A Validated HPLC-UV-ESI-IT-MS Method for the Quantification of Carnosol in Lepechinia mutica, a Medicinal Plant Endemic to Ecuador

The diphenolic diterpene carnosol was isolated from several species of the family Lamiaceae, including Lepechinia mutica, a medicinal plant endemic to Ecuador. The compound has exhibited high antioxidant, anti-inflammatory, antimicrobial, neuroprotective, and antifungal properties, as well as promising cytotoxicity against prostate, breast, skin, leukemia, and human colon cancer cell lines. In this paper, we developed and validated a simple, accurate, and reliable analytical HPLC-UV-ESI-IT-MS method, carried out on a C18 column, which is potentially suitable to quantify carnosol in plant extracts. The procedure complied with the established ICH validation parameters of analytical range (linearity in the range of 0.19–5.64 μg/g dried leaves; REAVERGE = 4.9%; R2 = 0.99907), analysis repeatability (RSD = 2.8–3.6%), intermediate precision (RSD = 1.9–3.6%), accuracy (estimated as % carnosol recovery in the range of 81 to 108%), and robustness. Finally, the LOD (0.04 µg/mg) and LOQ (0.19 μg/mg) values of carnosol/dried leaves were determined. Using this validated method, the content of carnosol in L. mutica was estimated to be 0.81 ± 0.04 mg/g of dried leaves (0.081%).

