The sensing approach is easily controllable by means of a smartphone and may presumably be applied to a range of biological targets. Toxins secreted from pathogens can also be detected in bacterial cultures using paper-based electrochemical sensors. This is the case of pyocyanin, a toxin solely produced by Pseudomonas aeruginosa, whose detection was performed using an inexpensive approach involving electrode printing of carbon ink on photo paper and square wave voltammetry [18]. It is worth noting that some authors have also exploited the bacterial enzyme activity (expression of β-glucosidase by Enterococcus spp. and the production of β-galactosidase and β-glucuronidase by E. coli) for their determination at paper-based electrochemical sensors [19].
The use of biological indicators to determine the toxicological effect of environment pollutants is an interesting research area where microorganisms combined with paper-based electrochemical sensors provide important advantages. In this context, a µPAD for highly integrated biotoxicity measurements was prepared involving screen-printing with conductive carbon ink and chromatographic paper. The µPAD contained three functional units for injection, separation, and detection zones with hydrophobic barriers. The E. coli cell incubation and the fluid-cell separation were integrated as special microfluidic units, and an interesting scheme for determination making use of inhibition of the microorganism respiratory chain was utilized. The procedure involved the addition of benzoquinone (BQ) to react with the electron or hydrogen carriers including enzymes, co-enzymes, prosthetic groups, or co-factors, which compose the respiratory chain, to form hydroquinone (HQ). When toxic environment inhibits the cellular respiratory chain, the quantity of HQ decreases leading to lower electrochemical current. The relationship between toxicant and HQ production provided the fundamentals for the biotoxicity assay. As proofs of concept; two heavy metals, Cu 2+ and Pb 2+ in water and soil; the antibiotic penicillin in soil; and the pesticides acetamiprid, triazolone, and acephate in vegetable juices were detected [20].

Clinical Applications
Sensitive and selective sensors constructed with screen-printed electrode (SPEs) have been developed for different analytes of clinical relevance. These sensors exhibit great advantages allowing fabrication of attractive designs for single and multiple determination even in the absence of biological elements. In this context, the special features of paper as support material for the preparation of diagnostic devices, together with those of screen-printed platforms, represents an important advance for easy self-testing and point-of-care (POC) assessment. Table 2 summarizes the analytical characteristics and the main properties of some recent and representative methods applied to analytes of clinical interest in biological samples . Some selected examples are discussed below.
An illustrative example is a wax-printed paper-based device reported by Martins et al. [27] for the electrochemical detection of 3-nitrotyrosine (3-NT), a biomarker of oxidative stress. The paper was modified to become a hydrophobic support, and then carbon and silver conductive inks were applied to generate a three electrode-system on a small spot. Square wave voltammetry (SWV) was employed to determine 3-NT in a range from 500 nM to 1 mM with a low LOD of 49.2 nM. More recently, a disposable paper-based printed electroanalytical strip has been reported for the rapid and high-throughput detection of glutathione in blood [28]. The detection involved a thiol-disulfide exchange reaction giving an electroactive product easily oxidizable at a Prussian Blue/carbon black nanocomposite screen-printed onto a wax-patterned filter paper. The resulting configuration, where the paper provides a reagents-free device, allowed the detection of glutathione up to a concentration 10 mM, with a LOD value of 60 µM, and was employed to quantify blood glutathione at physiological levels.
Hydrogen peroxide is an important biomarker associated with respiratory and pulmonary diseases such as asthma and lung cancer. A disposable cellulose paper-based electrochemical sensor integrated into a commercial respiratory mask was reported for on-site testing of H 2 O 2 in exhaled breath ( Figure 2) [29]. The device involved a Prussian-Blue-mediated carbon electrode for H 2 O 2 detection and a carbon blank electrode for subtracting the background currents. In the presence of the analyte, the oxidation product formed from Prussian Blue was electrochemically reduced providing amperometric responses related to the H 2 O 2 concentration. This configuration did not exhibit influence from environmental conditions or interferents due to differential measurements. In addition, the use of paper as flexible substrate and hygroscopic porous support eliminated the need for additional membranes.

Clinical Applications
Sensitive and selective sensors constructed with screen-printed electrode (SPEs) have been developed for different analytes of clinical relevance. These sensors exhibit great advantages allowing fabrication of attractive designs for single and multiple determination even in the absence of biological elements. In this context, the special features of paper as support material for the preparation of diagnostic devices, together with those of screen-printed platforms, represents an important advance for easy self-testing and point-of-care (POC) assessment. Table 2 summarizes the analytical characteristics and the main properties of some recent and representative methods applied to analytes of clinical interest in biological samples . Some selected examples are discussed below.
An illustrative example is a wax-printed paper-based device reported by Martins et al. [27] for the electrochemical detection of 3-nitrotyrosine (3-NT), a biomarker of oxidative stress. The paper was modified to become a hydrophobic support, and then carbon and silver conductive inks were applied to generate a three electrode-system on a small spot. Square wave voltammetry (SWV) was employed to determine 3-NT in a range from 500 nM to 1 mM with a low LOD of 49.2 nM. More recently, a disposable paper-based printed electroanalytical strip has been reported for the rapid and high-throughput detection of glutathione in blood [28]. The detection involved a thiol-disulfide exchange reaction giving an electroactive product easily oxidizable at a Prussian Blue/carbon black nanocomposite screen-printed onto a wax-patterned filter paper. The resulting configuration, where the paper provides a reagents-free device, allowed the detection of glutathione up to a concentration 10 mM, with a LOD value of 60 μM, and was employed to quantify blood glutathione at physiological levels.
