- freely available
Sensors 2007, 7(12), 3287-3298;
Abstract: An important step towards understanding the working principles of the mammalian hearing sensor is the concept of an active cochlear amplifier. Theoretical arguments and physiological measurements suggest that the active cochlear amplifiers originate from systems close to a Hopf bifurcation. Efforts to model the mammalian hearing sensor on these grounds have, however, either had problems in reproducing sufficiently close essential aspects of the biological example (Magnasco, M.O. Phys. Rev. Lett. 90, 058101 (2003); Duke, T. & Jülicher, F. Phys. Rev. Lett. 90, 158101 (2003)), or required complicated spatially coupled differential equation systems that are unfeasible for transient signals (Kern, A. & Stoop, R. Phys. Rev. Lett. 91, 128101 (2003)). Here, we demonstrate a simple system of electronically coupled Hopf amplifiers that not only leads to the desired biological response behavior, but also has real-time capacity. The obtained electronic Hopf cochlea shares all salient signal processing features exhibited by the mammalian cochlea and thus provides a simple and efficient design of an artificial mammalian hearing sensor.
Understanding hearing in general and more specifically the mammalian hearing organ, the cochlea, is a long-standing human endeavour. The first basic step was taken by H.L.F. Helmholtz, who proposed in 1863  the existence of a one-to-one correspondence between a position along the cochlear duct and a stimulation frequency to which, at the given position, the response is maximal (the tonotopic principle). Important steps that followed were von Békésy's discovery of traveling waves along the basilar membrane (BM) as the carriers of the auditory information (1928 ) and Gold's conjecture of an active amplification process within the cochlea (1948 ). The latter phenomenon was corroborated by the discovery of otoacoustic emissions (1978 ), the autonomous generation of sounds by the cochlea itself. Subsequent experiments have revealed that the active amplification is located in the outer hair cells (OHC) attached to the basilar membrane [5, 6]. It has been known for some time that systems close to bifurcation instabilities can act as small-signal amplifiers [7, 8], where each bifurcation type ads its own amplification signature.
Physiological measurements have shown that among mammals, the hearing characteristics vary little  and the measurement details  imply that the instability responsible for active amplification is of Hopf type. This led Eguíluz et al.  to propose the use of Hopf-type amplifiers as the basic elements for cochlear modeling (for a related result, see ). They argued that the Hopf nonlinearities could correctly capture the basic aspects of mammalian hearing: compression of the dynamic range; sharper tuning for lower intensity sounds, and the generation of combination tones. In turtles and bullfrogs, cells that are homologous to the mammalian hair cells (see ) have already been shown to display amplification profiles that are characteristic for systems close to a Hopf instability .
In the cochlea, pressure variations generated by incoming sounds are transformed into incompressible and inviscid hydrodynamic waves. As these waves move down the cochlea, they cause small BM displacements . Using x to denote the distance from the stapes along the unrolled cochlea, the system can be linearly described by a water-surface wave with fluid depth h, density ρ, surface mass density m and exponentially decreasing transversal stiffness . In this description, using a dispersion relation, the differential equation ruling the spatial energy distribution e(x) obtains the form 
The characteristics of the active contribution a are derived from a ωch-rescaled Hopf differential equation
2. Experimental setup
Electronic sensor design
For stationary inputs simulations of the differential equation (1) provide results that are close to those from the physiological example, if an appropriate coupling scheme is used . In the simulation approaches pursued, the optimal coupling scheme however remained elusive to us. Only when using a simplification of the previously very detailed biologically motivated coupling schemes, our electronic implementation yielded results of the desired accuracy. The construction of the device now provides us with a handle to reconsider critically the biophysical basis used for the simulation approach, and to draw thereof corresponding biophysical conclusions.