Molecules 2023, 28, x FOR PEER REVIEW 2 of 13 breast adenocarcinoma) cell lines, and against a Vero (African green monkey kidney) cell line [8].Carnosol exhibits higher cytotoxicity against gastric tumor AGS cells than normal fibroblasts [5].Moreover, carnosol induces apoptosis and prevents the growth of new vessels from tumors.This diterpene also contributes to the high antioxidant and anti-tumorogenesis properties of rosemary (Rosmarinus officinalis L.), and it is efficacious in the chemoprevention and/or chemotherapy of hormone-responsive cancers [4,[8][9][10].Carnosol (1) also inhibited neuroinflammation and produced anti-inflammatory effects in models of rheumatoid arthritis.The cytotoxicity and anti-inflammatory properties seem to be associated with the modulation of multiple deregulated pathways, including NF-kB, phosphatidylinositol-3-kinase, androgen and estrogen receptors, and PKC [10], and inhibitory effects on prostaglandin PGE2 formation [11].Finally, carnosol showed potent antifungal activity against the dermatophyte Microsporum canis and against Pyricularia oryzae [7].This fungus is responsible for blast disease, also known as rice rotten neck, which is the most important disease in rice cultures worldwide.Although clinical uses of L. mutica, as well as carnosol (1), have not been reported so far, compounds 1 and 2 are used as antioxidant preservatives in food and nonfood products, where they are labeled "extracts of rosemary" (E392) [2].Moreover, a prototype pharmacokinetic study was recently carried out with the determination of carnosol (1) in the plasma samples of a healthy volunteer after administration of rosemary extract [12].
Our research group isolated carnosol (1) during an extensive investigation of volatile and non-volatile constituents of extracts from Lepechinia mutica (Benth.)Epling and other Lepichinia species [7,[13][14][15].L. mutica is endemic to Ecuador [16,17], where the leaves are used as a traditional remedy to treat headache and nervous affections [18,19].In addition to the significant antifungal properties [7], we found that carnosol exhibited selective inhibitory activity against butyrylcholinesterase with an IC50 value less than that of the wellknown ChE inhibitors donepezil and allicin.These results make carnosol a potential therapeutic for the treatment and/or the prevention of AD [13,20].
The unique profile of biological activities displayed by carnosol (1) and the widespread ethnomedical uses [2] make the implementation and validation of a simple and reliable analytical method to quantify this diterpenoid in plant extracts highly desirable.In this paper, we describe an accurate high-performance liquid chromatography (HPLC)-UV-diode array detection (DAD)-electrospray ionization (ESI)-ion trap (IT)-mass spectrometry (MS) procedure, which we believe to be of general applicability for the quantification of carnosol in plant extracts.As a proband example, we quantified the content of carnosol (1) in an EtOAc extract of L. mutica.
HPLC-UV-ESI-IT-MS is an analytical technique with great versatility, sensitivity, and analytical power, where the individual capabilities of each technique are enhanced synergistically.While liquid chromatography separates mixtures of many components, mass spectrometry provides spectral information that may help to identify or confirm the suspected identity of each separated component.This hyphenated technique is thus widely used to separate, identify, and quantitate different secondary metabolites occurring in foods, drugs, plant extracts, and other biological matrices [21].HPLC-MS is especially Although clinical uses of L. mutica, as well as carnosol (1), have not been reported so far, compounds 1 and 2 are used as antioxidant preservatives in food and nonfood products, where they are labeled "extracts of rosemary" (E392) [2].Moreover, a prototype pharmacokinetic study was recently carried out with the determination of carnosol (1) in the plasma samples of a healthy volunteer after administration of rosemary extract [12].
Our research group isolated carnosol (1) during an extensive investigation of volatile and non-volatile constituents of extracts from Lepechinia mutica (Benth.)Epling and other Lepichinia species [7,[13][14][15].L. mutica is endemic to Ecuador [16,17], where the leaves are used as a traditional remedy to treat headache and nervous affections [18,19].In addition to the significant antifungal properties [7], we found that carnosol exhibited selective inhibitory activity against butyrylcholinesterase with an IC 50 value less than that of the well-known ChE inhibitors donepezil and allicin.These results make carnosol a potential therapeutic for the treatment and/or the prevention of AD [13,20].
The unique profile of biological activities displayed by carnosol (1) and the widespread ethnomedical uses [2] make the implementation and validation of a simple and reliable analytical method to quantify this diterpenoid in plant extracts highly desirable.In this paper, we describe an accurate high-performance liquid chromatography (HPLC)-UVdiode array detection (DAD)-electrospray ionization (ESI)-ion trap (IT)-mass spectrometry (MS) procedure, which we believe to be of general applicability for the quantification of carnosol in plant extracts.As a proband example, we quantified the content of carnosol (1) in an EtOAc extract of L. mutica.
HPLC-UV-ESI-IT-MS is an analytical technique with great versatility, sensitivity, and analytical power, where the individual capabilities of each technique are enhanced synergistically.While liquid chromatography separates mixtures of many components, mass spectrometry provides spectral information that may help to identify or confirm the suspected identity of each separated component.This hyphenated technique is thus widely used to separate, identify, and quantitate different secondary metabolites occurring in foods, drugs, plant extracts, and other biological matrices [21].HPLC-MS is especially suitable for the analysis of most polar and thermally labile metabolites that cannot be processed by gas chromatography-mass spectrometry.

Specificity of the Method
A sample of carnosol (1) was isolated from an EtOAc extract of Lepechinia mutica leaves following the procedure described in a previous paper [7].After crystallization, purified carnosol was used as the standard in the subsequent experiments.The identity of the isolated sample was confirmed by IR, NMR, UV spectra, and a comparison with the literature [1,7,22].
To get the best separation efficiency, different HPLC columns, compositions and elution time of the mobile phase, and column temperatures were tested and compared.Under optimized HPLC conditions (see Section 3.4), the chromatogram of the purified sample (Figure 2) showed a major peak, eluted with a retention time of 16.14 min, due to carnosol, which was flanked by two minor peaks, with a retention time of 15.78 min and 16.74 min, respectively.The first minor peak was tentatively identified as rosmarinic acid from the [M + H] + ion at m/z 361, while the other peak was unidentified.The peaks in the chromatogram were highly symmetric and base-separated.Moreover, the duration of the analysis was reasonable.
Molecules 2023, 28, x FOR PEER REVIEW 3 of 13 suitable for the analysis of most polar and thermally labile metabolites that cannot be processed by gas chromatography-mass spectrometry.