Hydrogen peroxide is an important biomarker associated with respiratory and pulmonary diseases such as asthma and lung cancer. A disposable cellulose paper-based electrochemical sensor integrated into a commercial respiratory mask was reported for on-site testing of H2O2 in exhaled breath ( Figure 2) [29]. The device involved a Prussian-Blue-mediated carbon electrode for H2O2 detection and a carbon blank electrode for subtracting the background currents. In the presence of the analyte, the oxidation product formed from Prussian Blue was electrochemically reduced providing amperometric responses related to the H2O2 concentration. This configuration did not exhibit influence from environmental conditions or interferents due to differential measurements. In addition, the use of paper as flexible substrate and hygroscopic porous support eliminated the need for additional membranes.

Configuration
Technique and Method Detection Analyte/Sample Analytical Characteristics Ref.
Biosensors 2020, 10, x FOR PEER REVIEW 7 of 31 containing the sensor chip. Reproduced and adapted with permission of American Chemical Society [29].
Molecularly imprinted polymers (MIPs) used as electrode surface modifiers allow high selective recognition, although they sometimes lack the required sensitivity due to the poor conductivity of building materials. To improve the analytical performance, nanostructured configurations yielding larger currents and fast responses have been proposed. The resulting nano-sized MIPs have been combined with paper-based analytical devices to obtain three-dimensional electrochemical PADs (3D-ePADs), which provide additional advantages such as lower cost and smaller sample and reagents volumes. An interesting example is the method involving filter papers prepared by alkyl ketene dimer (AKD)-inkjet printing of a circular hydrophobic detection zone coupled with screenprinted graphite electrodes drop coated with Fe3O4@Au@SiO2-MIP nanocomposites (Figure 3), for the voltammetric determination of serotonin. Linear sweep voltammetry at +0.39 V provided a linear range from 0.01 to 1,000 mM with a LOD of 0.002 mM. The resulting MIPs exhibited strong affinity for the analyte, and the electrochemical sensor showed electrocatalytic activity toward the oxidation of serotonin. The sensor was successfully applied to the analysis of pharmaceutical capsules and urine samples [30]. Human C-reactive protein (CRP) is a nonspecific pentameric protein produced by hepatocytes in the liver upon stimulation by endogenous proinflammatory cytokines. CRP is an important biomarker for various cardiovascular diseases and its determination requires sensitive and accurate methods with high selectivity for application in complex clinical samples. Pinyorospathum et al. [31] developed a single step method for the determination of CRP in human serum involving a AuNPmodified SPCE self-assembled with PADs tethered with a biomimetic polymer consisting of thiolterminated poly(2-methacryloyloxyethyl phosphorylcholine) (PMPC-SH). The approach took advantage of the specific binding of protomers subunits forming the CRP structure to the Human C-reactive protein (CRP) is a nonspecific pentameric protein produced by hepatocytes in the liver upon stimulation by endogenous proinflammatory cytokines. CRP is an important biomarker for various cardiovascular diseases and its determination requires sensitive and accurate methods with high selectivity for application in complex clinical samples. Pinyorospathum et al. [31] developed a single step method for the determination of CRP in human serum involving a AuNP-modified SPCE self-assembled with PADs tethered with a biomimetic polymer consisting of thiol-terminated poly(2-methacryloyloxyethyl phosphorylcholine) (PMPC-SH). The approach took advantage of the specific binding of protomers subunits forming the CRP structure to the phosphorylcholine group in the presence of calcium ion [32]. Figure 4A shows that PMPC-SH copolymer reacts with AuNPs; then, [Fe(CN) 6 ] 3−/4− current at the resulting PMPC-SH/AuNPs-SPCE is measured by DPV to further subtracting it from the response in the presence of CRP. Figure 4B-F shows the preparation of the PAD in three parts-the middle where SPCE is placed, the green flap for the storage of calcium ions and dropping the sample, and the purple flap used for the detection with the [Fe(CN) 6 ] 3−/4− redox probe. The current decreased in the presence of CRP and Ca 2+ over the 5 to 5,000 ng mL −1 CRP concentration range with a LOD value of 1.6 ng mL −1 . The use of a PMPC-modified surface reduced the nonspecific adsorption of proteins, and the sensor response was not interfered by bilirubin, myoglobin, or albumin. The sensor was successfully applied to the determination of CRP in certified human serum.
Biosensors 2020, 10, x FOR PEER REVIEW 8 of 31 phosphorylcholine group in the presence of calcium ion [32]. Figure 4A shows that PMPC-SH copolymer reacts with AuNPs; then, [Fe(CN)6] 3−/4− current at the resulting PMPC-SH/AuNPs-SPCE is measured by DPV to further subtracting it from the response in the presence of CRP. Figure 4B-F shows the preparation of the PAD in three parts-the middle where SPCE is placed, the green flap for the storage of calcium ions and dropping the sample, and the purple flap used for the detection with the [Fe(CN)6] 3−/4− redox probe. The current decreased in the presence of CRP and Ca 2+ over the 5 to 5,000 ng mL −1 CRP concentration range with a LOD value of 1.6 ng mL −1 . The use of a PMPCmodified surface reduced the nonspecific adsorption of proteins, and the sensor response was not interfered by bilirubin, myoglobin, or albumin. The sensor was successfully applied to the determination of CRP in certified human serum. Electrochemical enzyme paper biosensors (EPADs) have been shown to provide an adequate microenvironment for direct measurements, while physical adsorption of the enzyme did not affect its native structure, function, and electrocatalytic activity [32]. An illustrative example is a biosensor for phenylketonuria (PKU) screening based on the determination of phenylalanine (Phe). The biosensor was implemented by immobilization of phenylalanine dehydrogenase (PheDH) over paper microzones placed onto an electrochemically reduced graphene oxide (ERGO)-modified SPCE. The detection of the NADH formed in the presence of NAD + provided a sensitive, low-cost, and fast method for PKU monitoring in neonatal blood samples [33]. An original paper-based biosensor was developed for the detection of acetylcholinesterase (AChE) [34]. This enzyme catalyzes the hydrolysis of acetylcholine neurotransmitter and its abnormal function can promote and accelerate the aggregation of amyloid-betapeptides closely related to Alzheimer's disease. The bioelectrode was fabricated by immobilization of acetylthiocholine on a sheet prepared with double adhesive tape. The enzyme samples were dropped on the backside of the electrode where, after hydrolysis, amperometric detection was performed and provided a LOD value of 0.1 U mL −1 AChE.