In addition to providing suboptimal results, for transient signals our simulations were confronted with excessive computational demand. This was our main motivation to resort to an electronic implementation that, while being a simplification of the software model, should take care to preserve the insights gained. For the design, we decomposed the cochlea into n sections of characteristic frequencies ωsi, i = 1, … , n, and endowed each one with properties of the passive hydrodynamic behavior and an active Hopf amplifier. Major challenges were to properly connect the passive/active components towards a section, and the sections towards a cascade representing the entire cochlea. The Hopf part (2) can be approximated in circuitry by using a combination of integrative summers and multipliers, resistors R and a capitance C . Using complex notation for the voltages vz = vx + jvy, the Hopf equation reads
In order to build a generic section, it was sufficient to specify the detuning between the passive frequency ωsi and the Hopf amplifier frequency, ϕs(i) := ωsi/ωchi < 1, i = 1, … , n, where n is the number of sections. This was our first design parameter. The second design parameter was the relationship between the characteristic frequencies of subsequent section frequencies
For the realization of the cochlea reported below, we chose, for simplicity, the two parameters independent from the section, as Ψ(i) = Ψ, ϕs(i) = ϕs, ⩝i. For a frequency range to be covered, we used the second design parameter to determine the number of composing sections and to evaluate the characteristic frequencies. From Eqs. (6), the corresponding electronic components were obtained, which leaves as the only free parameter the Hopf parameter μ, in the electronic realization expressed by the gain control vμ. As was outlined above, the Hopf parameter determines both the amplification strength and the tuning width (smaller |μ|'s lead to larger amplifications combined with sharper tuning widths). We chose an electronic realization that requires five sections to cover one octave. This choice provides an amplification profile close to that of the biophysical measurements, at a minimum number of components. The circuits were implemented using conventional hardware, obtainable in any electronic store. As a consequence, the supply voltage was chosen at 10 V. The electronic parts used were chosen according to cost and availability criteria only. A more careful design optimizing the impedances between the sections would substantially reduce the noise accumulated along the cochlea. Primary investigations have corroborated the expectation that the cochlea can easily be miniaturized and be cast in microelectronics. Complex sounds (such as the ‘click’ stimulations used) were obtained via a microphone (Sennheiser E845 with an appropriate pre-amplifier) or via the synthetic waveform generator (Agilent 33220A), followed by an analogue Hilbert transform that generated the real and the imaginary components of the signal. For the active tuning of the μ-parameter array we used the National Instruments 6723 I/O card.
In what follows we report on a realization using design parameters ϕs = 1.05−1 and Ψ = 0.84. This leads to a range of one octave (say the range from 1480 to 2960 Hz) by means of five components. The final cochlea prototype constructed embraced 24 sections.
Steady-state signal response
From the constructed electronic device, we measured the amplitude Amp = |v0| generated in response to varying input frequencies and stimulation strengths (pure-tone stimulations). The comparison of the measurements after the first, the second, the tenth and the twentieth section demonstrates that by passing through the sections, the signal is gradually shaped (see Fig. 2), where after a few sections, the response already attains its characteristic form, see Fig. 3. Using more sections in the electronic device shapes the measured frequency response ever closer to the biophysical measurements (compare with , Fig. 1).
In Fig. 3, the response from four-section cochlea covering the range range from 5920 to 9956 Hz is taken for two values of the Hopf parameter μ, in order to demonstrate the influence μ has on the amplification strength. Close to bifurcation, a discretization effect is observed that limits the minimal number of sections that can be used given a required gain. The strength of the latter is a function of the frequency distance between adjacent sections (Ψ) and of the resonance widths of the active amplifiers (i.e., of μ). An array of about 30 sections leads to a hearing sensor that for practical purposes is comparable to the mammalian hearing organ.
Moreover, in experiments with stimulations by more than one tone, two-tone suppression and combination-tone generation, as the salient nonlinear phenomena of mammalian hearing [24–27], are reproduced with great fidelity, see Fig. 4.
Transient signal response
The response to transient, broadband stimulation is of similar quality. In Fig. 5, the results for click stimulations are displayed. A comparison with the corresponding biophysical measurements  reveals a close match. Also other biophysical experiments with transient signals reported in  are reproduced with ease. In Fig. 6, the instantaneous frequency response is displayed for different stimulus levels. The measured response reproduces the logarithmic increase towards the steady-state response near the central frequency (see inset), as well as the fact that the steady-state response is somewhat higher than the central frequency. Our measurements extend beyond the first wave packet, which leads at the packet interfaces to systematic outliers of the measured instantaneous frequencies. Even the phase functions taken from the device match with the ones taken from the biological example, see Fig. 7. Response phases increasingly lag as the frequency of stimulation increases. As in the biological example, a shallow slope at low frequencies is observed that turns into a steeper slope towards and beyond the central frequency.
