- freely available
Viruses 2013, 5(9), 2272-2281; doi:10.3390/v5092272
Published: 19 September 2013
Abstract: Hantaviruses are members of the family Bunyaviridae and are an emerging cause of disease worldwide with high lethality in the Americas. In Brazil, the diagnosis for hantaviruses is based on immunologic techniques associated with conventional RT-PCR. A novel one-step SYBR Green real-time RT-PCR was developed for the detection and quantitation of Araraquara hantavirus (ARAV) and Rio Mamore hantavirus (RIOMV). The detection limit of assay was 10copies/µL of RNA in vitro transcribed of segment S. The specificity of assay was evaluated by melting curve analysis, which showed that the Araraquara virus amplified product generated a melt peak at 80.83 ± 0.89 °C without generating primer-dimers or non-specific products. The assay was more sensitive than conventional RT-PCR and we detected two samples undetected by conventional RT-PCR. The one-step SYBR Green real-time quantitative RT-PCR is specific, sensible and reproducible, which makes it a powerful tool in both diagnostic applications and general research of ARAV and RIOMV and possibly other Brazilian hantaviruses.
Hantaviruses are members of the Bunyaviridae family and possess a trisegmented negative-strand RNA genome, categorized according to small (S), medium (M) and large (L) segments . Hantaviruses are transmitted to humans by the inhalation of aerosols of excreta or direct contact with infected rodents, although human-to-human transmission has been documented only for the Andes virus in South America . Severe diseases caused by hantaviruses are a serious public health problem worldwide. In the Americas, hantavirus infections produce a severe disease known as Hantavirus Pulmonary Syndrome (HPS), which is characterized by fever, respiratory failure, shock and a high case fatality ratio [1,2].
The first HPS cases in Brazil were diagnosed in 1993. Since then, approximately 1,600 HPS cases have been reported, with a 39% case fatality ratio . Currently six hantaviruses are found in several species of wild rodents in Brazil: Juquitiba virus (Oligoryzomys nigripes); Araraquara virus (ARAV) (Necromys lasiurus); Castelo dos Sonhos virus (Oligoryzomys aff. moojeni and Oligoryzomys utiaritensis); Laguna Negra-like virus (Calomys aff. callosus and Calomys laucha); Anajatuba virus (Oligoryzomys fornesi) and Rio Mamore virus (RIOMV) (Oligoryzomys sp) [1,4,5,6].
Serological methods are commonly used for hantavirus diagnosis, including enzyme-linked immunosorbent assays (ELISAs), immunofluorescence assays and immunoblot assays [1,7,8]. Additionally, hantavirus isolation in Vero E6 cell cultures and detection of hantavirus antibodies by plaque reduction neutralization are also used for diagnosis. Nonetheless, both methods require labor-intensive, time-consuming and biosafety level 3 conditions, although these are hardly used in diagnostics [1,7].
Nucleic acid-based methods, such as polymerase chain reaction preceded by reverse transcription (RT-PCR), are suitable for the detection of hantaviruses [9,10,11]. In Brazil, a conventional RT-PCR was able to detect Araraquara hantavirus genomes in blood and tissue samples of humans and rodents . Nevertheless, molecular techniques can evolve into faster, more sensitive methods that allow for the detection and quantitation of hantaviruses in several samples.
Recently, real-time RT-PCR assays have been developed using TaqMan probes for the diagnosis and/or quantification of hantaviruses such as Puumala virus, Andes virus, Hantaan virus, New York virus, Tula virus, Seoul virus, Dobrava virus and Sin Nombre virus [10,11,13,14,15]. In addition, a real-time RT-PCR assays using SYBR Green system for diagnosis of Dobrava virus in patients with HFRS were developed . In Brazil, using this same method, a real-time RT-PCR for detection of Juquitiba virus infection in rodents captured in the Atlantic rainforest was developed, but did not allow for viral quantitation .
In summary, real-time RT-PCR has many advantages over conventional RT-PCR, including rapidity, higher sensitivity and quantitative measurement . Therefore, in this study, we developed and validated a quantitative SYBR Green-based real-time RT-PCR assay for ARAV and RIOMV, being the most virulent among hantavirus genus and most commonly found in Brazil.