The combination of SPEs with simple paper-based microfluidics (μ-PEI) exhibits several advantages for the preparation of electrochemical biosensors compared with conventional analytical Electrochemical enzyme paper biosensors (EPADs) have been shown to provide an adequate microenvironment for direct measurements, while physical adsorption of the enzyme did not affect its native structure, function, and electrocatalytic activity [32]. An illustrative example is a biosensor for phenylketonuria (PKU) screening based on the determination of phenylalanine (Phe). The biosensor was implemented by immobilization of phenylalanine dehydrogenase (PheDH) over paper microzones placed onto an electrochemically reduced graphene oxide (ERGO)-modified SPCE. The detection of the NADH formed in the presence of NAD + provided a sensitive, low-cost, and fast method for PKU monitoring in neonatal blood samples [33]. An original paper-based biosensor was developed for the detection of acetylcholinesterase (AChE) [34]. This enzyme catalyzes the hydrolysis of acetylcholine neurotransmitter and its abnormal function can promote and accelerate the aggregation of amyloid-betapeptides closely related to Alzheimer's disease. The bioelectrode was fabricated by immobilization of acetylthiocholine on a sheet prepared with double adhesive tape. The enzyme samples were dropped on the backside of the electrode where, after hydrolysis, amperometric detection was performed and provided a LOD value of 0.1 U mL −1 AChE.
The combination of SPEs with simple paper-based microfluidics (µ-PEI) exhibits several advantages for the preparation of electrochemical biosensors compared with conventional analytical devices fabricated with other substrates (glass, silicon, or polymers). The resulting devices are inexpensive, easy to fabricate, and compatible with a variety of chemical or biochemical applications [35]. Cellulose papers with high surface area have become useful substrates in combination with SPEs for prototyping new point of care testing devices (POCTs) involving microfluidic systems in clinical diagnostics. Among the recent designs, those involving immunoassays constitute a challenge where the stability of the immunoreagents and the preparation of a surface capable of promoting electronic transfer to effectively enhance the assay sensitivity and selectivity, are critical factors. An illustrative method is that reported for the determination of alpha-fetoprotein (AFP). The method used paper-based microfluidic channels to integrate sampling, detection, and adsorption zones, as well as an rGO-tetraethylene pentamine (TEPA)/AuNPs nanocomposite for immobilization of specific AFP antibodies and sensitive detection ( Figure 5) [35]. AFP is one of the most important biomarkers in diagnosing hepatocellular carcinoma, and in the case of pregnant women, it is the first serologic biomarker to detect birth defects in a developing baby [36,37]. The immunoreaction was performed by applying the tested solution to the sample zone and letting it elute slowly to the detection zone where AFP was captured. Then, gold nanorods decorated with horseradish peroxidase and detector antibodies (HRP-GNRs-Ab 2 ) were dropped onto the sample zone to form a sandwich-type configuration on the working electrode and SWV detection was carried out in the presence of H 2 O 2 . The calibration plot showed a wide linear range (0.01-100.0 ng mL −1 ) with a low LOD value of 0.005 ng mL −1 .
Biosensors 2020, 10, x FOR PEER REVIEW 9 of 31 devices fabricated with other substrates (glass, silicon, or polymers). The resulting devices are inexpensive, easy to fabricate, and compatible with a variety of chemical or biochemical applications [35]. Cellulose papers with high surface area have become useful substrates in combination with SPEs for prototyping new point of care testing devices (POCTs) involving microfluidic systems in clinical diagnostics. Among the recent designs, those involving immunoassays constitute a challenge where the stability of the immunoreagents and the preparation of a surface capable of promoting electronic transfer to effectively enhance the assay sensitivity and selectivity, are critical factors. An illustrative method is that reported for the determination of alpha-fetoprotein (AFP). The method used paperbased microfluidic channels to integrate sampling, detection, and adsorption zones, as well as an rGO-tetraethylene pentamine (TEPA)/AuNPs nanocomposite for immobilization of specific AFP antibodies and sensitive detection ( Figure 5) [35]. AFP is one of the most important biomarkers in diagnosing hepatocellular carcinoma, and in the case of pregnant women, it is the first serologic biomarker to detect birth defects in a developing baby [36,37]. The immunoreaction was performed by applying the tested solution to the sample zone and letting it elute slowly to the detection zone where AFP was captured. Then, gold nanorods decorated with horseradish peroxidase and detector antibodies (HRP-GNRs-Ab2) were dropped onto the sample zone to form a sandwich-type configuration on the working electrode and SWV detection was carried out in the presence of H2O2.
The calibration plot showed a wide linear range (0.01-100.0 ng mL −1 ) with a low LOD value of 0.005 ng mL −1 . Figure 5. Schematic displays of the modification and assay procedure for the implementation of an immunosensor for the determination of alpha-fetoprotein (AFP) using paper-based microfluidic Figure 5. Schematic displays of the modification and assay procedure for the implementation of an immunosensor for the determination of alpha-fetoprotein (AFP) using paper-based microfluidic channels to integrate sampling, detection and adsorption zones, and a reduced graphene oxide (rGO)-tetraethylene pentamine (TEPA)/AuNPs nanocomposite for immobilization of specific AFP antibodies. Reproduced with permission of Elsevier [35].
