Manual Wheelchair Equipped with a Planetary Gear-Research Methodology and Preliminary Results

Featured Application: A wheelchair with a planetary gear for adjusting the torque demand to needs of the operator. The results of the preliminary tests show that the muscular effort of the person propelling a wheelchair increases as the gear ratio increases. Varying propelling conditions change the muscle load of the operator and this alters the kinematics of the upper limbs’ movement and tilt of the torso as well. Occurring changes affect the position of the center of gravity—which in turn change the conditions of the resistance to motion. The use of such a wheelchair facilitates movement on soft ground or inclined planes and gives the possibility of increasing the range of movement. Abstract: The purpose of the study was to create a research methodology for testing the newly developed wheelchair drive, which allows the operator to choose the gear ratio and, thus, makes it possible to change the propulsion torque value. The aim was to choose such conditions in the experiment, that would result in great enough changes in the participant’s muscle load and body kinematics for it to be possible to register them with applied measuring methods. Surface electromyography was used to assess the effort that was required for the propulsion of a wheelchair under different conditions. Additionally, upper limb motion capture measurements were also performed. The preliminary results show that the muscular effort of the participant propelling the wheelchair increases with the load—resulting from both the gear ratio and the inclination angle. At the same time, the position of the motion range of upper limb individual segments changes signiﬁcantly. Simultaneously, the mean value of the shoulder displacement and its angle of rotation decreases.


Introduction
The wheelchair's purpose is to enable the independent existence of people who are affected by mobility disabilities. The experience of wheelchair users shows that these devices are important for their mental health because they promote increased mobility and allow social interaction development [1,2]. One of the important aspects for the users of these devices is increasing their overall movement distance [3]. Another incentive should concern the torque on the drive wheels [4], which is particularly important when navigating difficult road conditions. Factors that impede the movement of vehicles, such as wheelchairs, include a strong wind, unpaved or dirty surfaces [5,6], hills [7], unevenness, thresholds or stairs. All these factors translate into increased resistance when driving a wheelchair [8] and, thus, require an increased effort from users when propelling a wheelchair.
Manual wheelchairs equipped with technical solutions such as suspension systems to overcome obstacles without special architectural modifications are available [9]. Whereas

The Description of Developed Design
The study was carried out on an operator of a wheelchair equipped with multi-speed gears integrated with powered wheels. At the core of the described design is a planetary gear that gives the user the possibility to smoothly change gears. The planetary gear itself (sun gear with external teeth, gears and carriers) is a compact mechanism that can move along the main wheel axle. This provides the ability to change the drive parameters while linking the handrims delivering the rotational torque to the hubs driving the large wheels.
A view of the assembled system is presented in Figure 1. The view of the diagram of the designed planetary gear is presented in Figure 2. The system works as follows. The torque generated by the operator's limbs is introduced to the system through handrims (10). The handrims (10) are coupled with the internal gear hub (11) to enable the power transmission. The torque ratio changes as a result of switching the operating mode of the planetary gear (mounted into the hub), described later in the study. From the internal gear hub (11), the torque is transmitted via a system of spokes mounted to the hub flange (9) onto tires (8). As a result, the handrims (10) can rotate at a different speed than the tires (8), thus realizing different ratios.