2. Materials and Methods
2.1. Serum Sample and Viral RNA Extraction
The RNA of samples of human and wild rodents was extracted using the QIAamp viral RNA extraction kit (Qiagen, Hilden, Germany) according to the manufacturer’s protocol. The RNA was recovered in 50 µL of RNase-free water with 40 units of recombinant RNase-OUT recombinant inhibitor (Invitrogen, Carlsbad, CA, USA) and stored at −70 °C.
2.2. Plasmid Cloning
An amplicon of ~264 nucleotides of the S segment of ARAV generated using primers, as described by Moreli et al. , was cloned in the pCR-TOPO 2.1 vector and introduced in Escherichia coli DH5α One Shot (Invitrogen, Carlsbad, CA, USA) according to the manufacturer’s protocol. From a selected transformant containing the desired construct, plasmid DNA was isolated using the QIAprep Spin Miniprep Kit (Qiagen, Hilden, Germany). The plasmid was sequenced to confirm the insertion and showed 99% sequence identity to the RNA S segment of ARAV (EF571895.1).
2.3. In Vitro Transcription
Plasmids containing the insert were linearized by digestion with BamH I enzyme (Fermentas, Vilnius, Lithuania). The linearized DNA plasmid was subsequently purified using the QIAquick PCR purification kit (Qiagen, Hilden, Germany) and in vitro transcribed using T7 RNA polymerase (Invitrogen, Carlsbad, CA, USA) following the manufacturer’s instructions. Residual DNA was removed by treatment with DNAse I (Invitrogen, Carlsbad, CA, USA) for 2 hours. Subsequently, RNA purification was performed using RNeasy kit (Qiagen, Hilden, Germany) and stored at −70 °C. In addition the products were subjected to PCR with primers described by Moreli et al.  to verify the complete degradation of plasmid DNA.
2.4. Standardization of the One-Step SYBR Green I Real-Time Quantitative RT-PCR
Quantitative RT-PCR was performed using the StepOnePlus™ Real-Time PCR System (Applied Biosystems, Foster City, CA, USA) and the SuperScript III Platinum One-Step qRT-PCR Kit (Invitrogen, Carlsbad, CA, USA) according to the manufacturer’s instructions. The optimal reaction mixture contained 2 µL of RNA template; 0.5 µL of MgSO2; 0.35 µL of each primer (ARAV forward and ARAV reverse ) (10 mM); 10 µL of SYBR buffer (2×); 0.5 µL of SuperScript III Platinum and 7.7 µL of MilliQ water in a final 20 µL volume. The temperature cycles to optimal reaction was 50 °C for 20 min for the production of cDNA; 95 °C for 5 min to activate the Taq polymerase and separate double-stranded DNAs; and 45 cycles at 95 °C for 15 s for denaturation; 55 °C for 25 s for primer annealing and 72 °C for 35 s for extension. The melting curve of the reaction (TM) was used to determine the specificity of amplified products. The TM curve was obtained by performing a thermal cycle of 95 °C for 15 s, decreased to 60 °C for 1 min, and increased again to 95 °C for 15 s. The reactions were carried out in a 96-well optical reaction plates or MicroAmp reaction tubes (Applied Biosystems, Foster City, CA, USA). The size of all obtained amplicons was determined by electrophoresis in 1.5% agarose gel.
2.5. Quantitation of the ARAV RNA Based on a Standard Curve
RNA concentration was determined by spectrophotometry in a Thermo Scientific NanoDrop™ 1000 Spectrophotometer (NanoDrop Technologies, Wilmington, DE, USA). The measurements were performed in duplicate and the concentration, in copies/μL, was converted to copy number using the following formula: RNA copy number (copies/μL) = (RNA concentration (g/μL)/number of nucleotides of transcript × 340) × 6.022 × 1023. The linear range of quantitation of the one-step real-time RT-PCR assay was determined by using ten-fold serial dilutions of RNA transcribed by develop of standard curve. For each new reaction, a standard curve was developed based in dilutions of in vitro-transcribed RNA and generation of a standard curve.
