Negative Reagent Ions for Real Time Detection Using SIFT-MS
AbstractDirect analysis techniques have greatly simplified analytical methods used to monitor analytes at trace levels in air samples. One of these methods, Selected Ion Flow Tube-Mass Spectrometry (SIFT-MS), has proven to be particularly effective because of its speed and ease of use. The range of analytes accessible using the SIFT-MS technique has been extended by this work as it introduces five new negatively charged reagent ions (O−, OH−, O2−, NO2−, and NO3−) from the same microwave powered ion source of moist air used to generate the reagent ions traditionally used (H3O+, NO+, and O2+). Results are presented using a nitrogen carrier gas showing the linearity with concentration of a number of analytes not readily accessible to positive reagent ions (CO2 from ppbv to 40,000 ppmv, sulfuryl fluoride and HCl). The range of analytes open to the SIFT-MS technique has been extended and selectivity enhanced using negative reagent ions to include CCl3NO2, SO2F2, HCN, CH3Cl, PH3, C2H4Br2, HF, HCl, SO2, SO3, and NO2. View Full-Text
Scifeed alert for new publicationsNever miss any articles matching your research from any publisher
- Get alerts for new papers matching your research
- Find out the new papers from selected authors
- Updated daily for 49'000+ journals and 6000+ publishers
- Define your Scifeed now
Hera, D.; Langford, V.S.; McEwan, M.J.; McKellar, T.I.; Milligan, D.B. Negative Reagent Ions for Real Time Detection Using SIFT-MS. Environments 2017, 4, 16.
Hera D, Langford VS, McEwan MJ, McKellar TI, Milligan DB. Negative Reagent Ions for Real Time Detection Using SIFT-MS. Environments. 2017; 4(1):16.Chicago/Turabian Style
Hera, David; Langford, Vaughan S.; McEwan, Murray J.; McKellar, Thomas I.; Milligan, Daniel B. 2017. "Negative Reagent Ions for Real Time Detection Using SIFT-MS." Environments 4, no. 1: 16.
Note that from the first issue of 2016, MDPI journals use article numbers instead of page numbers. See further details here.