The transmission is equipped with a splined couplers assembly (Figure 2a), so that the torque from the planetary gear is delivered to the hub by a splined disc (1) and a splined ring (2). The transmission consists of a bearing system (Figure 2b), which determines the positions of the carrier (3) with the planetary gears, the sun gear (4) and the ring gear (7). A set of four bearings: two radial (5) and two axial (6) were used for the positioning all of the moving elements. Radial bearings maintain the axis of rotation of the carrier with the axis of the sun gear. In contrast, the axial bearings reduce the friction between the carrier and the ring gear assembly. In addition, the axial bearings position the ring gear assembly relative to the carrier and the sun gear assembly. torque generated by the operator's limbs is introduced to the system through handrims (10). The handrims (10) are coupled with the internal gear hub (11) to enable the power transmission. The torque ratio changes as a result of switching the operating mode of the planetary gear (mounted into the hub), described later in the study. From the internal gear hub (11), the torque is transmitted via a system of spokes mounted to the hub flange (9) onto tires (8). As a result, the handrims (10) can rotate at a different speed than the tires (8), thus realizing different ratios.  The transmission is equipped with a splined couplers assembly (Figure 2a), so that the torque from the planetary gear is delivered to the hub by a splined disc (1) and a splined ring (2). The transmission consists of a bearing system (Figure 2b), which determines the positions of the carrier (3) with the planetary gears, the sun gear (4) and the ring gear (7). A set of four bearings: two radial (5) and two axial (6) were used for the positioning all of the moving elements. Radial bearings maintain the axis of rotation of the carrier with the axis of the sun gear. In contrast, the axial bearings reduce the friction between the carrier and the ring gear assembly. In addition, the axial bearings position the ring gear assembly relative to the carrier and the sun gear assembly.
A kinematic diagram of the solution is shown in Figure 3. The planetary gear works for all gears with the sun gear (14) locked. The gear shift depends on the input and output of the transmission. In the first gear- Figure 3a, which gives a reduction in the rotational speed-the torque from the handrims is transferred to the sun wheel (14) with internal teeth (12) through the planetary gear (13) and is delivered from the carrier (15). In the second, neutral gear- Figure 3b-the torque is delivered to the sun gear (14) through the planetary gear (14) with internal teeth (12) and it is passed on from here. In this case, the carrier (14) also rotates, but it is not picked up by the hub. In the last, third gear of the mechanism- Figure 3c, which is a multiplier of rotational speed-the torque from the handrims is delivered to the planetary gear carrier (15) and is picked up from the sun gear (14) through the planetary gear (13) with internal teeth (12). A kinematic diagram of the solution is shown in Figure 3. The planetary gear works for all gears with the sun gear (14) locked. The gear shift depends on the input and output of the transmission. In the first gear- Figure 3a, which gives a reduction in the rotational speed-the torque from the handrims is transferred to the sun wheel (14) with internal teeth (12) through the planetary gear (13) and is delivered from the carrier (15). In the second, neutral gear- Figure 3b-the torque is delivered to the sun gear (14) through the planetary gear (14) with internal teeth (12) and it is passed on from here. In this case, the carrier (14) also rotates, but it is not picked up by the hub. In the last, third gear of the mechanism-Figure 3c, which is a multiplier of rotational speed-the torque from the handrims is delivered to the planetary gear carrier (15) and is picked up from the sun gear (14) through the planetary gear (13) with internal teeth (12). teeth (12) through the planetary gear (13) and is delivered from the carrier (15). In the second, neutral gear- Figure 3b-the torque is delivered to the sun gear (14) through the planetary gear (14) with internal teeth (12) and it is passed on from here. In this case, the carrier (14) also rotates, but it is not picked up by the hub. In the last, third gear of the mechanism-Figure 3c, which is a multiplier of rotational speed-the torque from the handrims is delivered to the planetary gear carrier (15) and is picked up from the sun gear (14) through the planetary gear (13) with internal teeth (12). The internal gear hub prototype was made to a scale of 2:1. This was decided upon because of the assembly and testing processes simplification. For the same reason, the prototype was equipped with only a simplified gear shift mechanism. The device was mounted on a V300 Vermeiren wheelchair. The assembly required the use of a suitable adapter connecting the central axle of the hub with the wheelchair frame. The device was compatible with 24 inch, double-chamber rims and 3 mm diameter spokes.