Another transient signal of interest are low-frequency amplitude modulated sounds. That is because such signals play a significant role in clinical procedures for the differential diagnosis of sensory hearing loss. Moreover, such signals may play a major role in speech intelligibility. Also in this case, the correspondence between the physiological observations and the corresponding response generated by our electronic device is extremely close, see Fig. 8.
Active tuning towards signals
The least perfect correspondence between the signals delivered by our device and the physiological example is found in Fig. 5. In the electronic device, the response to sound stimulations is asymmetric (stronger in the beginning, lesser at the end of the signal), whereas in the biological example this is almost perfectly symmetrical. This discrepancy, however, can be fixed and understood by the following arguments. The physiological measurements indicate a response-dependent value of μ, potentially implemented on the level of hair cells with its channel dynamics (small |μ| at signal onset, larger values upon increased response). Before the arrival of a strong stimulating sound, the biological cochlea appears to be tuned for weaker sounds, using small values of the tuning parameter |μ|. With the arrival of a strong click, the amplification seems to be quickly decreased, and then kept constant. The observed behavior corresponds well to the behavior (latencies) one would expect if the hair-cells are tuned by means of its efferent connections, monitored by instructions from the auditory centers.
By closely following the biological example in our design, it is in the nature of the design of our cochlea that such a dynamical adaption of μ can be implemented with little additional effort. Even more importantly, on slower time-scales, the auditory neuronal feedback loop might tune μ in order to suppress unwanted, and to enhance desired, signal components. Theoretical experiments exploiting these features are successful and raise the expectation that by proceeding on this way, we may substantially reduce the cocktail-party problem. Preliminary results demonstrating this effect have already been obtained and corroborate the effectiveness of this feature.
We have demonstrated that the Hopf cochlear amplifier concept together with the cochlear biophysics thus provides the design of an extremely sensitive and robust hearing sensor, where salient nonlinear signal processing characteristics (compressive nonlinearity, high sensitivity, two-tone suppression, combination-tone generation) are naturally implemented. The device has all ingredients to serve as a perfect hearing replacement device (cochlear implant), once we succeed in similarly improving the stimulation electrodes needed for implants in a qualitatively comparable manner. Currently, such an electrode is being developed in collaboration with EMPA.
This work was supported by the Swiss National Science Foundation SNF (grant 205321-108427 to R.S.).
- Helmholtz, H.L.F. Die Lehre von den Tonempfindungen als physiologische Grundlage für die Theorie der Musik; Vieweg, Braunschweig: Germany, 1863. [Google Scholar]
- von Békésy, G. Zur Theorie des Hörens. Die Schwingungsform der Basilarmembran. Phys. Z. 1928, 29, 793–810. [Google Scholar]
- Gold, T. Hearing. The physical basis of the action of the cochlea. Proc. R. Soc. London Biol. Sci. 1948, 135, 492–498. [Google Scholar]
- Kemp, D.T. Stimulated acoustic emissions from within the human auditory system. J. Acoust. Soc. Am. 1978, 64, 1386–1391. [Google Scholar]
- Brownell, W.E.; Bader, C.R.; Bertrand, D.; de Ribaupierre, Y. Evoked mechanical responses of isolated cochlear outer hair cells. Science 1985, 227, 194–196. [Google Scholar]