2.6. Specificity, Reproducibility and Detection Limit of the Reaction
The specificity of the one-step SYBR Green I quantitative RT-PCR was checked with other RNA viruses, including Coxsackie B5, Influenza A, Human Respiratory Syncytial virus A, Parainfluenza-2, Metapneumovirus, Oropouche and Rhinovirus-39. RIOMV in cell culture have also been tested.
The reproducibility of the one-step SYBR Green real-time quantitative RT-PCR was analyzed based on the TM, slope and R2 values obtained from decimal dilutions (108–103 copies/μL) of the ARAV RNA in vitro-transcribed assayed in duplicate. Likewise, the detection limit of the reaction was determined by testing between 3.7 × 1011 copies/μL to 3.7 × 10−3 copies/μL of in vitro transcribed ARAV RNA in triplicate.
2.7. Comparison of the One-Step SYBR Green I Real-Time Quantitative RT-PCR with Conventional RT-PCR
The efficiency of amplification and detection of ARAV genome by the one-step SYBR Green real-time quantitative RT-PCR was compared with conventional RT-PCR . We extracted the RNA from serum samples of HPS patients H1-H20 and macerated lung tissues from Necromys lasiurus rodents N1-N10 using the QIAamp viral RNA extraction kit (Qiagen, Hilden, Germany) and tested our method with both samples. In addition, the serum samples were tested for IgG and IgM to hantavirus via ELISA using the recombinant nucleoprotein (N) of ARAV as an antigen .
3.1. The Standard Curve of the One-Step SYBR Green I Real-Time Quantitative RT-PCR
The concentration of ARAV RNA in vitro-transcribed was 8.4 ng/μL, and it contained 3.7 × 1011 copies/μL of in vitro transcribed ARAV RNA. Equivalent ΔRn amplification plot curves were obtained when testing ARAV RNA at different dilutions in duplicate with values of 0.055305. Decimal dilutions of the ARAV RNA were tested, and their CT values were plotted as the standard curve of the reaction. The parameters obtained for the SYBR Green I one-step real-time quantitative RT-PCR were: slope −3.075 (the ideal value is −3.200), percentage efficiency (EFF) 101.4%; correlation coefficient (R2) 0.995 (the ideal value is <1.0) and Y-inter 35.312. The CT values obtained from ARAV RNA dilutions were reproducible. Therefore, only one TM peak between 80.83 ± 0.89 °C was observed for many of the tested ARAV RNA samples, as shown in the melting curves in Figure 1A and with the samples in Figure 1B.
3.2. Reproducibility, Detection Limit and Specificity of the Reaction
The reaction was reproducible based on triplicate analysis of in vitro transcribed ARAV RNA. Positive samples showed only a small variation in parameters: TM: 80.64 ± 0.67 °C; slope: 3.047 ± 0.068; R2: 0.977 ± 0.021 and CT with variations of 3%–7% in each triplicate, as previously described by the manufacturers. The negative control was water or buffer instead of sample template (buffer EB). The positive control was ARAV RNA in vitro-transcribed, equivalent to that used in the standard curve.
The detection limit of the test was 10 copies/µL of RNA based on triplicate analysis of decimal dilutions of in vitro transcribed ARAV RNA (3.7 × 1011 copies/µL to 3.7 × 10−3 copies/µL of in vitro transcribed ARAV RNA).
However, this one-step SYBR Green I real-time quantitative RT-PCR was able to amplify the genome of RIOMV, showing similar TM peaks obtained for ARAV. The one-step SYBR Green I real-time quantitative RT-PCR did not amplify the genomes of others viruses including Coxsackie B5, Influenza A, Human Respiratory Syncytial virus A, Parainfluenza-2, Metapneumovirus, Oropouche and Rhinovirus-39.