EMG Measurements
Surface electromyography (EMG) was used to conduct the research on the efficiency of the discussed mechanism. Efficiency is understood here as the selection possibility of the drive system parameters so as to change the values of the forces that must be generated by the operator to drive the wheelchair (a change of the muscle load to meet the actual torque demand). Four upper limb muscles were selected for the measurements, which take part in the manual propulsion of the wheelchair. They included: the deltoid muscle anterior (A) and posterior (B), triceps brachii (C) and extensor carpi radialis longus (D). The muscles that take an active part in propelling the wheelchair were selected on the basis of the previously conducted research, which was described in [31]. A miniaturized EMG Mini DTS system for wireless testing, compatible with Noraxton's MyoResearch software was used to carry out the measurements. Round electrodes with gel (20 mm in diameter) were used in the study, which were placed in the central part of the abdomen of the examined muscles. Measurements were taken at a frequency of 1500 Hz. The research was based on the SENIAM project methodology [32] as well as information contained in the "ABC EMG" [33]. A detailed description of the methodology used to perform these measurements can be found in [34].

Subjects
View of the dynamometer, the examined wheelchair with the participant and the arrangement of the electrodes is presented in Figure 4.
One female participant, aged 25 years, with a body weight of 61 kg and the value of BMI 20.4 kg/m 2 was examined. The examined person was a student at the University of Physical Education in Poznań and volunteered as a test subject. She confirmed the voluntary willingness to participate in the study with a signature on the consent form and read the patient information form. The research has been positively evaluated by the Bioethical Commission at the Karol Marcinkowski Medical University in Poznan Poland, Resolution No. 1100/16 of 10 November 2016, under the guidance of prof. MD Chęciński P. for the research team led by Ph.D. Wieczorek B. The authors obtained a written consent of the examined person for the publication of research results with her participation. The examined person was not affected by a disability or physical impairment. This choice was due to safety issues and is justified by the prototype nature of the tested device. In an unlikely, but possible, emergency, the participant must be able to evacuate quickly from the test stand. This can only be guaranteed with a fully mobile person. The participant had some experience in handling a wheelchair because she had participated in numerous previous studies of a similar nature. The study aim was to prove the efficiency of the newly developed solution of the manual wheelchair propulsion system. To achieve this end, it was considered sufficient to use one person as the study was of a qualitative rather than a quantitative nature. It was considered crucial to determine whether changes in the parameters of the propulsion system, such as, for example, the gear ratio or wheelchair inclination, could produce measurable and unambiguous results. Studies on a larger population aimed at determining the quantitative nature of changes in the recorded results are currently in progress and their results will be published in the future.
the drive system parameters so as to change the values of the forces that must be generated by the operator to drive the wheelchair (a change of the muscle load to meet the actual torque demand). Four upper limb muscles were selected for the measurements, which take part in the manual propulsion of the wheelchair. They included: the deltoid muscle anterior (A) and posterior (B), triceps brachii (C) and extensor carpi radialis longus (D). The muscles that take an active part in propelling the wheelchair were selected on the basis of the previously conducted research, which was described in [31]. A miniaturized EMG Mini DTS system for wireless testing, compatible with Noraxton's MyoResearch software was used to carry out the measurements. Round electrodes with gel (20 mm in diameter) were used in the study, which were placed in the central part of the abdomen of the examined muscles. Measurements were taken at a frequency of 1500 Hz. The research was based on the SENIAM project methodology [32] as well as information contained in the "ABC EMG" [33]. A detailed description of the methodology used to perform these measurements can be found in [34].

Subjects
View of the dynamometer, the examined wheelchair with the participant and the arrangement of the electrodes is presented in Figure 4.

Test Conditions and Statistical Analysis
In order to carry out the experiments, the appropriate methodology and research plan were developed and its diagram is shown in Figure 5. The study consisted of two phases: the preparation phase (A1-A3) and the actual tests (B1-B4). In the first step, the person being examined was prepared by attaching measuring electrodes (A1) and performing exercises necessary to perform MVC normalization on the test results (a commonly used amplitude analysis technique for EMG signals) [32] (A2). Subsequently, the person was placed with the wheelchair on the dynamometer (A3) and the test was started (B1). After completing the task, the wheelchair operator rested for 15 min (B2). At the same time, the wheelchair gear ratio and/or inclination (B3) were changed. The inclination angle change, thanks to the used test stand, was performed in such a way that the larger wheel did not change its position in relation to the support rollers. The tests were repeated successively, until all variants were exhausted and they came to an end (B4).