Oher configurations involving label-free paper-based immunosensors with immobilized capture antibodies have been reported for the detection of hormones [40,42], CRP [43], interferon (IFN-γ) [45], cancer biomarkers [44,46], and cardiac biomarkers [41,47]. Among them, it is worth mentioning the platform described by Ruecha et al. [45] involving a paper-based microfluidic device coupled with a label-free electrochemical impedimetric immunosensor for the detection of IFN-γ in serum. This multifunctional cytokine, originally characterized by its viral activities, is primarily secreted by natural killer T cells as a part of the innate immune response to intracellular pathogens [48] and plays a crucial role related to inflammatory and antoimmune diseases. In Ruecha's method, a wax-printing strategy was implemented to fabricate the paper electrode, which was screened with graphene modified ink to deposit polyaniline (PANI) and further covalent immobilization of the specific anti-IFN-γ antibodies. The increase of the charge transfer resistance with the cytokine loading provided a linear relationship with logarithmic concentrations of IFN-γ in the 5-1000 pg mL −1 range with a LOD of 3.4 pg mL −1 .
Although the number of paper-based electrochemical biosensing platforms has increased in the last few years, the vast majority of the reported methods involve the use of enzymes and antibodies as recognition elements, and so far, they have not been expanded to nucleic acid-based assays. In this context, Liu et al. [38] prepared a paper modified with signal molecule-labelled DNA and a screen-printed electrode along with target recognition solutions to achieve the detection of multiple biomarkers. The method is based on the target-induced synthesis of Mg 2+ -dependent DNAzyme for catalyzing the cleavage of substrate DNA from paper, taking advantage of the high specific target-triggered polymerization/nicking and DNAzyme-catalyzed signal amplification. The performance of this method was evaluated using a microRNA recognition probe for lung cancer-specific miR-21, a phosphorylated hairpin probe for targeting alkaline phosphatase (ALP), and a DNA aptamer for carcinoembryonic antigen (CEA). Ferrocene-labeled DNA (Fc-DNA) was immobilized on paper by functionalizing it with aldehyde groups and further Schiff-based reaction ( Figure 6A). Then, the paper-electrochemical biosensor was prepared by sticking the Fc-DNA modified paper onto a plastic slide and carbon nanotubes modified SPE. As an example, the fundamentals of miR-21 detection are illustrated in Figure 6B. After incubation with the recognition solution that contains the ssDNA probe (P1), KF polymerase and nicking endonuclease Nt.BbvCI, the polymerization via KF activity is initiated to extend the 3'-end, providing dsDNA with recognition site for endonuclease whose activity to cut one strand of dsDNA generates new replication sites. Then, the Mg 2+ -dependent DNAzyme strand is displaced and released. This cycle produces a large amount of DNAzyme strands that fold into the catalytically active loop structure and bind to immobilized Fc-DNA resulting in the release of DNAzyme strands and cleaved Fc-shorter ssDNA from the paper, which diffuses to the surface of CNTs-SPE giving a DPV response.
Among paper-based electrochemical DNA sensors, configurations developed for the detection of human papillomavirus (HPV) [54] and human immunodeficiency virus (HIV) [55] are particularly relevant. Teengam et al. [54] reported a graphene-PANI modified electrode with immobilized anthraquinone-labeled pyrrolidinyl peptide nucleic acid probe (AQ-PNA) for the detection of a synthetic 14-base oligonucleotide target with the sequence of HPV type 16 DNA by electrochemical measurement of the AQ response by SWV. A linear range of 10-200 nM and a LOD value of 2.3 nM were obtained. The performance of this biosensor was tested with the detection of PCR-amplified DNA. On the other hand, Cinti et al. [55] developed a series of paper-based strips for the electrochemical detection of single and double stranded DNA, which were successfully applied to a synthetic PCR amplified dsDNA sequence related to HIV in serum samples. Paper-based AuNPs-SPE platforms and triplex forming oligonucleotides (TFO) including Methylene Blue (MB) were used as the recognizing probes.

Wearable Printed Electrodes for Biosensing Applications
Due to the booming research activity in the field of wearable and/or flexible printed electrodes, excellent reviews have been recently reported highlighting the versatility and tremendous potential of these devices [61][62][63][64][65][66][67][68][69]. Therefore, this section is just limited to give a rough overview of late advances and prospects to draw the current landscape of wearable and flexible printed electrodes (not implantable) for biosensing. Accordingly, only a few of the most representative methods reported during 2018 and 2019, applied mainly to clinical diagnosis and environment monitoring, are critically discussed. Table 3 summarizes the main features of these methods.
SPEs can be easily printed in a variety of shapes (flower, skull, marijuana, panda bear, etc., Figure 7a-d) and sizes and can be modified with different biological elements and nanomaterials. Leveraging on these advantages, screen printing has been employed to construct affordable wearable printed electrochemical sensors to provide real-time information on both the wearer's health and performance (opening the door to individualized medicine) and the surrounding environment. In the physiological monitoring field, the active sensor surface is in close contact with the epidermis (oral mucosa in the mouth, stratum corneum or skin) to detect relevant biomarkers such as glucose [70] and ethanol [71] in different informative biofluids (saliva, sweat, tears). However, for monitoring the wearer's environment, the sensor faces away from the epidermis in the direction of the surrounding to detect risk of exposure to chemicals [63]. Moreover, the recently explored robotic assisted strategy implies that the robot fingertips are kept in close contact with the target sample [72].