Specificity of the Method
A sample of carnosol (1) was isolated from an EtOAc extract of Lepechinia mutica leaves following the procedure described in a previous paper [7].After crystallization, purified carnosol was used as the standard in the subsequent experiments.The identity of the isolated sample was confirmed by IR, NMR, UV spectra, and a comparison with the literature [1,7,22].
To get the best separation efficiency, different HPLC columns, compositions and elution time of the mobile phase, and column temperatures were tested and compared.Under optimized HPLC conditions (see Section 3.4), the chromatogram of the purified sample (Figure 2) showed a major peak, eluted with a retention time of 16.14 min, due to carnosol, which was flanked by two minor peaks, with a retention time of 15.78 min and 16.74 min, respectively.The first minor peak was tentatively identified as rosmarinic acid from the [M + H] + ion at m/z 361, while the other peak was unidentified.The peaks in the chromatogram were highly symmetric and base-separated.Moreover, the duration of the analysis was reasonable.The identity of the major peak in the chromatogram as carnosol (1) was confirmed by the UV absorption spectrum (Figure 3a), which showed two characteristic bands with maximum absorbances at about 210 and 285 nm.Moreover, the UV spectra taken at the peak upslope and downslope, normalized, and compared, were identical, indicating that The identity of the major peak in the chromatogram as carnosol (1) was confirmed by the UV absorption spectrum (Figure 3a), which showed two characteristic bands with maximum absorbances at about 210 and 285 nm.Moreover, the UV spectra taken at the peak upslope and downslope, normalized, and compared, were identical, indicating that the peak was spectrally pure or that the peak corresponded to a single compound.On the other hand, the ESI-MS spectra, taken at different points of the peak, consistently showed a pseudo-molecular ion peak  Analyses conducted at regular time intervals indicated that the sample of the standard carnosol was stable and did not degrade under normal experimental conditions for the entire time of the experiments.Moreover, the purity of the standard was estimated to be 94.22% from the HPLC chromatogram (Figure 2).

Linearity Study of Standard Carnosol
The linearity of an analytical procedure is the ability to obtain a response that is directly proportional to the concentration (amount) of the analyte in samples within a given range [23][24][25][26].To determine the linearity of our method, a series of methanol solutions of the standard carnosol (1) were prepared, covering the range of 1-300 µg/mL.The corresponding areas of the UV absorption band at about 285 nm were electronically calculated, and the resulting graph (mAU vs. concentration) yielded the coefficient of determination, R 2 = 0.99907 (see Figure S1 in the SI).However, R 2 alone is not adequate to demonstrate linearity, since values greater than 0.999 can result even from data exhibiting curvature.Therefore, additional criteria are needed to confirm linearity [27,28], and % RE values of 15-20% were suggested as an acceptance criterion for linearity [28].Since carnosol concentrations within the range of 2-5 µg/mL were associated with % RE values ranging from 84 to 226, carnosol analysis was considered linear in the range of 10-300 µg/mL, where the values of % REAVERAGE and % REMAX were determined as 4.9 and 17.1, respectively.

Lowest Limit of Detection (LOD) and Lowest Limit of Quantitation (LOQ) of Standard Carnosol
The lowest limit of detection (LOD) [23][24][25][26] is the lowest quantity of a substance that can be distinguished from the absence of that substance (a blank value) with a certain confidence level (generally 99%).In other words, the LOD value is the concentration of an analyte that gives rise to a signal statistically significantly larger than the signal arising from the repeated measurements of a blank.Thus, it is the lowest concentration of an Analyses conducted at regular time intervals indicated that the sample of the standard carnosol was stable and did not degrade under normal experimental conditions for the entire time of the experiments.Moreover, the purity of the standard was estimated to be 94.22% from the HPLC chromatogram (Figure 2).