The test consisted of performing a series of 30 propulsion phases, i.e., pushing off the wheels of a wheelchair. This number was selected so that it was possible to carry out all the defined variants of the experiment without the participant's physical discomfort (so as not to lead to excessive muscle exhaustion). Tests were performed for each of the 27 system configurations. They were obtained by changing: the gear ratio of the planetary propulsion system (p1-3) and the angle of inclination of the wheelchair together with the patient (k1-3). Such measurements were repeated in three series (s1-3). The push frequency (occurrence of the propulsion phases) was f = 40 BPM (occurrence approximately every 1.5 s) and it was sonically imposed with the use of a metronome-the examined person's task was to follow the impact rhythm as accurately as possible. Thanks to this, the average speed of the wheelchair was controlled. It was hypothesized that the participant's muscle effort would change with the gear ratio of the drive system and angle of inclination of the wheelchair. Table 1 shows variants of the markings and input values for the performed experiments. The markings assumed according to it are consistently preserved throughout the paper.
In order to carry out the experiments, the appropriate methodology and research plan were developed and its diagram is shown in Figure 5. The study consisted of two phases: the preparation phase (A1-A3) and the actual tests (B1-B4). In the first step, the person being examined was prepared by attaching measuring electrodes (A1) and performing exercises necessary to perform MVC normalization on the test results (a commonly used amplitude analysis technique for EMG signals) [32] (A2). Subsequently, the person was placed with the wheelchair on the dynamometer (A3) and the test was started (B1). After completing the task, the wheelchair operator rested for 15 min (B2). At the same time, the wheelchair gear ratio and/or inclination (B3) were changed. The inclination angle change, thanks to the used test stand, was performed in such a way that the larger wheel did not change its position in relation to the support rollers. The tests were repeated successively, until all variants were exhausted and they came to an end (B4). The test consisted of performing a series of 30 propulsion phases, i.e., pushing off the wheels of a wheelchair. This number was selected so that it was possible to carry out all  The measured EMG signals were rectified and then smoothed using RMS algorithms with a window width of 150 ms. Later, a maximum voluntary contraction test (MVC) was performed. This post-processing method utilizes a reference value to normalize the subsequent EMG data series ( Figure 5, marking A1). The output is displayed as a percentage of the MVC value, which can be used to easily establish a common ground when comparing the data between repetitions and subjects. A set of five dedicated exercises were carried out to test the maximum contraction of any one, which were selected on the basis of the previous studies [31]. The recorded data were successively normalized, using the arithmetic mean of the amplitude of the highest signal segment with a constant duration of 1000 ms as the reference value.
Subsequently, each experiment was divided into individual propulsion phases. On the basis of the measurements, the surface area (SA) was determined numerically, using the trapezoid method, under each obtained graph and the signal amplitude (MVC MAX ), separately for each propulsion cycle. Out of 30 values determined in this way (n), the 3 largest and the smallest ones (the most deviating from the average) were rejected, assuming that they were burdened with a large error. The arithmetic mean (x), (of the measurements processed in this way) was used as an estimator of the searched value. The standard deviation of the arithmetic mean (σ) was assumed as the estimator error.
Together with an increase in an operator's effort, the EMG signal reaches higher values; thus, as the load increases, the area under the graph (SA) also increases, which allows to assess the impact of the propulsion system ratio and the angle of inclination of a wheelchair on the work to be performed by the examined person for each propulsion cycle. The units that have particular axes of the graphs for the EMG test results were the ordinate axis (MVC), given as a percentage, and for abscissa it was time; therefore, the resulting unit of the value of SA is [s ·%].
The statistical analysis was carried out at the significance level set at α = 0.05. The Anderson-Darling test was used to determine the normal distribution of the measured data. Next, a two-parameter analysis of variance (ANOVA) was used to determine the effect of the gear ratio and the angle of inclination on the dependent variable, which was assumed to be the area under the standardized EMG (SA) signal graph. A Tukey's HSD post-hoc test was applied when a significant difference was detected in the between-subject factor. In addition, the mean of the amplitudes of the EMG signal values (MVC MAX ) was determined for individual propulsion cycles.