3.3. Comparison of the One-Step SYBR Green Real-Time RT-PCR with Conventional RT-PCR
The results obtained by conventional RT-PCR were compared with those of the SYBR Green I one-step real-time quantitative RT-PCR, and real-time quantitative RT-PCR appeared to be a more sensitive method. A hantavirus genome was detected and quantified by real-time quantitative RT-PCR in two serum samples of HPS patients that were negative by conventional RT-PCR, as shown in Table 1. Five lung samples from wild rodents that were positive by conventional RT-PCR were also positive by the real-time quantitative RT-PCR.
|Table 1. Comparison of conventional RT-PCR and one-step SYBR Green I real-time RT-PCR for diagnosis of Araraquara virus.|
|N. S||IgG ELISA||IgM ELISA||RT-PCR||One-step SYBR Green I real-time RT-PCR Copies/µL of RNA 1||Conversion for Copies/mL of serum or Copies/mg of tissue|
|H3||IgG-1,600||Negative||Positive||6.93 × 103 copies/µL||2.47 × 106 copies/mL|
|H5||IgG-3,200||Negative||Positive||8.85 × 102 copies/µL||3.15 × 105 copies/mL|
|H6||IgG-800||Negative||Positive||1.06 × 104 copies/µL||3.78 × 106 copies/mL|
|H7||IgG-3,200||Positive||Negative||1.03 × 102 copies/µL||3.67 × 104 copies/mL|
|H9||IgG-1,600||Positive||Positive||2.72 × 103 copies/µL||9.71 × 105 copies/mL|
|H11||IgG-6,400||Negative||Negative||7.4 × 101 copies/µL||2.64 × 104 copies/mL|
|H12||IgG-1,600||Negative||Positive||1.55 × 103 copies/µL||5.53 × 105 copies/mL|
|H13||IgG-1,600||Positive||Positive||1.34 × 103 copies/µL||4.78 × 105 copies/mL|
|H14||IgG-800||Positive||Positive||8.72 × 102 copies/µL||3.11 × 105 copies/mL|
|H16||IgG-1,600||Positive||Positive||3.87 × 103 copies/µL||1.38 × 106 copies/mL|
|N1||-||-||Positive||1.39 × 103 copies/µL||2.48 × 103 copies/mg|
|N3||-||-||Positive||8.9 × 104 copies/µL||1.58 × 105 copies/mg|
|N4||-||-||Positive||1.32 × 104 copies/µL||2.35 × 104 copies/mg|
|N8||-||-||Positive||1.78 × 103 copies/µL||3.17 × 103 copies/mg|
|N10||-||-||Positive||2.68 × 102 copies/µL||4.78 × 102 copies/mg|
N. S: Number sample; H: Serum samples of HPS patients; N: Lung tissues from Necromys lasiurus rodents. 1 The hantavirus load of clinical samples obtained by plotting on a standard curve, expressed in copies/µL of RNA.
PCR has been used extensively worldwide for the diagnosis of viral infections, including those caused by hantaviruses . Real-time PCR has several advantages over conventional PCR. It is faster, more sensitive, and reproducible, allows for quantitative measurement, has lower hands-on-time and lower risk of contamination [9,18].
Recently, some quantitative real-time RT-PCRs, using TaqMan, were developed for quantitation of hantavirus [10,14,18]. Here, we report, for the first time, a quantitative real time SYBR Green-based technique for the detection and quantitation of ARAV and RIOMV, being an important cause of HPS in the Southeast and Central Plateau of Brazil .
The main reason for the choice of a detection method based on SYBR Green is that RNA viruses present a diagnostic challenge. After consecutive replications, RNA viruses can generate genomic alterations, making them difficult to detect. Thus, the use of TaqMan probes, which are very specific, may produce false negative diagnoses due to genomic alterations in the binding sites with the probe .
The one-step SYBR Green I real-time quantitative RT-PCR was highly sensitive, being able to detect 10 copies/µL of RNA. The test was also reproducible, based on equivalent melting curves and amplicon sizes obtained from hantavirus-infected samples in multiple reactions [13,20,21]. Moreover, the latter is a lower-cost reaction compared with other real-time RT-PCRs, such as TaqMan [9,21].
The amplicons obtained with the one-step SYBR Green I quantitative RT-PCR were undoubtedly part of the hantavirus genome because the primers used amplify part of the N gene of ARAV. This amplification occurs in a conserved region of the N gene; in fact, a previous analysis showed that these primers were also able to amplify the genomes of other South American hantaviruses . Thus, the amplification of RNA from RIOMV showed TM peaks similar to ARAV.
