A Method for Direct Measurement of the First-Order Mass Moments of Human Body Segments

We propose a simple and direct method for measuring the first-order mass moment of a human body segment. With the proposed method, the first-order mass moment of the body segment can be directly measured by using only one precision scale and one digital camera. In the dummy mass experiment, the relative standard uncertainty of a single set of measurements of the first-order mass moment is estimated to be 1.7%. The measured value will be useful as a reference for evaluating the uncertainty of the body segment inertial parameters (BSPs) estimated using an indirect method.


Introduction
The knowledge of body segment inertial parameters (BSPs) is required in the field of biomechanics. These parameters can be accurately estimated using dual x-ray absorptiometry (DXA) [1][2][3]. Although some DXA-based two-dimensional computation methods can be considered a direct measurement, the instrumentation for DXA is expensive and is not available to most researchers. In addition, the DXA-

OPEN ACCESS
based three-dimensional BSP computation [4] is not a real direct measurement since it is necessary to assume mass densities to reconstruct a three-dimensional mass distribution. On the other hand, many simple models for estimating BSPs have also been proposed [5,6], and data derived from DXA are sometimes used as the standard.
In this paper, a simple and direct method for measuring the first-order mass moment, M (1) = ML, is proposed, where M is the mass of a body segment and L is the length from the axis of rotation to the position of the center of mass (CoM) of the segment.
In the proposed method, the first-order mass moment of a human body segment M (1) is obtained from the two positions of the CoM of the entire body, in which the body segment under test has two different rotation angles. Only one precision scale and one digital camera are used in the experiment.
The first-order mass moment obtained with the proposed method can be used to evaluate the validity of the values of the BSPs obtained from other non-direct measurement methods. The moment value is also used for the posture error analysis in measuring human body masses in spacecraft with Space Scale [7][8][9]. Figure 1 shows the principle of the top-down view of the proposed method. The center of mass of the entire body can be expressed using the center of mass of the body segment S 1 and that of the rest of the body S 2 as follows:

Methods
where CoM, CoM 1 , and CoM 2 are the center of mass of the entire body, of the body segment S 1 , and of the rest of the body S 2 , respectively. M, M 1 , and M 2 are the mass of the entire body, of the body segment S 1 , and of the rest of the body S 2 , respectively.
where suffixes α and β correspond to the two different states of the body obtained when the segment under test S 1 has two different rotation angles. When the rotation occurs in the horizontal plane, CoM 1,β can be expressed using the rotation matrix R(θ) and CoM 1,α as follows: where θ is the rotation angle of the segment under test S 1 from the initial state (expressed using suffix α) to the final state (expressed using suffix β). On the basis of Equations (2) and (3), CoM 2 M 2 can be eliminated as follows: where I is the identity matrix. Then, the first-order mass moment M (1) 1 of the segment under test S 1 can be calculated using the rotation matrix R(θ) -1 ; the center of mass of the entire body before and after the rotation of segment S 1 , i.e., CoM α and CoM β ; and the mass of the entire body M as follows: The values of CoM at the two different states are measured using a common technique [10,11]. Figure 2 shows the setup for measuring the CoM of an entire body.
The center of rotation of segment S 1 , CoR, is set as the origin of the coordinate system, O. The entire body is placed on a horizontal board, which is supported by the following three points: Support Point-1, Support Point-2, and Support Point-3. The weights applied at these three points are measured using a precision scale (capacity 3100 g, resolution: 0.1 g, precision: 1.5 g, model: GX-30KR, manufacturer: A and D Co., Ltd. (Japan)) one by one, while the attitude of the body is fixed. During the measurement, the three support points are supported by the scale or the height adjustment blocks whose total height is equal to that of the scale. CoM is calculated as follows: where P P1 , P P2 , and P P3 are the horizontal positions of Support Point-1, Support Point-2, and Support Point

Experiments
The first-order mass moment M (1) of a dummy mass is evaluated for demonstrating the performance of the proposed method. Figure 3 shows the overlapped pictures, which correspond to a set of measurements.  An aluminum bar (mass: 3.013 kg, length: 77.2 cm) is used as the segment under test S 1 , which simulates an arm or a leg under the test. The other aluminum block with a mass of 52.066 kg is also used as segment S 2 , which simulates the rest of the human body. Both S1 and S2 are placed on the horizontal board with a mass of 11.627 kg. The center of rotation CoR and the origin of the coordinate system O are defined at the end of segment S 1 . The first-order mass moment M (1) of segment S 1 around the origin O (the center of rotation CoR) is calculated to be 116.3 kg·cm. In the experiment, 30 sets of measurements are conducted with a rotation angle between 30° and 180°.

Experiments
The first-order mass moment M (1) of a leg of a human body is measured using the proposed method. Figure 5 shows the overlapped pictures, which correspond to a set of measurements. In the experiments, 15 sets of measurements are conducted with a constant rotation angle of 48.8°. The total mass of the subject human body, including the clothes, and that of the frame are 63.460 kg and 15.569 kg, respectively. The rotation center RoC is estimated from the overlapped pictures such as Figure 5.     (1) and their mean value M (1) mean versus the order of the set of measurements i.

Discussion
The relative standard uncertainty of a single set of measurements of M (1) in the dummy mass experiment is estimated to be 1.7%. The standard deviations of the values of M (1) in the experiments using the dummy mass for 30 measurements with various rotation angles and the human body for 15 measurements with a constant rotation angle are both 1.4% of the mean value. Although these error values are not directly comparable, the random errors of both the experiments are thought to be at the same level. This indicates that the change of posture of the other part of the human body during one set of measurements can be ignored even when the leg is rotated. With regard to the bias error, the bias error in the human body experiment might be slightly larger than that in the dummy mass experiment since the accuracy of the estimation of the rotation center CoR in the human body experiment might be lesser than that in the dummy mass experiment. In the experiments, the rotation canter CoR and the rotation angle θ are manually measured using the printed overlapped images that are taken by the fixed digital camera with the help of a straightedge and protractor. The accuracy of estimating the rotation center will be improved with more careful consideration [12,13], such as considering 3-dimensional anatomical structure.
The measured value will be useful as a reference for evaluating the uncertainty of the body segment inertial parameters (BSPs) estimated using an indirect method, in which the mass and the center of mass of a segment are estimated [5,6]. In the evaluation, the value of the first-order mass moment M (1) of a body segment, which is directly measured using the proposed method, is compared with the value, which is calculated from the mass, the center of mass and the center of rotation estimated using the indirect method.
In the experiment, only one scale is used for measuring the weights applied at the three supporting points of the board. Therefore, during the measurements of the three points, the subject should not be moved; this requires a certain amount of effort in the case of the human subject. If three scales are used, then the weights at the three points can be obtained immediately and the measurement time can be reduced.

Conclusions
A simple and direct method for measuring the first-order mass moment M (1) of the human body segment has been proposed. By using the proposed method, the first-order mass moment of the body segment can be directly measured by using only one precision scale and one digital camera. The relative standard uncertainty of a single set of measurements of M (1) in the dummy mass experiment is estimated to be 1.7%. The measured value will be useful as a reference for evaluating the uncertainty of the body segment inertial parameters (BSPs) estimated using an indirect method.