Motion Capture Measurements
The trajectory of the upper limb movement was determined using the motion capture procedure, in which a visual detection of the ArUco marker was performed with the use of OpenCV libraries [35]. The study used three moving markers attached to the different upper limb segments: the wrist (ID1), elbow (ID3) and shoulder (ID5) (as presented in post-hoc test was applied when a significant difference was detected in the between-subject factor. In addition, the mean of the amplitudes of the EMG signal values ( ) was determined for individual propulsion cycles.

Motion Capture Measurements
The trajectory of the upper limb movement was determined using the motion capture procedure, in which a visual detection of the ArUco marker was performed with the use of OpenCV libraries [35]. The study used three moving markers attached to the different upper limb segments: the wrist (ID1), elbow (ID3) and shoulder (ID5) (as presented in  The determined coordinates of the nonstationary markers in relation to the stationary marker (ID0) allowed for calculating the kinematic parameters describing the propelling and return phase of the movement of the upper limb during propulsion (as presented in Figure 7). In the motion capture analysis of the upper limb, the path of each marker during the propulsion and return phases and the total angle of the arc drawn by the wrist , were determined. Moreover, the performed analysis made it possible to observe the starting and ending position of the wrist. This position in the study was defined using angular coordinates. In the case of the beginning of the propulsion and return phases, the coordinates were: the length of initial leading radius and the initial radius inclination angle , while for the end positions, these were, respectively, the length of the final leading radius and the final radius inclination angle . The last parameters analyzed in the study were the surface areas within which the used markers were contained on the course of five full propulsion cycles. To determine the perimeter of these surface areas ( ), the alpha shape algorithm [36] was used, in which the alpha coefficient was in the range of 0.7-0.9. The analysis was performed for three measurement tests consisting of five full propulsion cycles. It was investigated how the surface areas of paths traveled by the individual markers changed during propulsion. Additionally, the total length and width of the designated shapes were also determined (as illustrated in Figure 7). The determined coordinates of the nonstationary markers in relation to the stationary marker (ID0) allowed for calculating the kinematic parameters describing the propelling and return phase of the movement of the upper limb during propulsion (as presented in Figure 7). In the motion capture analysis of the upper limb, the path of each marker during the propulsion and return phases s and the total angle of the arc drawn by the wrist ϕ, were determined. Moreover, the performed analysis made it possible to observe the starting and ending position of the wrist. This position in the study was defined using angular coordinates. In the case of the beginning of the propulsion and return phases, the coordinates were: the length of initial leading radius R START and the initial radius inclination angle ϕ START , while for the end positions, these were, respectively, the length of the final leading radius R END and the final radius inclination angle ϕ END . The last parameters analyzed in the study were the surface areas within which the used markers were contained on the course of five full propulsion cycles. To determine the perimeter of these surface areas (A ID ), the alpha shape algorithm [36] was used, in which the alpha coefficient was in the range of 0.7-0.9. The analysis was performed for three measurement tests consisting of five full propulsion cycles. It was investigated how the surface areas of paths traveled by the individual markers changed during propulsion. Additionally, the total length L and width H of the designated shapes were also determined (as illustrated in Figure 7).  Figures 8 and 9 show the selected values of (SA) for different gear ratios. All the calculated values and statistical data are presented in Table 2.     Figures 8 and 9 show the selected values of (SA) for different gear ratios. All the calculated values and statistical data are presented in Table 2       According to the analysis of the collected data, all designated areas (SA) were characterized by a normal distribution, although in one case the value (p val ) for the Anderson-Darling test was very close to the assumed significance level α (for sample s2k3p3, muscle D: p val = 0.05).