Linearity Study of Standard Carnosol
The linearity of an analytical procedure is the ability to obtain a response that is directly proportional to the concentration (amount) of the analyte in samples within a given range [23][24][25][26].To determine the linearity of our method, a series of methanol solutions of the standard carnosol (1) were prepared, covering the range of 1-300 µg/mL.The corresponding areas of the UV absorption band at about 285 nm were electronically calculated, and the resulting graph (mAU vs. concentration) yielded the coefficient of determination, R 2 = 0.99907 (see Figure S1 in the SI).However, R 2 alone is not adequate to demonstrate linearity, since values greater than 0.999 can result even from data exhibiting curvature.Therefore, additional criteria are needed to confirm linearity [27,28], and % RE values of 15-20% were suggested as an acceptance criterion for linearity [28].Since carnosol concentrations within the range of 2-5 µg/mL were associated with % RE values ranging from 84 to 226, carnosol analysis was considered linear in the range of 10-300 µg/mL, where the values of % RE AVERAGE and % RE MAX were determined as 4.9 and 17.1, respectively.

Lowest Limit of Detection (LOD) and Lowest Limit of Quantitation (LOQ) of Standard Carnosol
The lowest limit of detection (LOD) [23][24][25][26] is the lowest quantity of a substance that can be distinguished from the absence of that substance (a blank value) with a certain confidence level (generally 99%).In other words, the LOD value is the concentration of an analyte that gives rise to a signal statistically significantly larger than the signal arising from the repeated measurements of a blank.Thus, it is the lowest concentration of an analyte in a sample that can be detected but not necessarily quantified.On the other hand, the LOQ value is defined as the lowest concentration of an analyte that can be quantitatively determined with acceptable precision and accuracy.The limit of carnosol detection (LOD) was determined to be 2 µg/mL, based on a signal-to-noise ratio of 3.4.Furthermore, the limit of carnosol quantification (LOQ) was set at 10 µg/mL, which corresponded to the lowest concentration at which linearity was observed.These results demonstrate that the HPLC-UV-ESIMS method is sensitive for the quantitative determination of carnosol in plant extracts.

Quantification of Carnosol in Lepechinia mutica
Having demonstrated good linearity for the reference sample, we estimated the content of carnosol (1) in an EtOAc extract of dried leaves of L. mutica using the same optimized HPLC conditions as those used for the standard.A sample of the residue resulting from evaporation of the extract was redissolved in MeOH-H 2 O, 9:1 v/v, freed from chlorophylls, and analyzed in three replicates.Carnosol was eluted with a retention time of 16.17 ± 0.04 min (Figure 4).The UV and ESI-MS spectra taken at the peak upslope and downslope revealed peak homogeneity.The mean concentration of carnosol (1), estimated using the calibration curve, was 0.81 ± 0.04 mg/g (0.081% w/w) of the dried leaves.Thus, the calculated linear range, LOD, and LOQ of carnosol were 0.19-5.64,0.04, and 0.19 mg/g of the dried leaves, respectively.
tively determined with acceptable precision and accuracy.The limit of carnosol detection (LOD) was determined to be 2 µg/mL, based on a signal-to-noise ratio of 3.4.Furthermore, the limit of carnosol quantification (LOQ) was set at 10 µg/mL, which corresponded to the lowest concentration at which linearity was observed.These results demonstrate that the HPLC-UV-ESIMS method is sensitive for the quantitative determination of carnosol in plant extracts.