The specificity of the reaction was confirmed by the TM curve, which was consistently specific for the amplicon obtained; the mean peak Tm obtained with curves specific for hantavirus amplicons will enable easier discrimination of a primer-dimers and other viruses [13,20,21]. In addition, no amplification of genomes from other RNA viruses was obtained using the reaction.
The standard curve of the SYBR Green I real-time quantitative RT-PCR was based on reactions using a vector containing 263 nucleotides of the S segment of the ARAV. The vector had been previously treated with DNAse I, preventing the amplification of remnants of plasmid DNA and erroneous measurements. Furthermore, DNAse I treatment prevented fluorophore binding to any sequence of double-stranded DNA [9,20]. The hantavirus load of clinical samples was quantified by plotting their CT reaction values on a standard curve based on the CT values of 108–103 copies/µL of ARAV RNA in vitro-transcribed [10,22].
The one-step SYBR Green I quantitative RT-PCR was more sensitive for hantavirus diagnosis than conventional RT-PCR. From the sera of 20 human HPS patients, a hantavirus genome was amplified by 10 by the one-step SYBR Green I quantitative RT-PCR, including two samples that were not amplified by conventional RT-PCR. These two samples had low viral loads (1.03 × 102 and 7.4 × 101 copies/µL of RNA or 3.67 × 104 and 2.64 × 104 copies/mL of serum) that were likely below the detection capacity of conventional RT-PCR (Table 1).
The analysis of viral load obtained from serum samples of HPS patients, infected with ARAV, demonstrated high plasma levels of viral RNA during acute infection (2.64 × 104–3.78 × 106 copies/mL), similar to those found in others studies [23,24], which analyzed plasma viral load in patients infected with Sin Nombre virus and was found to range between 104.7–107.5 and 1.3 × 104–1.8 × 106 copies/mL (Table 1).
Our data corroborates with results that demonstrated high plasmatic viral load of Puumala virus and Dobrava virus in patients with hemorrhagic fever with renal syndrome (HFRS) and Nephropathia epidemica (NE) (102–108 and 1.71 × 102–3 × 106 copies/mL) [13,25]. These reports demonstrate that IgG antibody values are inversely correlated with virus concentration. This result can be an association between high antibody titles with low viremia (Table 1).
The analysis of viral load obtained from macerated lung samples of Necromys lasiurus showed variable levels of viral RNA (4.78 × 102–1.58 × 105 copies/mg) (Table 1). A study with experimental infection of Peromyscus maniculatus showed that the highest titers of viral RNA are found 21 days post infection, and decline 60 days post infection .
The one-step SYBR Green I quantitative RT-PCR was performed more rapidly than conventional RT-PCR, taking approximately 2 h. This assay could be useful for studies of the pathogenesis of hantaviruses and for the analysis of the effect of antiviral drugs for ARAV and RIOMV [11,13]. Furthermore, it may be that this technique is able to detection of other hantavirus South American, due to the specificity of the primers used for these hantavirus described by Moreli et al. .
We are grateful to Eurico Arruda Neto, Miriã Ferreira Criado, and José Luiz Proença-Modena, for kindly donating the RNA samples of respiratory viruses and Oropouche virus. This study was supported by FAPESP - Fundação de Amparo a Pesquisa do Estado de São Paulo (Grant Number: 2008/50617-6; 2009/17518-9; 2011/05067-0; 2011/16866-1), Brazil.
Conflicts of Interests
The authors declare no conflict of interest.
References and Notes
- Jonsson, C.B.; Figueiredo, L.T.; Vapalahti, O. A global perspective on hantavirus ecology, epidemiology, and disease. Clin. Microbiol. Rev. 2010, 23, 412–441, doi:10.1128/CMR.00062-09.
- Hjelle, B.; Torres-Perez, F. Hantaviruses in the Americas and their role as emerging pathogens. Viruses 2010, 2, 2559–2586, doi:10.3390/v2122559.
- Figueiredo, L.T.; Moreli, M.L.; de-Sousa, R.L.; Borges, A.A.; de-Figueiredo, G.G.; Machado, A.M.; Bisordi, I.; Nagasse-Sugahara, T.K.; Suzuki, A.; Pereira, L.E.; et al. Hantavirus pulmonary syndrome, central plateau, southeastern, and southern Brazil. Emerg. Infect. Dis. 2009, 15, 561–567, doi:10.3201/eid1504.080289.