Results of EMG Measurements
The ANOVA test showed differences for all muscles and experiment variants between the individual parameters and their interactions. For nearly all muscles, a post-hoc test indicated a significant difference between the inclination angles and gear ratios. The deviations from this rule (statistically insignificant difference) were: s1 A variant p2 and p3 (p = 0.989), s3 A variant p2 and p3 (p = 0.537) and s3 B variant p1 and p2 (p = 0.668). The nature of the changes that were taking place can be observed in Figures 8 and 9. The effort of the wheelchair's operator (measured by the SA value) decreased as the angle of the inclination of the wheelchair decreased and the gear ratio increased. These changes were recorded consistently for each measurement test. The statistical analysis carried out allows for concluding that both the change in the angle of inclination of the wheelchair as well as the change in the gear ratio of its propulsion system, and their combination, influenced the determined values of the area (SA) under the MVC diagrams. The change in the gear ratio and the increase in the resistance to movement resulting from the angle of inclination affected the position of the wheelchair user in relation to the axis of the rear wheel. The nature of the kinematics of the movement of the upper limbs and muscle electromyography is similar to the results of the research by Laft et al., 2018, which analyzed the impact of the position of the rear wheel axle on a wheelchair's user [37]. The necessary statistical data are contained in Table 3. For all measurements, very small values (p val ) can be observed, so there is great certainty that the observed changes were significant. Table 3. The statistical data of the analysis of variance (ANOVA); F-F test value, p val -test probability of ANOVA analysis, Deltoid muscle anterior (A) and posterior (B), Triceps brachii (C), Extensor carpi radialis longus (D); markings of the angles of erection of the wheelchair: k1 = 0 • , k2 = 1.5 • and k1 = 5.4 • ; markings of experiment series: s1, s2 and s3; markings of the propulsion system gear ratios: p1 = 1.96, p2 = 1 and p3 = 0.51.    Table 4. As the analysis of the data provided shows, the changes occurring in the average EMG signal amplitude values were similar to the values of the areas.       The muscular effort required to propel the wheelchair increased with the angle of inclination of the wheelchair in relation to the ground level (k1 < k2 < k3). This conclusion could be drawn from the study by Yang et al. 2012 as well [38]. This is the result of the occurrence and subsequent increase in the so-called grade resistance. In addition, the increase in the inclination also causes a change in how the individual muscles of a person propelling a wheelchair are loaded because it affects the kinematics and geometry of the human-wheelchair system. Similar relationships were demonstrated by Qi et al. in 2013 [39]. The muscle load of the person propelling a wheelchair increases as the gear ratio increases (p3 < p2 < p1). This conclusion is confirmed by the studies by Wieczorek et al. concerning the testing of a wheelchair's driving force [40]. This is the expected result because, as is known, a speed reduction increases the value of the input torque, which is the multiplication of the diameter of the pulls and the propelling force generated by an operator's limbs. Figure 11 shows selected measurement results for the gear ratio variant p1 for the angles from k1 to k3. It presents a compilation of three distances travelled by the markers during the propulsion phases. Markers were placed on the joints of the forearm and upper arm. The analyzed trajectories were located on a plane parallel to the sagittal plane in the coordinate system with the origin in the axis of rotation of the wheelchair's drive wheel. Figure 12 shows the measurement of the upper limb trajectory for the variant of the inclination angle k1 and the gear ratios from p1 to p3. The analysis of the presented graphs allows for identification of significant differences caused by the change of the wheelchair's inclination angle and the change of its drive system gear ratio. As the wheelchair's inclination angle increased, the position (in the adopted coordinate system) of the range of motion of the upper limb's individual segments changed significantly. At the same time, the range of motion itself changed to a much lesser extent. The measured values concerning the wheelchair's inclination are consistent with the results of Lee et al. in 2017, where the influence of different directions of the handrim grip was analyzed [41]. There was also a noticeable relocation of the shoulder girdle shown by a marker (ID5). From Figure 11 it can be seen that as the angle of inclination of the wheelchair increased, the shoulder moved forward during propelling. This is consistent with the results of Morrow et al. in 2010 and Lafta in 2018 [42,43]. Such a change translates into a different load on the muscular system, which was noticeable in the analysis of the muscle activity. The path analysis of marker (ID1) allows for determining the total angle of the arc drawn by the wrist ϕ during