Quantification of Carnosol in Lepechinia mutica
Having demonstrated good linearity for the reference sample, we estimated the content of carnosol (1) in an EtOAc extract of dried leaves of L. mutica using the same optimized HPLC conditions as those used for the standard.A sample of the residue resulting from evaporation of the extract was redissolved in MeOH-H2O, 9:1 v/v, freed from chlorophylls, and analyzed in three replicates.Carnosol was eluted with a retention time of 16.17 ± 0.04 min (Figure 4).The UV and ESI-MS spectra taken at the peak upslope and downslope revealed peak homogeneity.The mean concentration of carnosol (1), estimated using the calibration curve, was 0.81 ± 0.04 mg/g (0.081% w/w) of the dried leaves.Thus, the calculated linear range, LOD, and LOQ of carnosol were 0.19-5.64,0.04, and 0.19 mg/g of the dried leaves, respectively.
The intense peak 2 in the HPLC chromatogram (Figure 4), with a retention time of 19.25 min, was tentatively assigned to carnosic acid (2) from the UV spectrum and the [M + H] + ion at m/z 333, which was in accordance with the MW = 332.44g/mol of compound 2 (C20H28O4).However, the lack of a standard in our hands prevented us from developing a validated quantification method for carnosic acid in the extract.

Analytical Method Validation
To validate the analytical method, the following parameters were determined according to the IUPAC and the International Conference on Harmonization (ICH) guidelines [23,24]: precision (repeatability and intermediate precision), accuracy, and robustness.The intense peak 2 in the HPLC chromatogram (Figure 4), with a retention time of 19.25 min, was tentatively assigned to carnosic acid (2) from the UV spectrum and the [M + H] + ion at m/z 333, which was in accordance with the MW = 332.44g/mol of compound 2 (C 20 H 28 O 4 ).However, the lack of a standard in our hands prevented us from developing a validated quantification method for carnosic acid in the extract.

Analytical Method Validation
To validate the analytical method, the following parameters were determined according to the IUPAC and the International Conference on Harmonization (ICH) guidelines [23,24]: precision (repeatability and intermediate precision), accuracy, and robustness.

Precision
The precision of an analytical procedure is defined as the closeness of agreement (degree of scatter) between a series of measurements obtained from multiple samplings of the same homogeneous sample under the prescribed conditions [23][24][25][26].Precision may be considered at the levels of repeatability and intermediate precision.

Repeatability
Repeatability is a measure of the variability in results experienced by a single analyst on a single instrument using the same operating conditions over a short timescale [23][24][25][26].During validation, repeatability is performed by analyzing multiple replicates of a homogeneous sample using the analytical method.The mean and the standard deviation (SD) are calculated, and the precision is expressed as the relative standard deviation (RSD), given by Equation (1): In our experiments, three samples (5, 10, and 15 mg) of residue A' from the EtOAc extract of dried leaves were analyzed separately in three consecutive days by an analyst (analyst 1), following the procedure described in Section 2.4.The concentration (mg/g) of carnosol (1) was determined in each analysis by interpolation of the calibration curve.The repeatability of the method was determined to be excellent, yielding a calculated relative standard deviation (RSD) ranging from 2.8 to 3.6% (see Table S1 in the SI).This range comfortably falls within the AOAC's recommended value of approximately 3.7% for analyte concentrations below 1 mg/g, indicating the reliability of the method [29].Moreover, the analysis of variance (ANOVA) indicated no significant difference between the results, as evidenced by an F-value of 2.87 and a p-value of 0.134.

Intermediate Precision
In analytical method validation, 'intermediate (measurement) precision', sometimes referred to as 'within-laboratory reproducibility', gives an estimate of the variation in results when measurements are made in a single laboratory but under conditions that are more variable than repeatability conditions.The aim is to obtain a precision estimate that reflects all sources of variation that may occur in a single laboratory under routine conditions (different analysts, extended timescale, different pieces of equipment, etc.) [23][24][25][26].To validate our analytical method, intermediate precision was estimated by comparing the results obtained by analyst 1 (see Section Repeatability) with those obtained by another analyst (analyst 2), who quantified carnosol in three samples (5, 10, and 15 mg) of residue A' in three consecutive days under identical experimental conditions.The relative standard deviations (RSDs) ranged between 1.9 and 3.6% (see Table S2 in the SI).
The ANOVA results indicated that there were no significant differences between the results of the two analysts across three concentrations spanning the analytical range of the method.Moreover, these RSD values were less than the recommended reproducibility threshold of 6% established by the Association of Official Agricultural Chemists (AOAC) [29].These findings provided confidence in the intermediate precision of the method, making it suitable for use in future studies and applications.