The spectacular growth witnessed in wearable printed sensors is largely due to the development of novel materials that imparted the resulting sensors the capabilities to fold, bend, stretch, and repair, ensuring their performance during on-body applications under extreme tensile stress [73]. Wearable electrochemical sensors have been implemented on head-to-toe wearable platforms and in connection to different biofluids, environments and analytes [64]. For the realization of wearable applications matching the non-planarity and mechanical properties of the human body, electrochemical sensors have been printed on temporary tattoo, bendable bandage, gloves, contact lens, water-soluble silk thin-film substrates (transferred to tooth enamel) or textile substrates (GORE-TEX and Neoprene) (Figure 7e-j) [63] or incorporated in mouthguards, eyeglasses, or rings (Figure 7k-m). The great progress experienced by electronics in terms of flexibility and miniaturization [65,[74][75][76], in the development of effective methods for stimulating/controlling of non-invasive bio-fluids collection and the proliferation of smart-phones and connected devices, together with a growing consumer demand for health awareness, and the imperative need for doctors to obtain as much objective and quality data from their patients as possible, have been crucial aspects in the development of fully implementable wearable electrochemical devices and in opening up new avenues for body-integrated electronics previously unattainable [74]. However, powering is still the main Achilles' heel of these devices and the size and weight of the power source may limit the wearability of the biodevices and hinder the wearer's activity. Therefore, the rational integration of power sources with biosensors is a desperate requirement and additional efforts are required to develop anatomically compliant, miniaturized, stretchable and flexible power sources [64,77].
In general, there are three different modes of integrating wearable biosensors and power supplies: (1) an external circuit connection which is bulky and cumbrous; (2) a flexible substrate-based integration; and (3) all-in-one integration [78]. The last two strategies, made possible by advances in device designs and micro/nanofabrication technologies, are more widely used. The second strategy implies each component is relatively independent and can be considered a general integration strategy applicable to diverse sensing systems without having to worry about structural compatibility between components but difficult to allow the level of miniaturization required. The third strategy is effective for miniaturized designs in which all the components suffer from similar deformations simultaneously. However, the endurability difference among them should be minimized to guarantee the normal function of the respective component.
Currently, wearable biodevices are powered mainly by (i) safe high energy wearable batteries; (ii) energy conversion devices (piezoelectric and triboelectric nanogenerators, which harvest the mechanical energy in human motions, such as walking, breathing, and waving arms; solar cells, which harness light energy; thermoelectric supercapacitors; biofuel; and water-voltage cells); (iii) energy storage devices (mechanically flexible energy storage elements, mainly supercapacitor, and lithium-ion battery); (iv) hybrid power supplies combining energy conversion with energy storage; and (v) wireless energy transfer (wireless coils, like RF antennas). Significant progress have also been made in self-power and energy-efficient or even energy-free systems devices, fueled by the development of high-efficiency energy acquisition approach and ultra-low power consumption technique [68,69,[78][79][80].
lithium-ion battery); iv) hybrid power supplies combining energy conversion with energy storage; and v) wireless energy transfer (wireless coils, like RF antennas). Significant progress have also been made in self-power and energy-efficient or even energy-free systems devices, fueled by the development of high-efficiency energy acquisition approach and ultra-low power consumption technique [68,69,[78][79][80]. As a previous step to on-body measurements, Payne et al. made an exhaustive study to characterize the effects of five different salts in physiologically relevant concentration ranges on the performance of a printed, flexible, wearable biosensor involving lactate oxidase and tetrathiafulvalene for the amperometric detection of lactate in sweat [87].
The extensive and pioneering work performed by Wang´s group in the development of wearable and flexible printed electrodes for biosensing in healthcare, food, and security fields should be noted. Wang's team proposed the use of fully integrated wearable bendable bandage and minimally invasive microneedle-based sensors modified with catechol (CAT) for rapid and decentralized screening of skin melanoma through the amperometric detection of the benzoquinone (BQ) generated in the presence of the tyrosinase (TYR) biomarker (Figure 8). The bandage sensor exhibited high resiliency against mechanical strains due to the use of stress-enduring inks for its printing. These As a previous step to on-body measurements, Payne et al. made an exhaustive study to characterize the effects of five different salts in physiologically relevant concentration ranges on the performance of a printed, flexible, wearable biosensor involving lactate oxidase and tetrathiafulvalene for the amperometric detection of lactate in sweat [87].
The extensive and pioneering work performed by Wang's group in the development of wearable and flexible printed electrodes for biosensing in healthcare, food, and security fields should be noted. Wang's team proposed the use of fully integrated wearable bendable bandage and minimally invasive microneedle-based sensors modified with catechol (CAT) for rapid and decentralized screening of skin melanoma through the amperometric detection of the benzoquinone (BQ) generated in the presence of the tyrosinase (TYR) biomarker (Figure 8). The bandage sensor exhibited high resiliency against mechanical strains due to the use of stress-enduring inks for its printing. These skin-worn sensors were interfaced to flexible electronic board that controlled the electrochemical operation and transmitted data wirelessly to a mobile device, and were used to screen biomarkers both on the skin surface (bandage sensor) or under the skin (microneedle device). They were applied to analyze TYR-containing agarose phantom gel and porcine skin [81]. These epidermal sensors allow skin cancer screening in less than 4 min obviating the need of painful solid biopsies and the associated delays and anxiety.