the propulsion phase. The performed measurements show that the value of this angle was influenced by the gear ratio (p1-p3). It was noticed that as the load increased, the angle of rotation of the wheel during the propulsion phase decreased. For the tested wheelchair, it was noticeable at the multiplication ratio, for which the transmission characteristics caused an increase in the resistance torque on the handrims in relation to the resistance torque applied to the drive wheel. The analysis of the variability of the position of individual markers is important because this was the carrier of the information about the geometry and kinematics of the system and this influences a number of biomechanical parameters, including the position of the center of gravity. This statement is confirmed by the further studies by Wieczorek and Kukla on the location of the center of gravity during the propulsion of a wheelchair [44,45]. In this aspect, however, the position of the marker on the shoulder should be considered particularly important, because its position was closely related to the torso inclination angle in relation to the horizontal plane while driving the wheelchair. Due to the weight ratio of the torso to the other body segments above the waist, its inclination influenced the position of the center of gravity in the most significant way [46,47]. The set of determined values is presented in Table 5.        Figure 13 shows the total displacement of the analyzed upper limb joint s and the total angle of rotation ϕ of the drive wheel during the propelling phase (ID1). The angle of rotation of the wheel during the propulsion phase resulted from the analysis of the movement of the marker attached to the wrist (ID1), because its trajectory coincided with the curvature of the handrims. Additionally, for the marker (ID1), the greatest changes in the displacement value s were observed. The provided wrist kinematics data can be used in other research work focusing on the kinematics of propelling a manual wheelchair. With the increase in the gear ratio and with the increase in the wheelchair's angle of inclination, both the average value of the displacement and the average value of the angle covered by the marker (ID1) on the wrist of the examined person decreased. This was further confirmed by the analysis of the mean values of L and H as well as the mean value of the surface area A ID1 -as can be observed in Figure 14. A decreasing trend in the above-mentioned values could be observed with the change of the gear ratio (hence, the driving torque). The increase in the mean value of the surface area A ID1 as a function of the inclination angle can be explained by the change in orientation of the entire path traveled by the markers with respect to the stationary coordinate system, with the origin in the axis of rotation of the larger wheel. The lengths of L and H were always measured parallel, respectively, to the x and y axes which resulted in projecting their characteristic dimensions according to the principles of trigonometry. This change can be observed especially well by the comparison of the characteristic dimensions of L and H in the context of the ID3 marker in Figure 11. The set of determined values is presented in Table 6. This conclusion does not correlate with the studies by Lalumiere et al. 2013, which showed that overcoming terrain obstacles increases the forward flexion of the torso and increases the range of movement of the upper limbs and their loads [48]; however, this was about a different type of load. In the presented studies, it had a constant value for the entire duration of the experiment, while in [48] this character was of a more impactful nature as the curb ascent platform was used. For this reason, the load was temporary but of a great value.

Summary and Conclusions
The innovativeness of the presented works is related to the application of a unique mechanical system that performs functions rarely found in manual wheelchairs. The research approach novelty lies in the assessment and comparison method of the recorded results. The results of the EMG research were combined with the effects of the motion

Summary and Conclusions
The innovativeness of the presented works is related to the application of a unique mechanical system that performs functions rarely found in manual wheelchairs. The research approach novelty lies in the assessment and comparison method of the recorded results. The results of the EMG research were combined with the effects of the motion capture process. This is a new approach that allows for analyzing the interrelationship of  Most likely, the observed changes were a result of the increasing load on the muscles of the examined person [49,50]. In the case of a lesser load, the movements of the torso and upper limb were more extensive and the propulsion movement itself was smoother. On the other hand, when it was necessary to implement a greater driving torque at a given frequency of the propulsion phases (i.e., attempts to maintain a constant speed of movement), the greater load on the muscles of the tested person resulted in a smaller range of the upper body parts movements, especially the torso.
