A Device for Automatically Measuring and Supervising the Critical Care Patient’S Urine Output
AbstractCritical care units are equipped with commercial monitoring devices capable of sensing patients’ physiological parameters and supervising the achievement of the established therapeutic goals. This avoids human errors in this task and considerably decreases the workload of the healthcare staff. However, at present there still is a very relevant physiological parameter that is measured and supervised manually by the critical care units’ healthcare staff: urine output. This paper presents a patent-pending device capable of automatically recording and supervising the urine output of a critical care patient. A high precision scale is used to measure the weight of a commercial urine meter. On the scale’s pan there is a support frame made up of Bosch profiles that isolates the scale from force transmission from the patient’s bed, and guarantees that the urine flows properly through the urine meter input tube. The scale’s readings are sent to a PC via Bluetooth where an application supervises the achievement of the therapeutic goals. The device is currently undergoing tests at a research unit associated with the University Hospital of Getafe in Spain. View Full-Text
Scifeed alert for new publicationsNever miss any articles matching your research from any publisher
- Get alerts for new papers matching your research
- Find out the new papers from selected authors
- Updated daily for 49'000+ journals and 6000+ publishers
- Define your Scifeed now
Otero, A.; Palacios, F.; Akinfiev, T.; Fernández, R. A Device for Automatically Measuring and Supervising the Critical Care Patient’S Urine Output. Sensors 2010, 10, 934-951.
Otero A, Palacios F, Akinfiev T, Fernández R. A Device for Automatically Measuring and Supervising the Critical Care Patient’S Urine Output. Sensors. 2010; 10(1):934-951.Chicago/Turabian Style
Otero, Abraham; Palacios, Francisco; Akinfiev, Teodor; Fernández, Roemi. 2010. "A Device for Automatically Measuring and Supervising the Critical Care Patient’S Urine Output." Sensors 10, no. 1: 934-951.