Biosensors 2020, 10, x FOR PEER REVIEW 17 of 31 skin-worn sensors were interfaced to flexible electronic board that controlled the electrochemical operation and transmitted data wirelessly to a mobile device, and were used to screen biomarkers both on the skin surface (bandage sensor) or under the skin (microneedle device). They were applied to analyze TYR-containing agarose phantom gel and porcine skin [81]. These epidermal sensors allow skin cancer screening in less than 4 min obviating the need of painful solid biopsies and the associated delays and anxiety. The same group reported a strategy using a single wearable and flexible epidermal platform for the simultaneous yet independent noninvasive sampling and analysis of two different epidermal biofluids (sweat and skin interstitial fluid (ISF) with a blood-like composition) at two physically separate locations. This approach involves parallel operation of iontophoretic delivery of the sweatinducing pilocarpine into the skin and reverse iontophoretic ISF extraction across the skin at anode and cathode, respectively (Figure 9a,b) [83]. The developed wearable device was implemented using a cost-effective screen-printing technique with body-compliant temporary tattoo materials for disposable single use and conformal wireless readout circuits, and integrated amperometric GOx and AOx biosensors (Figure 9c). It was used for real-time monitoring of alcohol and glucose levels in sweat and ISF, respectively, from individuals consuming food and alcoholic drinks. The same group reported a strategy using a single wearable and flexible epidermal platform for the simultaneous yet independent noninvasive sampling and analysis of two different epidermal biofluids (sweat and skin interstitial fluid (ISF) with a blood-like composition) at two physically separate locations. This approach involves parallel operation of iontophoretic delivery of the sweat-inducing pilocarpine into the skin and reverse iontophoretic ISF extraction across the skin at anode and cathode, respectively (Figure 9a,b) [83]. The developed wearable device was implemented using a cost-effective screen-printing technique with body-compliant temporary tattoo materials for disposable single use and conformal wireless readout circuits, and integrated amperometric GOx and AOx biosensors (Figure 9c). It was used for real-time monitoring of alcohol and glucose levels in sweat and ISF, respectively, from individuals consuming food and alcoholic drinks. Biosensors 2020, 10, x FOR PEER REVIEW 18 of 31 Flexible epidermal tattoo and textile-based electrochemical biosensors using stretchable organophosphorus hydrolase (OPH) enzyme electrodes have been developed for continuous vaporphase detection of organophosphorus (OP) threats [82] (Figure 10a). These wearable sensors were fabricated with elastomeric inks and displayed resiliency toward mechanical stress expected from the wearer's activity without compromising the biosensing performance. They were coupled with a fully integrated conformal flexible electronic interface providing square-wave voltammetry (SWV) detection of the enzymatically-generated nitrophenol product ( Figure 10b) and wireless data transmission. The sensor achieved a LOD of 12 mg L −1 in terms of OP air density. The same group proposed also a wearable tattoo OPH-pH biosensor for real-time on-body potentiometric monitoring of G-type nerve agent simulants (using fluorine-containing OP nerve agent simulant diisopropyl fluorophosphate, DFP, as model) in both liquid and vapor phases. The OPH biocatalytic recognition phase was coupled on a flexible printed transducer with a pH-responsive poly(aniline) PANI layer for monitoring the proton release during the enzymatic hydrolysis of DFP by OPH. This skin-worn OP potentiometric sensor withstands severe mechanical strains without compromising the analytical performance and displays a wide dynamic range, fast response, high selectivity towards DFP and good reproducibility. These wearable OP biosensing devices hold considerable promise for real-time on-body detection and warning exposure to chemical threats such CWAs and pesticides in a variety of scenarios for triggering timely countermeasure actions, changing dramatically the protection of civilians, farmers, and military personnel. Flexible epidermal tattoo and textile-based electrochemical biosensors using stretchable organophosphorus hydrolase (OPH) enzyme electrodes have been developed for continuous vapor-phase detection of organophosphorus (OP) threats [82] (Figure 10a). These wearable sensors were fabricated with elastomeric inks and displayed resiliency toward mechanical stress expected from the wearer's activity without compromising the biosensing performance. They were coupled with a fully integrated conformal flexible electronic interface providing square-wave voltammetry (SWV) detection of the enzymatically-generated nitrophenol product ( Figure 10b) and wireless data transmission. The sensor achieved a LOD of 12 mg L −1 in terms of OP air density. The same group proposed also a wearable tattoo OPH-pH biosensor for real-time on-body potentiometric monitoring of G-type nerve agent simulants (using fluorine-containing OP nerve agent simulant diisopropyl fluorophosphate, DFP, as model) in both liquid and vapor phases. The OPH biocatalytic recognition phase was coupled on a flexible printed transducer with a pH-responsive poly(aniline) PANI layer for monitoring the proton release during the enzymatic hydrolysis of DFP by OPH. This skin-worn OP potentiometric sensor withstands severe mechanical strains without compromising the analytical performance and displays a wide dynamic range, fast response, high selectivity towards DFP and good reproducibility. These wearable OP biosensing devices hold considerable promise for real-time on-body detection and warning exposure to chemical threats such CWAs and pesticides in a variety of scenarios for triggering timely countermeasure actions, changing dramatically the protection of civilians, farmers, and military personnel. Wang's team reported recently a wearable eyeglasses-based tear biosensing system for noninvasive monitoring of key biomarkers [85]. The wearable tear bioelectronic platform integrates a microfluidic electrochemical detector into an eyeglasses nose-bridge pad and the wireless electronic circuitry into the eyeglasses frame and yielded a fully portable, convenient-to-use fashionable sensing device placed outside the eye region addressing drawbacks of sensor systems involving direct contact with the eye as the contact lenses platform (Figure 11a). The concept was used for real-time noninvasive detection of alcohol, glucose and multiple vitamins in tears in connection with enzymatic (AOx and GOx) biosensing fluidic system (alcohol and glucose) and rapid voltammetric scanning (vitamins). This tear alcohol sensing strategy exhibited good correlation to concurrent blood alcohol concentration (BAC) in the monitoring of alcohol intake in individuals over multiple drinking courses.
A novel sensor ring concept, comprising a powerful wireless electronic board embedded into a ring platform and a readily replaceable printed dual-sensor electrode cap, was developed for the simultaneous, direct and rapid field detection of D9-tetrahydrocannabinol (THC) and alcohol in diluted saliva [86]. The ring sensing platform contained a voltammetric THC sensor based on a multiwalled carbon nanotubes MWCNTs/carbon electrode and an amperometric alcohol biosensor Wang's team reported recently a wearable eyeglasses-based tear biosensing system for non-invasive monitoring of key biomarkers [85]. The wearable tear bioelectronic platform integrates a microfluidic electrochemical detector into an eyeglasses nose-bridge pad and the wireless electronic circuitry into the eyeglasses frame and yielded a fully portable, convenient-to-use fashionable sensing device placed outside the eye region addressing drawbacks of sensor systems involving direct contact with the eye as the contact lenses platform (Figure 11a). The concept was used for real-time non-invasive detection of alcohol, glucose and multiple vitamins in tears in connection with enzymatic (AOx and GOx) biosensing fluidic system (alcohol and glucose) and rapid voltammetric scanning (vitamins). This tear alcohol sensing strategy exhibited good correlation to concurrent blood alcohol concentration (BAC) in the monitoring of alcohol intake in individuals over multiple drinking courses.