However, the indicated dependencies require further research. It is necessary to determine the quantitative effect of the gear ratio and inclination angle on the parameters of wheelchair propulsion biomechanics. An increased muscle load of the operator may also affect the propulsion pattern of the upper limb [51,52], but also the distribution of muscle strength into the tangential (driving) and centripetal force [53]. Qi et al., In 2021, noted that the fatigue of a wheelchair user, e.g., by climbing uphill, against the wind or on terrain with a high rolling resistance coefficient, affects the directional stability of the wheelchair and the user. Difficulties in moving often cause greater deviations of the measured parameters and there is also a change in the position of the center of gravity, which, according to Qi et al. in 2021, significantly affects the stability of movement in a wheelchair [54]. McLaurin and Brubaker in 1991 also observed the dependence of the stability of the movements in steering a wheelchair and the position of the center of gravity. It was noted that the additional resistance also influenced the frequency of movements and their stability. The conclusions of these authors are consistent with the research presented in this paper [55].
The limitation of the design of wheelchairs using gears is an increased resistance to motion, as demonstrated by Warguła et al. 2021 [56]; however, advanced gear shifting systems in the transmission can compensate for these losses and represent a new trend in wheelchair gear applications [57].

Summary and Conclusions
The innovativeness of the presented works is related to the application of a unique mechanical system that performs functions rarely found in manual wheelchairs. The research approach novelty lies in the assessment and comparison method of the recorded results. The results of the EMG research were combined with the effects of the motion capture process. This is a new approach that allows for analyzing the interrelationship of a number of biomechanical parameters at once. On the basis of the recorded data, it can be stated that the developed technical solution allows for adjusting the torque demand to the requirements and needs of the operator; thus, increasing the ergonomics of wheelchair usage. The analysis of the obtained results allows for concluding that the muscular effort of the participant propelling the wheelchair increases as the gear ratio increases (p3 < p2 < p1). This trend can also be observed in relation to the inclination of the wheelchair in (k1 < k2 < k3)-the greater the angle value, the greater the muscular effort.
An increasing muscle load while propelling the wheelchair changes the kinematics of the upper limbs' movement by reducing its displacement and area of motion. This also results in a reduced tilting movement in the torso during handrim propelling. The changes indicated above affect the position of the center of gravity of the human-wheelchair system, which in turn affects the resistance to motion. This is due to the different loads on the front and rear axles of the wheelchair, which have different rolling resistance coefficients.
A potential limitation of the presented research is the use of EMG and motion capture measurements only from one hand and the issue of bilateral symmetry during manual wheelchair propulsion remains open [58,59]. The results of recent research indicate that a larger scale of asymmetry may occur between the left and right limbs for one person than in the averaged data set for the selected side for a group of people [58][59][60][61]; however, the effect of this phenomenon in the presented studies would not affect their final conclusions. Regardless of the chosen limb, the muscle load and range of motion of the limb would change with the load, but the magnitude of these changes could be different between the left and right hand. This is an important issue that will need to be taken into account when conducting research on a larger population. This problem could be addressed by taking measurements from both hands or always from only dominant or nondominant ones.
The use of a stationary ergometer could be considered another limitation. These devices do not have the ability to perfectly reproduce the real conditions of moving around in a wheelchair. On the other hand, they allow for a controlled change of the movement parameters in a laboratory environment. It has been shown that the results obtained from tests on ergometers in the context of propulsion mechanics during steady-state propulsion are consistent with those recorded while moving on the ground [62].
It should be recognized that the aim of the works concerning the measurement issues have been achieved. It was possible to register and evaluate the selected biomechanical parameters resulting from changes in the load on the human-wheelchair system. The natural direction of the evolution of the presented research seems to be to conduct it on a larger population of people. These studies should be oriented both on the quantitative and qualitative determination of the changes taking place in the muscle effort and range of movement of the upper limbs. It may also be reasonable to compare the developed solution with other drive mechanisms that change the gear ratio between the handrims and the wheels of a wheelchair, but with the means of different mechanical transmissions, utilizing, for example, chain or belt transmissions.