- Travassos da Rosa, E.S.; Medeiros, D.B.; Nunes, M.R.; Simith, D.B.; de Souza Pereira, A.; Elkhoury, M.R.; Lavocat, M.; Marques, A.A.; Via, A.V.; D’Andrea, P.; et al. Pygmy rice rat as potential host of Castelo dos Sonhos Hantavirus. Emerg. Infect. Dis. 2011, 17, 1527–1530, doi:10.3201/eid1708.101547.
- Rosa, E.S.; Mills, J.N.; Padula, P.J.; Elkhoury, M.R.; Ksiazek, T.G.; Mendes, W.S.; Santos, E.D.; Araujo, G.C.; Martinez, V.P.; Rosa, J.F.; et al. Newly recognized hantaviruses associated with hantavirus pulmonary syndrome in northern Brazil: Partial genetic characterization of viruses and serologic implication of likely reservoirs. Vector Borne Zoonotic Dis. 2005, 5, 11–19, doi:10.1089/vbz.2005.5.11.
- Firth, C.; Tokarz, R.; Simith, D.B.; Nunes, M.R.; Bhat, M.; Rosa, E.S.; Medeiros, D.B.; Palacios, G.; Vasconcelos, P.F.; Lipkin, W.I. Diversity and distribution of hantaviruses in South America. J. Virol. 2012, 86, 13756–13766, doi:10.1128/JVI.02341-12.
- Vaheri, A.; Vapalahti, O.; Plyusnin, A. How to diagnose hantavirus infections and detect them in rodents and insectivores. Rev. Med. Virol. 2008, 18, 277–288, doi:10.1002/rmv.581.
- Vapalahti, O.; Mustonen, J.; Lundkvist, A.; Henttonen, H.; Plyusnin, A.; Vaheri, A. Hantavirus infections in Europe. Lancet Infect. Dis. 2003, 3, 653–661, doi:10.1016/S1473-3099(03)00774-6.
- Mackay, I.M. Real-time PCR in the microbiology laboratory. Clin. Microbiol. Infect. 2004, 10, 190–212, doi:10.1111/j.1198-743X.2004.00722.x.
- Kramski, M.; Meisel, H.; Klempa, B.; Kruger, D.H.; Pauli, G.; Nitsche, A. Detection and typing of human pathogenic hantaviruses by real-time reverse transcription-PCR and pyrosequencing. Clin. Chem. 2007, 53, 1899–1905, doi:10.1373/clinchem.2007.093245.
- Jakab, F.; Sebok, J.; Ferenczi, E.; Horvath, G.; Szucs, G. First detection of Dobrava hantavirus from a patient with severe haemorrhagic fever with renal syndrome by SYBR Green-based real time RT-PCR. Scand. J. Infect. Dis. 2007, 39, 902–906, doi:10.1080/00365540701387072.
- Araujo, J.; Pereira, A.; Nardi, M.S.; Henriques, D.A.; Lautenschalager, D.A.; Dutra, L.M.; Ometto, T.L.; Hurtado, R.F.; Maues, F.; Nava, A.; et al. Detection of hantaviruses in Brazilian rodents by SYBR-Green-based real-time RT-PCR. Arch. Virol. 2011, 156, 1269–1274, doi:10.1007/s00705-011-0968-1.
- Evander, M.; Eriksson, I.; Pettersson, L.; Juto, P.; Ahlm, C.; Olsson, G.E.; Bucht, G.; Allard, A. Puumala hantavirus viremia diagnosed by real-time reverse transcriptase PCR using samples from patients with hemorrhagic fever and renal syndrome. J. Clin. Microbiol. 2007, 45, 2491–2497, doi:10.1128/JCM.01902-06.
- Trombley, A.R.; Wachter, L.; Garrison, J.; Buckley-Beason, V.A.; Jahrling, J.; Hensley, L.E.; Schoepp, R.J.; Norwood, D.A.; Goba, A.; Fair, J.N.; et al. Comprehensive panel of real-time TaqMan polymerase chain reaction assays for detection and absolute quantification of filoviruses, arenaviruses, and new world hantaviruses. Am. J. Trop. Med. Hyg. 2010, 82, 954–960, doi:10.4269/ajtmh.2010.09-0636.