Accuracy (% Recovery)
The accuracy, sometimes termed trueness, of an analytical procedure expresses the nearness (closeness of agreement) between the value that is accepted as an accepted reference value and the measured value.Accuracy can also be described as the extent to which the test results generated by the method and the true (or accepted) reference value agree [23][24][25][26].
The accuracy of the developed method was evaluated by adding known amounts of standard carnosol to samples of L. mutica extract and comparing the extracted amounts to the added amounts.Each of the two analysts 1 and 2 carried out nine experiments in three consecutive days (see Section 3.8.2).The graphs of the percent carnosol recovery (% R) for the two groups of analyses (Figure 5) showed a normal distribution of values, ranging from 81% to 108%, with a mean value of 96% over the entire range.These values were within the range acceptable by the AOAC validation guidelines [29], indicating that the method was deemed precise and accurate.the added amounts.Each of the two analysts 1 and 2 carried out nine experiments in t consecutive days (see Section 3.8.2).The graphs of the percent carnosol recovery (% R the two groups of analyses (Figure 5) showed a normal distribution of values, ran from 81% to 108%, with a mean value of 96% over the entire range.These values w within the range acceptable by the AOAC validation guidelines [29], indicating tha method was deemed precise and accurate.

Robustness
The robustness of an analytical procedure is a measure of its capacity to remain affected by small but deliberate variations in method parameters, and it provides an i cation of its reliability during normal usage [24].Among the large number of potent variable parameters, we assumed flow rate, temperature, UV detection wavelength, bile phase composition, and gradient changes to be the most reasonably variable par eters in the analytical conditions.The resolution of 0.8 between the peaks of rosma acid and carnosol was considered the most important performance characteristic o method.According to the one-factor-at-a-time (OFAT) approach [30], the selected par eters were changed one by one in both directions from the nominal (optimal) value.effects of the changes in the parameters on the peak resolution were monitored by rec ing the HPLC chromatogram for each experiment.The results are reported in Table 1

Robustness
The robustness of an analytical procedure is a measure of its capacity to remain unaffected by small but deliberate variations in method parameters, and it provides an indication of its reliability during normal usage [24].Among the large number of potentially variable parameters, we assumed flow rate, temperature, UV detection wavelength, mobile phase composition, and gradient changes to be the most reasonably variable parameters in the analytical conditions.The resolution of 0.8 between the peaks of rosmarinic acid and carnosol was considered the most important performance characteristic of the method.According to the one-factor-at-a-time (OFAT) approach [30], the selected parameters were changed one by one in both directions from the nominal (optimal) value.The effects of the changes in the parameters on the peak resolution were monitored by recording the HPLC chromatogram for each experiment.The results are reported in Table 1.The RSD values among measurements ranged from 2.7 to 4.9% (Table 1), which indicated the robustness of the method.

General Information
All solvents were reagent grade or HPLC grade and were purchased from Sigma-Aldrich (St. Louis, MO, USA).Formic acid was supplied by Sigma-Aldrich.A highperformance liquid chromatography (HPLC) Dionex UltiMate 3000 instrument, with a photodiode array detector model DAD-3000 (RS), coupled to an Amazon Speed Ion-Trap mass spectrometer analyzer (AmaZon speed, Bruker, Billerica, MA, USA) with an electrospray ionization source (ESI), was used.The system was controlled using Bruker Daltonics HyStar 3.2 software with an Ethernet data interface.An Agilent Technology Eclipse Plus stainless steel column (Santa Clara, CA, USA) filled with a C18 reversed phase, particle size 5 µm, internal diameter 2.1 mm, and length 150 mm, was used in all the HPLC separations.The UV diode array detector (DAD) was set at 285 nm.