A novel sensor ring concept, comprising a powerful wireless electronic board embedded into a ring platform and a readily replaceable printed dual-sensor electrode cap, was developed for the simultaneous, direct and rapid field detection of D9-tetrahydrocannabinol (THC) and alcohol in diluted saliva [86]. The ring sensing platform contained a voltammetric THC sensor based on a multi-walled carbon nanotubes MWCNTs/carbon electrode and an amperometric alcohol biosensor involving Prussian-blue (PB) mediator, coated with AOx/chitosan reagent layer on the ring cap (Figure 11b). The dual-analyte THC/alcohol ring sensor system showed no cross talk and high sensitivity (0.5 µM THC and 0.2 mM alcohol). THC and alcohol were determined simultaneously in the same diluted saliva sample within 3 min without any interference from the matrix. This new wearable THC/alcohol ring sensor, readily expanded to detecting other drugs of abuse, is very promising for rapid testing of suspected drivers and for alerting users to their own levels before driving.
Biosensors 2020, 10, x FOR PEER REVIEW 20 of 31 involving Prussian-blue (PB) mediator, coated with AOx/chitosan reagent layer on the ring cap ( Figure 11b). The dual-analyte THC/alcohol ring sensor system showed no cross talk and high sensitivity (0.5 μM THC and 0.2 mM alcohol). THC and alcohol were determined simultaneously in the same diluted saliva sample within 3 min without any interference from the matrix. This new wearable THC/alcohol ring sensor, readily expanded to detecting other drugs of abuse, is very promising for rapid testing of suspected drivers and for alerting users to their own levels before driving. The same group have also reported in a pioneering way advances in wearable chemical sensor technology and flexible electronics to develop chemical sensing robotic fingers (printed on the robotic glove) for rapid discrimination between sweetness, sourness, and spiciness, via electrochemical monitoring of glucose, ascorbic acid, and capsaicin in different drinks (juices, sport and soft drinks and coffee) and extracts (green chili, red paprika and red pepper) ( Figure 12) [72]. It is worth remarking that although it was out of the period to which this section has been restricted this group proposed also glove type wearable devices for use in forensic analysis (gunshot residues and nitroaromatic explosive compounds) [88]. This chemical sensing robotic skin is a key demonstration to spur future development of wearable printed electrodes, which offers great opportunity for automated chemical sensing machinery, facilitating robotic decision in a wide range of applications and even in potentially hazardous environments for human counterparts. The same group have also reported in a pioneering way advances in wearable chemical sensor technology and flexible electronics to develop chemical sensing robotic fingers (printed on the robotic glove) for rapid discrimination between sweetness, sourness, and spiciness, via electrochemical monitoring of glucose, ascorbic acid, and capsaicin in different drinks (juices, sport and soft drinks and coffee) and extracts (green chili, red paprika and red pepper) ( Figure 12) [72]. It is worth remarking that although it was out of the period to which this section has been restricted this group proposed also glove type wearable devices for use in forensic analysis (gunshot residues and nitroaromatic explosive compounds) [88]. This chemical sensing robotic skin is a key demonstration to spur future development of wearable printed electrodes, which offers great opportunity for automated chemical sensing machinery, facilitating robotic decision in a wide range of applications and even in potentially hazardous environments for human counterparts. Wang's group has reported very recently an epidermal AAOx biosensor able to monitor the dynamics of vitamin C in sweat after the intake of vitamin C pills and fruit juices [89]. This method combines the use of a flexible vitamin C tattoo patch fabricated on a polyurethane substrate with a localized iontophoretic sweat stimulation system. Chrono-amperometric cathodic detection of the oxygen cosubstrate consumption during the enzymatic reaction, demonstrates very interesting potential for personalized nutrition solutions. Wang's group has reported very recently an epidermal AAOx biosensor able to monitor the dynamics of vitamin C in sweat after the intake of vitamin C pills and fruit juices [89]. This method combines the use of a flexible vitamin C tattoo patch fabricated on a polyurethane substrate with a localized iontophoretic sweat stimulation system. Chrono-amperometric cathodic detection of the oxygen cosubstrate consumption during the enzymatic reaction, demonstrates very interesting potential for personalized nutrition solutions.  Wang's group has reported very recently an epidermal AAOx biosensor able to monitor the dynamics of vitamin C in sweat after the intake of vitamin C pills and fruit juices [89]. This method combines the use of a flexible vitamin C tattoo patch fabricated on a polyurethane substrate with a localized iontophoretic sweat stimulation system. Chrono-amperometric cathodic detection of the oxygen cosubstrate consumption during the enzymatic reaction, demonstrates very interesting potential for personalized nutrition solutions.

General Considerations, Challenges to Face, and Future Prospects
The advances that have occurred in recent years in screen-printing technology provide unimaginable possibilities in electrochemical sensing and biosensing. Paper-based and wearable (bio)sensors are two of the areas that benefit from such advances. e-PADs show excellent opportunities for sensing and biosensing mainly in the environmental and clinical fields. For these purposes, different types of paper impregnated with the suitable reagents, modified with different nanomaterials (metal nanoparticles, carbon nanostructures, and nano-sized MIPs) either in the absence or in the presence of specific bioreceptors (enzymes, lectins, antibodies, and, to a much lesser extent, nucleic acids) have been proposed. The resulting devices exploit the interesting features offered by nanomaterials in terms of electrocatalytic properties, biocompatibility and high surface area, and the strong selectivity of biological molecules or MIPs, with the advantages of electrochemical detection. In the environmental field, e-PADs have been applied to the determination of heavy metals (Cu 2+ , Pb 2+ , Cd 2+ ), anions (ClO 3 − , NO 2 − ), gases (NO, NO 2 , DMMP), CWAs (mustard agent) microbial pathogens (E. coli), or other biotoxics (antibiotics or pesticides).