- Botten, J.; Mirowsky, K.; Kusewitt, D.; Ye, C.; Gottlieb, K.; Prescott, J.; Hjelle, B. Persistent Sin Nombre virus infection in the deer mouse (Peromyscus maniculatus) model: Sites of replication and strand-specific expression. J. Virol. 2003, 77, 1540–1550, doi:10.1128/JVI.77.2.1540-1550.2002.
- Moreli, M.L.; Sousa, R.L.; Figueiredo, L.T. Detection of Brazilian hantavirus by reverse transcription polymerase chain reaction amplification of N gene in patients with hantavirus cardiopulmonary syndrome. Mem. Inst. Oswaldo Cruz 2004, 99, 633–638, doi:10.1590/S0074-02762004000600018.
- Figueiredo, L.T.; Moreli, M.L.; Borges, A.A.; de Figueiredo, G.G.; Badra, S.J.; Bisordi, I.; Suzuki, A.; Capria, S.; Padula, P. Evaluation of an enzyme-linked immunosorbent assay based on Araraquara virus recombinant nucleocapsid protein. Am. J. Trop. Med. Hyg. 2009, 81, 273–276. 19635882
- Naslund, J.; Kerner, A.; Drobni, P.; Bucht, G.; Evander, M.; Ahlm, C. Detection of Puumala and Rift Valley Fever virus by quantitative RT-PCR and virus viability tests in samples of blood dried and stored on filter paper. J. Virol. Methods 2011, 178, 186–190, doi:10.1016/j.jviromet.2011.09.010.
- Papin, J.F.; Vahrson, W.; Dittmer, D.P. SYBR green-based real-time quantitative PCR assay for detection of West Nile Virus circumvents false-negative results due to strain variability. J. Clin. Microbiol. 2004, 42, 1511–1518, doi:10.1128/JCM.42.4.1511-1518.2004.
- Aitichou, M.; Saleh, S.S.; McElroy, A.K.; Schmaljohn, C.; Ibrahim, M.S. Identification of Dobrava, Hantaan, Seoul, and Puumala viruses by one-step real-time RT-PCR. J. Virol. Methods 2005, 124, 21–26, doi:10.1016/j.jviromet.2004.10.004.
- Pryor, R.J.; Wittwer, C.T. Real-time polymerase chain reaction and melting curve analysis. Methods Mol. Biol. 2006, 336, 19–32. 16916250
- Fronhoffs, S.; Totzke, G.; Stier, S.; Wernert, N.; Rothe, M.; Bruning, T.; Koch, B.; Sachinidis, A.; Vetter, H.; Ko, Y. A method for the rapid construction of cRNA standard curves in quantitative real-time reverse transcription polymerase chain reaction. Mol. Cell. Probes 2002, 16, 99–110, doi:10.1006/mcpr.2002.0405.
- Terajima, M.; Hendershot, J.D., 3rd.; Kariwa, H.; Koster, F.T.; Hjelle, B.; Goade, D.; DeFronzo, M.C.; Ennis, F.A. High levels of viremia in patients with the Hantavirus pulmonary syndrome. J. Infect. Dis. 1999, 180, 2030–2034, doi:10.1086/315153.
- Xiao, R.; Yang, S.; Koster, F.; Ye, C.; Stidley, C.; Hjelle, B. Sin Nombre viral RNA load in patients with hantavirus cardiopulmonary syndrome. J. Infect. Dis. 2006, 194, 1403–1409, doi:10.1086/508494.
- Saksida, A.; Duh, D.; Korva, M.; Avsic-Zupanc, T. Dobrava virus RNA load in patients who have hemorrhagic fever with renal syndrome. J. Infect. Dis. 2008, 197, 681–685, doi:10.1086/527485.
© 2013 by the authors; licensee MDPI, Basel, Switzerland. This article is an open access article distributed under the terms and conditions of the Creative Commons Attribution license (http://creativecommons.org/licenses/by/3.0/).