Standard Carnosol
Three lots of freshly collected leaves of L. mutica (600 + 400 + 200 g) were separately dried at 35 • C under forced air circulation.Subsequently, each dried lot was ground in liquid nitrogen and soaked in EtOAc (1.2 L) for 1 h at room temperature, using an ultrasonic bath at 35 • C in the last 15 min.The extraction was repeated three times for each lot to obtain, after solvent evaporation, three dense oily residues.They were pooled together to provide a total of 24.1 g (residue A).Subsequently, raw carnosol (1, 38.47 mg, 0.015% w/w dry leaves) was isolated from a sample of A (1 g) by medium-pressure column chromatography, according to the procedure described in a previous paper [7].Double recrystallization from EtOH yielded purified 1 as a white powder (5.12 mg, 0.0020% w/w dry leaves), which was used as a working standard in the experiments.The identity of isolated carnosol was confirmed using IR NMR, UV, and MS spectra (Figure 3), and comparison with the literature [1,7,22].A purity of 94.22% was estimated for the standard by comparing the area of the peak at 16.14 min with the total area of the peaks in the HPLC-MS chromatogram (Figure 2), without applying a correction factor.For preparing each test solution, the carnosol concentration was adjusted according to the purity of the standard.The standard carnosol (1) was stored under N 2 in sealed amber vials at −15 • C until analyses, which were performed within three weeks after the storage.

HPLC Analysis of Standard Carnosol
Different mobile phases, column types, and stationary phase size combinations, as well as different flow rates and column temperatures, were tested to obtain the most efficient chromatographic separation of carnosol from impurities.Under optimized HPLC conditions, the mobile phase was constituted of two mixtures: A (30% acetonitrile/70% water/0.1% formic acid) and B (60% acetonitrile/40% water/0.1% formic acid).They were mixed according to the following gradient program: 100% A, from 0 to 4 min, followed by 50% A/50% B, from 4 to 17 min, followed by 100% B, from 17 to 20 min, followed by 100% A, from 20 to 25 min.The flow rate and injection volume in each analysis, carried out at 22 • C, were 0.5 mL/min and 0.5 µL, respectively.With this method, the retention time of carnosol (1) was 16.14 ± 0.04 min (Figure 2).

MS Chromatographic Conditions
The ionization was set in positive mode, with the following parameters: voltage of the ionization capillary: 4.500 V; mass range: m/z 100-2000; nebulizer set at 26.0 psi; nitrogen drying temperature: 200 • C; dry gas (N 2 ): 6.0 L/min; rolling averages: 2 cts; averages: 2, ICC (ion charge control): on.The instrument was controlled using Hystar 3.2 software (Bruker, Billerica, MA, USA).To determine the analytical range and the LOD and LOQ values, 13 solutions (1, 2, 3, 4, 5, 10, 25, 50, 100, 150, 200, 250, and 300 µg/mL) of standard carnosol (1) were prepared by diluting with MeOH an intermediate solution (100 µg/mL) or a stock solution (1 mg/mL) of the standard in MeOH.Each solution, after having been filtered through a syringe filter (0.22 µm pore size), was injected (0.5 µL) into the HPLC-UV-ESI-IT-MS equipment.For each dilution, three replicates were carried out, and the UV spectrum at the top of the carnosol peak in the chromatogram was recorded (Figure 3).The mean area of the absorption band with a maximum at about 285 nm was measured, and a calibration curve (y = 2.0857x − 6.7954) of concentration (µg/mL) vs. absorption area (mAU, was plotted (Figure S1 in the SI).To define the linearity range, the lowest and highest concentrations producing a %RE lower than 20 were selected.The limit of detection (LOD) was established as the lowest concentration that yielded a signal-to-noise ratio greater than 3.The limit of quantification (LOQ) was set as the lowest concentration within the linearity range.