- Robles, L.; Ruggero, M.A. Mechanics of the Mammalian Cochlea. Physiological Reviews 2001, 81, 1305–1352. [Google Scholar]
- Wiesenfeld, K.; McNamara, B. Period-doubling systems as small-signal amplifiers. Phys. Rev. Lett. 1985, 55, 13–16. [Google Scholar]
- Derighetti, B.; Ravani, M.; Stoop, R.; Meier, P.F.; Brun, E.; Badii, R. Period-doubling lasers as small-signal detectors. Phys. Rev. Lett. 1985, 55, 1746–1748. [Google Scholar]
- Ruggero, M.A.; Temchin, A.N. Unexceptional sharpness of frequency tuning in the human cochlea. Proc. Natl. Acad. Sci. U.S.A. 2005, 102, 18614–18619. [Google Scholar]
- Geisler, C.D. From Sound to Synapse; Oxford Univ. Press: Oxford, UK, 1998. [Google Scholar]
- Eguíluz, V.M.; Ospeck, M.; Choe, Y.; Hudspeth, A.J.; Magnasco, M.O. Essential nonlinearities in hearing. Phys. Rev. Lett. 2000, 84, 5232–5535. [Google Scholar]
- Camalet, S.; Duke, T.; Jülicher, F.; Prost, J. Auditory sensitivity provided by self-tuned critical oscillations of hair cells. Proc. Natl. Acad. Sci. U.S.A. 2000, 97, 3183–3188. [Google Scholar]
- Göpfert, M.C.; Humphris, A.D.L.; Albert, J.T.; Robert, D.; Hendrich, O. Power gain exhibited by motile mechanosensory neurons in Drosophila ears. Proc. Natl. Acad. Sci. U.S.A. 2005, 102, 325–330. [Google Scholar]
- Martin, P.; Hudspeth, A.J. Compressive nonlinearity in the hair bundle's active response to mechanical stimulation. Proc. Natl. Acad. Sci. U.S.A. 2001, 98, 14386–14391. [Google Scholar]
- de Boer, E. The Cochlea. Springer Handbook of Auditory Research; Dallos, P., Popper, A.N., Fay, R.R., Eds.; Springer: New York, USA, 1996; pp. 258–317. [Google Scholar]
- Peterson, B.P.; Bogert, L.C. A dynamical theory of the cochlea. J. Acoust. Soc. Am. 1950, 22, 369–381. [Google Scholar]
- Kern, A.; Stoop, R. Essential Role of Couplings between Hearing Nonlinearities. Phys. Rev. Lett. 2003, 91, 128101. [Google Scholar]
- Whitham, G.B. Linear and Nonlinear Waves; Interscience Publishers: New York, USA, 1999. [Google Scholar]
- Lighthill, J. Waves in Fluids; Cambridge University Press: Cambridge, UK, 2002. [Google Scholar]
- Smith, R.J.; Dorf, R.C. Circuits, Devices and Systems; John Wiley & Sons: New York, USA, 1991. [Google Scholar]
- Zwislocki, J.J.; Nguyen, M. Place Code for Pitch: A Necessary Revision. Acta Otolaryngol. (Stockh.) 1999, 119, 140–145. [Google Scholar]
- Ruggero, M.A. Responses to sound of the basilar membrane of the mammalian cochlea. Curr. Opin. Neurobiol. 1992, 2, 449–456. [Google Scholar]
- Ruggero, M.A.; Narayan, S.S.; Temchin, A.N.; Recio, A. Mechanical bases of frequency tuning and neural excitation at the base of the cochlea. Proc. Natl. Acad. Sci. U.S.A. 2000, 97, 11744–11750. [Google Scholar]
- Robles, L.; Ruggero, M.A.; Rich, N.C. Two-tone distortion on the basilar membrane of the chinchilla cochlea. J. Neurophysiol. 1997, 77, 2385–2399. [Google Scholar]
- Ruggero, M.A.; Robles, L.; Rich, N.C. Two-tone suppression in the basilar membrane of the cochlea. J. Neurophysiol. 1992, 68, 1087–1099. [Google Scholar]
- Stoop, R.; Kern, A. Essential auditory contrast-sharpening is preneuronal. Proc. Natl. Acad. Sci. U.S.A. 2004, 101, 9179–9181. [Google Scholar]
- Stoop, R.; Kern, A. Two-tone suppression and combination tone generation as computations performed by the Hopf cochlea. Phys. Rev. Lett. 2004, 93, 268103. [Google Scholar]
- Recio, A.; Rhode, W.S. Basilar membrane responses to broadband stimuli. J. Acoust. Soc. Am. 2000, 108, 2281–2298. [Google Scholar]
- Rhode, W.S.; Recio, A. Basilar-membrane response to multicomponent stimuli in chinchilla. J. Acoust. Soc. Am. 2001, 110, 981–994. [Google Scholar]
© 2007 by MDPI http://www.mdpi.org