In the clinical field, e-PADs have been utilized for the determination of a wide variety of biomolecules including miRNAs, hormones (17β-estradiol, FSH), viruses (HPV, HIV), proteomic biomarkers of relevance in cancer and cardiovascular diseases (CRP, BNP, IFN-γ, CA125, cTnI, AFP, BChE activity, MMP9, and CEA), and other clinically relevant analytes (3-NT, glutathione, glucose, H 2 O 2 , serotonin, L-Tyr, Phe, acetylcholine and Cl − ). Paper-based electrochemical (bio)sensors have been employed to determine target analytes in highly variable matrices-soils, exhaust gases, waters and industrial sewage, cellular extracted DNA, blood, plasma, serum, urine, sweat, exhaled breath, and pharmaceutical capsules. Remarkable achievements include the development of disposable gPADs and the combination of SPEs with simple paper-based microfluidics (µ-PEI) with great interest in clinical diagnostics for prototyping new POCTs.
Apart from the interesting features derived from the use of paper as substrate such as porosity, capillarity, high surface area/volume ratio, disposability, lightness, flexibility, eco-friendliness, and low-cost, the filtration properties of this particular substrate have also been exploited. Moreover, some of these (bio)devices exhibit antifouling properties that are highly pursued to ensure the proper functioning of the devices in real world matrices and their use to determine low levels of analytes directly in such matrices involving simple and straightforward protocols.
Focusing on printed wearable devices devoted to electrochemical biosensing, new generation of printed wearable devices include soft, biocompatible, stretchable, and anatomically compliant devices with multifunctional characteristics that enable efficient bio-integration and withstand high tensile stress associated with on-body applications. Over the past two years, their applications have been geared mostly toward healthcare and environmental fields. To date, they have demonstrated preliminary potential for individual or multiplexed electrochemical determination in or near real time of relevant analytes in the agro-food, clinical and environmental areas (ethanol, drugs of abuse, lactate, glucose, vitamins, TYR, OP nerve agents) in different matrixes such as biofluids (sweat, saliva, tears, and ISF), vapor phases, and skin. So far, printed temporary tattoo and bandage sensors and printed sensors mounted in mouthguards, eyeglasses, or rings have been proposed for healthcare applications. Regarding environmental applications sensors have been printed on textile and gloves. Particularly noteworthy is the use of this type of sensors to screen biomarkers related with cancer in skin, the pioneering coupling with a parallel iontophoretic mechanism (extraction and delivery) for simultaneous sampling of different biofluids at separate locations, the enclosing within a microfluidic chamber for continuous monitoring and the development of a wearable taste-sensing robotic technology to discriminate between different flavors in liquid and solid food samples and an epidermal enzymatic biosensor for noninvasive nutrition status assessments.
However, it should be noted that, at present, the remarkable capabilities for taking preventive intervention of health and environmental risks, have been proved using proof-of-concept prototypes and for a limited number of samples, analytes, and biosensing approaches. Therefore, a thorough validation with large population studies and coordination and collaboration with medical practitioners to correctly interpret the data, and a better understanding of the correlations between analyte concentrations in the blood and noninvasive biofluids are required to underpin clinical acceptance. In addition, their extension to bioaffinity assays, particularly challenging since they require complex, multistep and usually not reversible protocols, is highly demanded to make other important biomarkers (proteins, DNAs, and RNAs) accessible to monitoring. Indeed, despite the great customization potential of these printed wearable electrochemical devices, important efforts are required for advancing them from prototypes to field devices and for their widespread commercial exploitation. Additional efforts are required to improve durability and robustness of wearables' batteries. Solutions are required to decrease the power consumption of devices for extended monitoring periods, which include battery consumption minimization, and developing of replaceable and flexible power source with continuous and long-time output or self-powering wearable devices.
It is evident that the development of wearable printed electrochemical devices, which so far has only scratched the surface of their tremendous potential, is poised to grow very rapidly over the next decade, bringing a considerable advance to the field of wearable devices. Although many challenges that impede the widespread adaptation of this field for commercial applications have been addressed due to the recent years endeavors in material science, microfluidics, nanotechnology, and biotechnology, together with the work in unison of researchers from diverse fields, there are still some outstanding issues before their full potential will be realized and exploited in our real lives. However, there is clearly room for them within many applications (forensic, food assessment, healthcare, security), where rapid screening and timely chemical information is critical.
The exciting new developments anticipated to come in the foreseeable future in both paper-based and wearable electrochemical (bio)sensors will certainly change and improve our daily lives providing eco-friendly, affordable and efficient solutions for smart healthcare (preventive medicine, precision medication and management of chronic diseases) and wellness moving the lab to our body (skin, mouth, and eyes), minimizing risks of exposure to chemical threats and drug impaired driving concerns. Taking into account the demands for ordinary users facing other issues such as the diversity of analytical targets in the practical applications and the selectivity, stability and recyclability of these screen-printing biodevices, additional future research can be predicted in this field. Moreover, continuous investment in material preparation and fabrication process perfection (including ingenious structural designs) and in achieving higher integration between the multifunction sensing units and auxiliary components (power supply, communication and even signal processing and displaying) will play a significant role in constructing cost-effective and consistent (bio)sensors, easily adopted by current society.
Indeed, today, it is timely to stress that the adaptation of mask filtration systems to include transducers for aerosol/viral detection, although very difficult, would represent a disruptive technology for the detection of pandemics such as the SARS-CoV-2 coronavirus we are experiencing.