Quantification of Carnosol in Lepechinia mutica Leaves
Three samples (250, 100, and 50 g) of dried leaves of Lepechinia mutica were separately extracted with EtOAc according to the procedure described in Section 3.3.After solvent evaporation, the resulting residues were pooled together to give a total of 50 g (residue A').A sample (10 mg) of A' was dissolved under sonication in 1.5 mL MeOH-H 2 O, 9:1 v/v, and quickly filtered through a regenerated cellulose syringe filter (0.2 µm pore size; Millipore, Bedford, MA, USA) to remove chlorophyll.A sample (5 µL) of the filtrate was injected into the HPLC-UV-ESI-IT-MS instrument.The analysis was repeated three times, and the mean concentration of carnosol = 43.1 ± 2.4 µg/mL was determined from the mean area of the UV band of (1), by interpolation of the calibration curve of the standard.The content C of carnosol (1) in the dried leaves of L. mutica was then calculated as 0.81 mg/g (0.081% w/w) using Equation (2): where: C: Content (mg/g) of carnosol (1)  Two analysts, 1 and 2, working independently with the same chromatographic equipment and following the procedure described in Section 3.7, determined the content of carnosol (mg/g dried leaves) in three batches (5, 10, and 15 mg) of extract A'.Each determination was carried out in triplicates in three consecutive days, and the mean, the standard deviation (SD), and the % RSD (=SD × 100/mean) were calculated for each group of experiments.The method repeatability was estimated from the data set obtained by analyst 1 (Table S1 in the SI), whereas the data sets obtained by analysts 1 and 2 (Tables S1 and S2 in the SI) were compared among themselves to estimate the intermediate precision.
[M + H] + at m/z 331.29 [C 20 H 27 O 4 ] + (Figure 3b), in accordance with the MW = 330.42g/mol of carnosol (C 20 H 26 O 4 ).Moreover, the ESI-MS 2 spectrum (Figure 3c) showed an ion at m/z 287.01, corresponding to the loss of CO 2 from the [M + H] + ion.Molecules 2023, 28, x FOR PEER REVIEW 4 of 13 the peak was spectrally pure or that the peak corresponded to a single compound.On the other hand, the ESI-MS spectra, taken at different points of the peak, consistently showed a pseudo-molecular ion peak [M + H] + at m/z 331.29 [C20H27O4] + (Figure 3b), in accordance with the MW = 330.42g/mol of carnosol (C20H26O4).Moreover, the ESI-MS 2 spectrum (Figure 3c) showed an ion at m/z 287.01, corresponding to the loss of CO2 from the [M + H] + ion.

Figure 5 .
Figure 5. Graphs of the carnosol % recovery (% R) for the two groups of analyses in the accu test.

3. 6 .
Linearity Range, Limit of Detection (LOD), and Limit of Quantitation (LOQ) of Standard Carnosol in the dried leaves Co: Mean concentration of carnosol (1) in the dissolved sample of A' = 43.1 • 10 −3 mg/mL V: Volume (1.5 mL) of MeOH-H 2 O, 9:1 v/v, used to dissolve the sample of A' W A : Weight of total residue A' = 50 • 10 3 mg W B : Weight of the sample of A' from which chlorophyll was removed = 10 mg W: Weight of dried leaves = 400 g 3.8.Validation of the Analytical Method 3.8.1.Precision (Repeatability and Intermediate Precision)

Table 1 .
Changes in the parameters used to evaluate the robustness of the analytical method.Graphs of the carnosol % recovery (% R) for the two groups of analyses in the accuracy test.

Table 1 .
Changes in the parameters used to evaluate the robustness of the analytical method.